日志码,描述,保护代码,保护项目,事件,保护项目（En）,事件(En),"日志灯（-1:记录，0:提示,1:警告,2:1级错误,3:2级错误,4:3级错误）",保护原因类别（0.其他 1.负压）,N2 ncCode,保护项目(繁体),事件(繁体),编号(N2项目上传JSH Web端),3PT ncCode-工步类型,WX_CellCode保护码,WX_ErrorCode错误码,火警类别（1手动喷淋保护2自动喷淋保护）,C_Protect,详细日志显示格式,保护码复用字段说明,详细日志说明,详细日志显示格式(繁体),详细日志显示格式(EN),详细日志说明(繁体),详细日志说明(EN),HC_ErrorCode错误码
0x00000001,LOG_FINISH,,,测试完成,,Finish,0,0,,,測試完成,,,,,,,,,,,,,,
0x00000002,LOG_STOP,,,用户停止,,Stop,0,0,,,用戶停止,,,,,,,,,,,,,,
0x00000003,LOG_EXIT,,,关闭程序,,Exit,0,0,,,關閉程序,,,,,,,,,,,,,,
0x00000004,LOG_JUMP,,,跳转,,Jump,0,0,,,跳轉,,,,,,,,,,,,,,
0x00000005,LOG_PROTECT,,,安全停止,,Protect,2,0,,,安全停止,,,,,,,,,,,,,,
0x00000006,LOG_CONTINUE,,,接续,,Continue,0,0,,,接續,,,,,,,,,,,,,,
0x00000007,LOG_RESET_STEP,,,重置工步,,Reset step,0,0,,,重置工步,,,,,,,,,,,,,,
0x00000008,LOG_START,,,用户启动,,Start,0,0,,,用戶啟動,,,,,,,,,,,,,,
0x00000009,LOG_QUERY,,,搜索单元,,Query,0,0,,,搜索單元,,,,,,,,,,,,,,
0x0000000A,LOG_COMM_ERR,,,串口错误,,COM error,1,0,,,串口錯誤,115116,,,,,,,,,,,,,
0x0000000B,LOG_POWEROFF,,,断电,,Power off,1,0,,,斷電,115117,,,,,,,,,,,,,
0x0000000C,LOG_POWERON,,,上电,,Power on,0,0,,,上電,,,,,,,,,,,,,,
0x0000000D,ERR_STEP_STORE,,,工步存储错误,,Step store error,2,0,,,工步存儲錯誤,,,,,,,,,,,,,,30
0x0000000E,ERR_STEP_RETRIEVE,,,工步读取错误,,Step read error,2,0,,,工步讀取錯誤,,,,,,,,,,,,,,30
0x0000000F,ERR_STEP_CHECKSUM,,,工步校验错误,,Step checksum error,2,0,,,工步校驗錯誤,,,,,,,,,,,,,,30
0x00000010,ERR_STEP_SEQUENCE,,,工步不连续,,Step discontinuous,2,0,,,工步不連續,,,,,,,,,,,,,,30
0x00000011,ERR_STEP_NUMBER,,,工步号超出范围,,Step number out of range,2,0,,,工步號超出範圍,,,,,,,,,,,,,,30
0x00000012,ERR_STACK_OVERFLOW,,,内部循环堆栈溢出,,Inner loop stack overflow,2,0,,,內部循環堆棧溢出,,,,,,,,,,,,,,
0x00000013,ERR_CONFIG_RETRIVE,,,上电取配置信息失败,,Get configuration information failed when power on,2,0,,,上電取配置信息失敗,,,,,,,,,,,,,,
0x00000014,ERR_NET_FAULT,,,网络阻塞,,Internet fault,2,0,,,網絡阻塞,,,,,,,,,,,,,,
0x00000015,ERR_DISLINK,,,服务器断开,,Server dislink,2,0,,,服務器斷開,,,,,,,,,,,,,,
0x00000016,ERR_LINK,,,服务器连接成功,,Server link,0,0,,,服務器連接成功,,,,,,,,,,,,,,
0x00000017,ERR_INST_OVERFLOW,,,串口指令队列溢出,,COM instruction queue overflow,2,0,,,串口指令隊列溢出,,,,,,,,,,,,,,
0x00000018,LOG_MID_RESTART,,,中位机重启,,Restart Device,0,0,,,中位機重啟,115118,,,,,,,,,,,,,
0x00000019,LOG_MID_FLASH_FULL,,,中位机flash满,,Mid flash full,0,0,,,中位機flash滿,,,,,,,,,,,,,,90
0x0000001A,LOG_CYCLE_NEST_OUT,,,嵌套循环超出三层,,Nested loops in excess of three layer,2,0,,,嵌套循環超出三層,,,,,,,,,,,,,,
0x0000001B,LOG_PAUSE,,,暂停,,Pause,0,0,,,暫停,,,,,,,,,,,,,,
0x0000001C,LOG_MID_LINK,,,中位机连接服务器成功,,ZWJ connet to server success,-1,0,,,中位機連接服務器成功,,,,,,,,,,,,,,
0x0000001D,LOG_MIGRATE,,,迁移,,Move,0,0,,,遷移,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：迁移原始通道,实际值1：迁移目标通道","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：遷移原始通道,實際值1：遷移目標通道",Set value: Migrate the original channel. Actual value 1: migrate the target channel,
0x0000001E,LOG_DATA_BACKUP_START,,,开始备份数据,,Start to backup data,0,0,,,開始備份數據,,,,,,,,,,,,,,
0x0000001F,LOG_DATA_BACKUP_COMPLETE,,,备份数据完成,,Data backup is complete,0,0,,,備份數據完成,,,,,,,,,,,,,,
0x00000020,LOG_COMM_ERR,,,串口错误,,COM error,1,0,,,串口錯誤,115119,,,,,,,,,,,,,
0x00000021,LOG_UPGRADE_FAILED,,,升级失败,,The upgrade was failed,2,0,,,升級失敗,,,,,,,,,,,,,,
0x00000022,LOG_UPGRADE_SUCCESS,,,升级成功,,The upgrade was successful,0,0,,,升級成功,,,,,,,,,,,,,,
0x00000023,LOG_UPGRADE_START,,,升级中位机,,Start to upgrade ZWJ,0,0,,,升級中位機,,,,,,,,,,,,,,
0x00000024,LOG_DATA_BACKUP_FAILED,,,备份数据失败,,Data backup was failed,2,0,,,備份數據失敗,,,,,,,,,,,,,,
0x00000025,LOG_RECV_DATA_SUCCESS,,,恢复数据成功,,Recover date was successful,0,0,,,恢復數據成功,,,,,,,,,,,,,,
0x00000026,LOG_SET_MANUAL_RTC,,,设置手动同步RTC时间,,Set up manually synchronize RTC time,0,0,,,設置手動同步RTC時間,,,,,,,,,,,,,,
0x00000027,LOG_SET_AUTO_RTC,,,设置自动同步RTC时间,,Set up automatically synchronize RTC time,0,0,,,設置自動同步RTC時間,,,,,,,,,,,,,,
0x00000028,LOG_SYNC_RTC,,,同步RTC时间,,Synchronize RTC time,-1,0,,,同步RTC時間,,,,,,,,,,,,,,
0x00000031,LOG_CC_FIRST,1.1,第一次充电(单步),在充电时间内充电电压电压下限设置值大于实际电压值,First charge(single),Do not charge to the setting voltage in specified time,2,0,,第一次充電(單步),在充電時間內充電電壓電壓下限設置值大於實際電壓值,115064,,E0,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x00000032,LOG_CC,1.2,CC恒流充电(单步),在规定时间内电压超出设定的上限值,CC charge(single),Voltage beyond the limit in specified time,2,0,CGB40|CGB41,CC恆流充電(單步),在規定時間內電壓超出設定的上限值,115065,FJB67-1,E1,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x00000033,LOG_CC_RAISERATE,1.3,CC充电电压上升速率(单步),在规定的时间上电压的△V达到设定的限值,CC charge voltage rise speed(single),△V beyond the limit in specified time,2,0,CGB44,CC充電電壓上升速率(單步),在規定的時間上電壓的△V達到設定的限值,115066,FJB73-1,E2,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：前面电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：前面電壓值","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: previous voltage value",20
0x00000034,LOG_CC_ABSOLUTE,1.4,CC电流超差保护(单步),电流超出设定的绝对值,CC current abs bias protect(single),Current beyond the abs limit,2,0,CGB42,CC電流超差保護(單步),電流超出設定的絕對值,115067,FJB70-1,E3,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电流变化设置值,实际值1：电流变化实际值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：電流變化設置值,實際值1：電流變化實際值","Set value: Current change Set value, actual value 1: actual current change value",10
0x00000035,LOG_CC_PERCENT,1.5,CC电流超差保护(单步),电流超出设定的百分比,CC current percentage bias protect(single),Current beyond the percentage limit,2,0,CGB42,CC電流超差保護(單步),電流超出設定的百分比,115068,FJB70-1,E4,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电流变化设置值,实际值1：电流变化实际值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：電流變化設置值,實際值1：電流變化實際值","Set value: Current change Set value, actual value 1: actual current change value",10
0x00000036,LOG_CC_ABSO_FLUC,1.6,CC电压下降异常点绝对值保护(单步),"在设置的容量范围,电压下降异常点大于设置的绝对值",CC voltage bias protect(single),Voltage bias beyond the abs limit in specified capcity,2,0,,CC電壓下降異常點絕對值保護(單步),"在設置的容量範圍,電壓下降異常點大於設置的絕對值",115069,,E5,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压值,实际值2：上次电压值,实际值3：本次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓值,實際值2：上次電壓值,實際值3：本次電壓值","Set value: Voltage set value, actual value 1: actual voltage value, actual value 2: last voltage value, actual value 3: this voltage value",40
0x00000037,LOG_CC_OUTL_FLUC,1.7,CC电压下降异常点个数保护(单步),"[V3]在设置的容量范围,电压下降异常点个数超过设置数,且异常下降值超过设置的绝对值",CC voltage error count bias protect,Error count beyond the limit in specified capcity,2,0,CGB43,CC電壓下降異常點個數保護(單步),"[V3]在設置的容量範圍,電壓下降異常點個數超過設置數,且異常下降值超過設置的絕對值",115070,FJB71-1,E6,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：错误个数设置值,实际值1：实际错误个数值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：錯誤個數設置值,實際值1：實際錯誤個數值","Set value: Number of errors Set value, actual value 1: Actual number of errors",40
0x00000038,LOG_CV_ABSOLUTE,1.8,CV电压超差保护(单步),[V7]电压超出设定的绝对值,CV voltage abs bias protect(single),Voltage beyond the abs limit,2,0,CGB69,CV電壓超差保護(單步),[V7]電壓超出設定的絕對值,115071,FJBA8-3,E7,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：本次电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：本次電壓值,實際值3：上次電壓值","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x00000039,LOG_CV_PERCENT,1.9,CV电压超差保护(单步),[V7]电压超出设定的百分比,CV voltage percentage bias protect(single),Voltage beyond the percentage limit,2,0,CGB69,CV電壓超差保護(單步),[V7]電壓超出設定的百分比,115072,FJBA8-3,E8,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：本次电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：本次電壓值,實際值3：上次電壓值","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x0000003A,LOG_CV_ABSO_FLUC,2.01,CV电流上升保护(单步),"在设置的容量范围,电流上升超出设置的绝对值",CV current bias protect,Current bias beyond the limit in specified capcity,2,0,CGB70,CV電流上升保護(單步),"在設置的容量範圍,電流上升超出設置的絕對值",115073,,E9,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：实际电流值,实际值2：上次电流值,实际值3：本次电流值","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流設置值,實際值1：實際電流值,實際值2：上次電流值,實際值3：本次電流值","Setting value: current setting value, actual value 1: actual current value, actual value 2: last current value, actual value 3: current value",40
0x0000003B,LOG_CV_OUTL_FLUC,2.02,CV电流上升偏差保护(单步),"在设置的容量范围,电流上升异常点个数超过设置数,且异常上升值超过设置的绝对值",CV current error count bias protect(single),Error count beyond the limit in specified capcity,2,0,CGB70,CV電流上升偏差保護(單步),"在設置的容量範圍,電流上升異常點個數超過設置數,且異常上升值超過設置的絕對值",115074,FJBA9-3,F0,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：错误个数设置值,实际值1：实际错误个数值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：錯誤個數設置值,實際值1：實際錯誤個數值","Set value: Number of errors Set value, actual value 1: Actual number of errors",40
0x0000003C,LOG_DC_ABSOLUTE,2.03,DC电流超差保护(单步),电流超出设定的绝对值,DC current abs bias protect(single),Current beyond the abs limit,2,0,CGB56,DC電流超差保護(單步),電流超出設定的絕對值,115075,FJB89-2,F1,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电流变化设置值,实际值1：电流变化实际值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：電流變化設置值,實際值1：電流變化實際值","Set value: Current change Set value, actual value 1: actual current change value",10
0x0000003D,LOG_DC_PERCENT,2.04,DC电流超差保护(单步),电流超出设定的百分比,DC current percentage bias protect(single),Current beyond the percentage limit,2,0,CGB56,DC電流超差保護(單步),電流超出設定的百分比,115076,,F2,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电流变化设置值,实际值1：电流变化实际值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：電流變化設置值,實際值1：電流變化實際值","Set value: Current change Set value, actual value 1: actual current change value",10
0x0000003E,LOG_DC_VOL_LOW,2.05,DC欠压保护(单步),电压低于500mV,DC lower voltage protect(single),Voltage lower than 500mV,2,0,,DC欠壓保護(單步),電壓低於500mV,115077,,F3,,,,,,,,,,,20
0x0000003F,LOG_DC_ABSO_FLUC,2.06,DC电压上升速率保护(单步),"在设置的容量范围,电压上升值大于设置的绝对值",DC voltage abs bias protect(single),Voltage abs bias beyond the limit in specified capcity,2,0,CGB57,DC電壓上升速率保護(單步),"在設置的容量範圍,電壓上升值大於設置的絕對值",115078,,F4,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压值,实际值2：上次电压值,实际值3：本次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓值,實際值2：上次電壓值,實際值3：本次電壓值","Set value: Voltage set value, actual value 1: actual voltage value, actual value 2: last voltage value, actual value 3: this voltage value",40
0x00000040,LOG_DC_OUTL_FLUC,2.07,DC电压上升速率保护(单步),"在设置的容量范围,电压上升异常点个数超过设置数,且异常上升值超过设置的绝对值",DC voltage bias protect(single),Error count beyond the limit in specified capcity,2,0,CGB57,DC電壓上升速率保護(單步),"在設置的容量範圍,電壓上升異常點個數超過設置數,且異常上升值超過設置的絕對值",115079,FJB90-2|FJBA6-2,F5,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：错误个数设置值,实际值1：实际错误个数值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：錯誤個數設置值,實際值1：實際錯誤個數值","Set value: Number of errors Set value, actual value 1: Actual number of errors",40
0x00000041,LOG_DC_DOWNRATE,2.08,DC放电电压下降速率(单步),在规定的时间上电压的△V超出设定的限值,DC dcharge voltage descent speed(single),△V beyond the limit in specified time,2,0,CGB66,DC放電電壓下降速率(單步),在規定的時間上電壓的△V超出設定的限值,115080,,F6,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：前面电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：前面電壓值","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: previous voltage value",20
0x00000042,LOG_REST,2.09,电压跳动保护(单步),上一工步最后一条电压与当前工步的实时电压的差值超过设定的值,voltage sudden change(single),The bias between the first voltage in current step and the last voltage in the step before the current step beyond the limit,2,0,CGB30,電壓跳動保護(單步),上一工步最後一條電壓與當前工步的實時電壓的差值超過設定的值,115081,FJB51-4,F7,,,电压波动保护,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压差设置值,实际值1：实际电压差值,实际值2：上个工步最后电压,实际值3：此工步第一个电压","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓差設置值,實際值1：實際電壓差值,實際值2：上個工步最後電壓,實際值3：此工步第一個電壓","Set value: voltage difference Set value, actual value 1: actual voltage difference, actual value 2: the last voltage of the last working step, actual value 3: the first voltage of this working step",20
0x00000043,LOG_VOL_UP,2.1,过压保护(全局),超出电压上限,Up voltage protect(all),Beyond the voltage limit,2,0,CGB32|CGB81,過壓保護(全局),超出電壓上限,115082,FJBC6-0,F8,,,过充电压保护,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,20
0x00000044,LOG_VOL_DOWN,2.11,欠压保护(全局),低于电压下限,Lower voltage protect(all),Lower than the voltage limit,2,0,CGB33,欠壓保護(全局),低於電壓下限,115083,FJBC7-0,F9,,,过放电压保护,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压下限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓下限值,實際值1：當前電壓值",Set value: lower voltage limit. Actual value 1: current voltage value,20
0x00000045,LOG_CUR_UP,2.12,过流保护(全局),超出电流上限,Up current protect(all),Beyong the current limit,2,0,CGB84,過流保護(全局),超出電流上限,115084,FJBD1-0,G0,,,过充电流保护,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3},实际值3:{4}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：电流实际值,实际值2：采样电路放大系数,实际值3：设置电流放大系数","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3},實際值3:{4}","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}, actual value 3:{4}","設置值：電流設置值,實際值1：電流實際值,實際值2：采樣電路放大系數,實際值3：設置電流放大系數","Set value: current set value, actual value 1: actual current value, actual value 2: sampling circuit amplification factor, actual value 3: current amplification factor",10
0x00000046,LOG_CUR_DOWN,2.13,欠流保护(全局),低于电流下限,Lower current protect(all),Lower than the current limit,2,0,,欠流保護(全局),低於電流下限,115085,FJBD2-0,G1,,,过放电流保护,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3},实际值3:{4}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：电流实际值,实际值2：采样电路放大系数,实际值3：设置电流放大系数","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3},實際值3:{4}","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}, actual value 3:{4}","設置值：電流設置值,實際值1：電流實際值,實際值2：采樣電路放大系數,實際值3：設置電流放大系數","Set value: current set value, actual value 1: actual current value, actual value 2: sampling circuit amplification factor, actual value 3: current amplification factor",10
0x00000047,LOG_CAPACITY,2.14,容量保护(全局),容量超出标称容量的设定值倍,Capcity protect(all),Capcity beyond the setting value times of the nominal capcity,2,0,CGB83,容量保護(全局),容量超出標稱容量的設定值倍,115086,FJBC8-0,G2,,,充放电总容量保护,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x00000048,LOG_POWER_VOL,2.15,上电电压保护(全局),上电时的电压不在设定的范围,Power on voltage protect(all),Voltage is not in the setting range when power on,2,0,,上電電壓保護(全局),上電時的電壓不在設定的範圍,115087,,G3,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x00000049,LOG_FORM_CAP,2.16,化成结束容量保护(全局),化成阶段完成后，最后一个恒流充电工步的最后一条数据的容量不在设定的范围，上限,formation end capcity protect(all),"When the formation test ends, the last capcity in last CCCahrge step beyond the limit，the upper limit.",2,0,,化成結束容量保護(全局),化成階段完成後，最後一個恆流充電工步的最後一條數據的容量不在設定的範圍，上限,115088,FJBD0-0,G4,,,过充保护容量,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：容量设置值,实际值1：容量实际值","時間:{0},設置值:{1}Ah,實際值1:{2}Ah","Time :{0}, set value :{1}Ah, actual value :{2}Ah","設置值：容量設置值,實際值1：容量實際值","Set value: Capacity Set value, actual value 1: actual capacity value",40
0x0000004A,LOG_VOL_PICUP,2.17,电压回检偏差保护(全局),电压回检偏差超出设定的偏差值,Voltage inspect protect(all),Voltage inspect bias is not in the setting range,2,0,CGB86,電壓回檢偏差保護(全局),電壓回檢偏差超出設定的偏差值,115089,FJBD3-0,G5,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：偏差电压设置值,实际值1：偏差电压实际值,实际值2：主通道电压值,实际值3：辅助通道电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：偏差電壓設置值,實際值1：偏差電壓實際值,實際值2：主通道電壓值,實際值3：輔助通道電壓值","Set value: Set value of deviation voltage, actual value 1: actual value of deviation voltage, actual value 2: voltage of the main channel, actual value 3: voltage of the auxiliary channel",20
0x0000004B,LOG_AUX_VOL_UP,2.18,辅助通道过压保护(全局),辅助通道电压超出设定的电压上限,Aux up voltage protect(all),Aux voltage beyond the voltage limit,2,0,,輔助通道過壓保護(全局),輔助通道電壓超出設定的電壓上限,115090,,G6,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x0000004C,LOG_AUX_VOL_DOWN,2.19,辅助通道欠压保护(全局),辅助通道电压低于设定的电压下限,Aux lower voltage peotect(all),Aux voltage lower than the voltage limit,2,0,,輔助通道欠壓保護(全局),輔助通道電壓低於設定的電壓下限,115091,,G7,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x0000004D,LOG_TEM_DOWN,2.2,低温保护(全局),低于温度下限,Lower temperature protect(all),Lower than the temperature limit,2,0,,低溫保護(全局),低於溫度下限,115092,FJBC4-0,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：溫度實際值","Set value: Set temperature value, actual value 1: actual temperature value",60
0x0000004E,LOG_TEM_UP,2.21,超温保护(全局),超出温度上限,Up temperature protect(all),Beyond the temperature limit,2,0,CGB80,超溫保護(全局),[T3]超出溫度上限,115093,FJBC5-0,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：溫度實際值","Set value: Set temperature value, actual value 1: actual temperature value",60
0x0000004F,LOG_FLOWNAME,2.22,流程名保护(全局),保护条件无作用于流程名,Flow name protect(all),Protection conditions were not acting on the flow name,2,0,,流程名保護(全局),保護條件無作用於流程名,,,H1,,,,,,,,,,,160
0x00000050,LOG_SMOKE_ALARM,2.23,烟雾报警(全局),探测到烟雾信号,Smoke alarm(all),Detect smoke signals,3,0,CGB94,煙霧報警(全局),探測到煙霧信號,115094,,,Y0,1,,,,,,,,,
0x00000051,LOG_MID_COMM,2.24,中位机通信断开(全局),中位机与上位机网络断开,Device communication interrupt exception(all),"The communication between ZWJ and SWJ interrupted,ZWJ active and SWJ disconnect",-1,0,,中位機通信斷開(全局),中位機與上位機網絡斷開,115095,,,Y1,,,,,,,,,,90
0x00000052,LOG_DATA_LOSS,2.25,数据丢失异常(全局),下位机数据收集异常，上位机数据丢失,Data loss exception(all),"XWJ data collection exception,SWJ data lose",1,0,,數據丟失異常(全局),下位機數據收集異常，上位機數據丟失,115096,,,Y2,,,,,,,,,,100
0x00000053,LOG_BATT_REVERSE,2.26,电芯反接保护(全局),电芯反接,Battery reverse protect(all),Battery recerse,2,0,,電芯反接保護(全局),電芯反接,,,,,,,,,,,,,,
0x00000054,LOG_BATTER,2.27,电芯保护(全局),设备电源打开未启动，对电池充放电,Battery protect(all),Charge when no starting,2,0,,電芯保護(全局),設備電源打開未啟動，對電池充放電,,,H2,,,,,,,,,,,
0x00000055,LOG_CPC_RAISERATE,2.28,CPC充电电压上升速率(单步),在规定的时间上电压的△V超出设定的限值,CPC charge voltage rise speed(single),△V beyond the limit in specified time,2,0,,CPC充電電壓上升速率(單步),在規定的時間上電壓的△V超出設定的限值,115097,,H3,,,,,,,,,,,20
0x00000056,LOG_CPC_ABSO_FLUC,2.29,CPC电压绝对波动保护(单步),在设置的容量范围，电压波动时的误差超出设定的绝对值,CPC voltage bias protect(single),Voltage bias beyond the limit in specified capcity,2,0,,CPC電壓絕對波動保護(單步),在設置的容量範圍，電壓波動時的誤差超出設定的絕對值,115098,,H4,,,,,2X4X中位机无此日志,,,,,,40
0x00000057,LOG_CPC_OUTL_FLUC,2.3,CPC电压异常点波动保护(单步),在设置的容量范围，异常点个数超出设定的个数，且异常偏差超出设定的绝对值,CPC voltage error count protect(single),Error count beyond the limit in specified capcity,2,0,,CPC電壓異常點波動保護(單步),在設置的容量範圍，異常點個數超出設定的個數，且異常偏差超出設定的絕對值,115099,,H5,,,,,2X4X中位机无此日志,,,,,,40
0x00000058,LOG_CPD_DOWNRATE,2.31,CPD充电电压下降速率(单步),在设置的时间上电压的△V超出设定的限值,CPD charge voltage descent speed(single),△V beyond the limit in specified time,2,0,,CPD充電電壓下降速率(單步),在設置的時間上電壓的△V超出設定的限值,115100,,H6,,,,,2X4X中位机无此日志,,,,,,20
0x00000059,LOG_CPD_ABSO_FLUC,2.32,CPD电压绝对波动保护(单步),在设置的容量范围，电压波动时的误差超出设定的绝对值,CPD voltage bias protect(single),Voltage bias beyond the limit in specified capcity,2,0,,CPD電壓絕對波動保護(單步),在設置的容量範圍，電壓波動時的誤差超出設定的絕對值,115101,,H7,,,,,2X4X中位机无此日志,,,,,,40
0x0000005A,LOG_CPD_OUTL_FLUC,2.33,CPD电压异常点波动保护(单步),在设置的容量范围，异常点个数超出设定的个数，且异常偏差超出设定的绝对值,CPD voltage error count protect(single),Error count beyond the limit in specified capcity,2,0,,CPD電壓異常點波動保護(單步),在設置的容量範圍，異常點個數超出設定的個數，且異常偏差超出設定的絕對值,115102,,H8,,,,,2X4X中位机无此日志,,,,,,40
0x0000005B,LOG_CRC_DOWNRATE,2.34,CRC充电电压下降速率(单步),在设置的时间上电压的△V超出设定的限值,CRC charge voltage descent speed(single),△V beyond the limit in specified time,2,0,,CRC充電電壓下降速率(單步),在設置的時間上電壓的△V超出設定的限值,115103,,H9,,,,,2X4X中位机无此日志,,,,,,20
0x0000005C,LOG_CRC_ABSO_FLUC,2.35,CRC电压绝对波动保护(单步),在设置的容量范围，电压波动时的误差超出设定的绝对值,CRC voltage bias protect(single),Voltage bias beyond the limit in specified capcity,2,0,,CRC電壓絕對波動保護(單步),在設置的容量範圍，電壓波動時的誤差超出設定的絕對值,115104,,I0,,,,,2X4X中位机无此日志,,,,,,40
0x0000005D,LOG_CRC_OUTL_FLUC,2.36,CRC电压异常点波动保护(单步),在设置的容量范围，异常点个数超出设定的个数，且异常偏差超出设定的绝对值,CRC voltage error count protect(single),Error count beyond the limit in specified capcity,2,0,,CRC電壓異常點波動保護(單步),在設置的容量範圍，異常點個數超出設定的個數，且異常偏差超出設定的絕對值,115105,,I1,,,,,2X4X中位机无此日志,,,,,,40
0x00000061,LOG_CCC_SUD_VOLDOWN,,CC电压突降(单步),"[V4]在恒流充电时,电压突降的值超过设置的绝对值时保护",CC charge voltage swell(single),CC charge voltage swell(single),2,0,CGB45,CC電壓突降(單步),"[V4]在恆流充電時,電壓突降的值超過設置的絕對值時保護",115106,FJB74-1,I2,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x00000062,LOG_CVC_ SUD_CURUP,,CV电流突升(单步),"在恒压充电时,电流突升的值超过设置的绝对值时保护",CV charge current dip(single),CV charge current dip(single),2,0,CGB71,CV電流突升(單步),"在恆壓充電時,電流突升的值超過設置的絕對值時保護",115107,FJBB1-3,I3,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：实际电流变化值,实际值2：当前电流值,实际值3：上次电流值","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流設置值,實際值1：實際電流變化值,實際值2：當前電流值,實際值3：上次電流值","Setting value: current setting value, actual value 1: actual current change value, actual value 2: current current value, actual value 3: last current value",10
0x00000063,LOG_CCD_ SUD_VOLUP,,DC电压突升(单步),"在恒流放电时,电压突升的值超过设置的绝对值时保护",CC discharge voltage dip(single),CC discharge voltage dip(single),2,0,,DC電壓突升(單步),"在恆流放電時,電壓突升的值超過設置的絕對值時保護",115108,,I4,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x00000064,LOG_CVD_ SUD_CURUP,,DV电流突升(单步),"在恒压放电时,电流突升的值超过设置的绝对值时保护",CV discharge current dip(single),CV discharge current dip(single),2,0,,DV電流突升(單步),"在恆壓放電時,電流突升的值超過設置的絕對值時保護",115109,,I5,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：实际电流变化值,实际值2：当前电流值,实际值3：上次电流值","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流設置值,實際值1：實際電流變化值,實際值2：當前電流值,實際值3：上次電流值","Setting value: current setting value, actual value 1: actual current change value, actual value 2: current current value, actual value 3: last current value",10
0x00000065,LOG_VOL_UP_REST,,Rest过压保护（单步）,"首次搁置电压保护,超出电压上限",over voltage protection（rest）,the voltage is over the upper limit when resting for the first time,2,0,,Rest過壓保護（單步）,"首次擱置電壓保護,超出電壓上限",115110,FJB55-4,I6,,,,,2X4X中位机无此日志,,,,,,20
0x00000066,LOG_VOL_DOWN_REST,,Rest欠压保护（单步）,"首次搁置电压保护,低于电压下限",under voltage protection（rest）,the voltage is under the lower limit when resting for the first time,2,0,,Rest欠壓保護（單步）,"首次擱置電壓保護,低於電壓下限",115111,FJB56-4,I7,,,,,2X4X中位机无此日志,,,,,,20
0x00000067,LOG_CUR_UP_REST,,Rest过流保护（单步）,超出电流上限,over current protection（rest）,the current is over the upper limit,2,0,CGB31,Rest過流保護（單步）,超出電流上限,115112,FJB54-4,I8,,,,,2X4X中位机无此日志,,,,,,10
0x00000068,LOG_CUR_DOWN_REST,,Rest欠流保护（单步）,低于电流下限,under current protection（rest）,the current is under the lower limit,2,0,CGB31,Rest欠流保護（單步）,低於電流下限,115113,FJB54-4,I9,,,,,2X4X中位机无此日志,,,,,,10
0x00000069,LOG_POWEROFF_PROTECT,,掉电保护,掉电,poweroff protection-poweroff,poweroff protection-poweroff,2,0,,掉電保護,掉電,115114,,,Q6,,,,,,,,,,
0x0000006A,LOG_POWER_ABNORMAL_OFF,,异常断电保护,异常断电,abnormal poweroff protection,abnormal poweroff,2,0,,異常斷電保護,異常斷電,115115,,,Q7,,,,,,,,,,
0x00000071,LOG_MEASUR_PRO,,,测量量保护,,Measure protect,2,0,,,測量量保護,,,,,,,,2X4X中位机无此日志,,,,,,160
0x00000072,LOG_MANUAL_START,,,用户手动启动,,Manual start,0,0,,,用戶手動啟動,,,J0,,,,,2X4X中位机无此日志,,,,,,
0x00000073,LOG_RUN_REST,,,用户在运行状态重启,,Restart when in the running state,0,0,,,用戶在運行狀態重啟,,,J1,,,,,2X4X中位机无此日志,,,,,,
0x00000074,LOG_PROT_SUSP_REST,,,用户在保护/暂停状态下重启,,Restart when in the protection or stoping state,0,0,,,用戶在保護/暫停狀態下重啟,,,J2,,,,,2X4X中位机无此日志,,,,,,
0x00000075,LOG_MANUAL_SUSP,,,用户手动暂停,,Manual suspended,0,0,,,用戶手動暫停,,,J3,,,,,2X4X中位机无此日志,,,,,,
0x00000076,LOG_RUN_PAUSE,,,运行到Pause工步,,Run puase,0,0,,,運行到Pause工步,,,,,,,,2X4X中位机无此日志,,,,,,
0x00000077,LOG_MANUAL_STOP,,,用户手动停止,,Manual stop,0,0,,,用戶手動停止,,,J4,,,,,2X4X中位机无此日志,,,,,,
0x00000078,LOG_RUN_END,,,运行到End工步,,Run end,0,0,,,運行到End工步,,,,,,,,2X4X中位机无此日志,,,,,,
0x00000079,LOG_RUN_JUMP,,,用户在运行状态跳转,,Jump when in the running state,0,0,,,用戶在運行狀態跳轉,,,,,,,,2X4X中位机无此日志,,,,,,
0x0000007A,LOG_PROT_SUSP_JUMP,,,用户在保护/暂停状态下跳转,,Jump when in the protection or stoping state,0,0,,,用戶在保護/暫停狀態下跳轉,,,,,,,,2X4X中位机无此日志,,,,,,
0x0000007B,LOG_MANUAL_CONT,,,用户手动接续,,Manual continue,0,0,,,用戶手動接續,,,,,,,,2X4X中位机无此日志,,,,,,
0x0000007C,LOG_POWOFF_CONT,,,断电保护后自动接续,,Auto continue when powoff,0,0,,,斷電保護後自動接續,,,,,,,,2X4X中位机无此日志,,,,,,
0x0000007D,LOG_RUN_RESET,,,用户在运行状态重置,,Reset when in the running state,0,0,,,用戶在運行狀態重置,,,,,,,,2X4X中位机无此日志,,,,,,
0x0000007E,LOG_PROT_SUSP_RESET,,,用户在保护/暂停状态下重置,,Reset when in the protection or stoping state,0,0,,,用戶在保護/暫停狀態下重置,,,,,,,,2X4X中位机无此日志,,,,,,
0x0000007F,LOG_RUN_MIGR,,,用户在运行状态迁移,,Move when in the running state,0,0,,,用戶在運行狀態遷移,,,,,,,,2X4X中位机无此日志,,,,,,
0x00000080,LOG_PROT_SUSP_MIGR,,,用户在保护/暂停状态下迁移,,Move when in the protection or stoping state,0,0,,,用戶在保護/暫停狀態下遷移,,,,,,,,2X4X中位机无此日志,,,,,,
0x00000081,LOG_POWER_BROWNOUT,,,电源掉电,,Power brownout,1,0,,,電源掉電,,,,Q8,,,,2X4X中位机无此日志,,,,,,
0x00000082,LOG_HDWARE_ERR,,,硬件出错,,hartware error,2,0,,,硬件出錯,,,,Q9,,,,2X4X中位机无此日志,,,,,,100
0x00000083,LOG_OPEN_PROT,,,开路测试保护,,Open circuit protect test,2,0,,,開路測試保護,,,J5,,,,,2X4X中位机无此日志,,,,,,160
0x00000084,LOG_SHORT_PROT,,,短路测试保护,,Short circuit protect test,2,0,,,短路測試保護,,,J6,,,,,2X4X中位机无此日志,,,,,,160
0x00000085,LOG_OTHER,,,其他错误,,other error,2,0,,,其他錯誤,,,,R0,,,,2X4X中位机无此日志,,,,,,
0x00000086,LOG_DATA_ERR,,,数据同步异常,,Data synchro abnorma,2,0,,,數據同步異常,,,,R1,,,,2X4X中位机无此日志,,,,,,
0x00000087,LOG_AUX_DIFF_VOLT_PRO,,,辅助通道电压差保护,,Aux channel voltage difference protect,2,0,,,輔助通道電壓差保護,,,J7,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：最大电压值,实际值3：最小电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：最大電壓值,實際值3：最小電壓值","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: maximum voltage value, actual value 3: minimum voltage value",20
0x00000088,NEILSBED_ABNORMAL,,,[S1]烟雾报警,,Smoke alarm,3,0,,,[S1]煙霧報警,,,,R2,1,,,,,,,,,
0x00000089,LOG_CV_CUR_ABSOLUTE,,,CV电流超差保护,,CV current absolute protect,2,0,,,CV電流超差保護,115120,,J8,,,,,,,,,,,10
0x0000008A,LOG_CC_VOL_ABSOLUTE,,,CC电压超差保护,,CC voltage absolute protect,2,0,,,CC電壓超差保護,115121,,J9,,,,,,,,,,,20
0x0000008B,LOG_FORMATION_VOL_CHANGE_RATE,,,化成电压变化率保护,,Formation vol change rate,2,0,,,化成電壓變化率保護,115122,,K0,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x0000008C,LOG_PRESSURE_LOWER,,,气压下限保护,,Pressure lower protect,2,1,,,氣壓下限保護,115123,FJB58-4,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x0000008D,LOG_PRESSURE_UPPER,,,[P1]气压上限保护,,Pressure upper protect,2,1,,,[P1]氣壓上限保護,115124,FJB58-4,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x0000008E,LOG_CAP_REACH,,,容量已经到达所设置容量值,,Capacity has reached the set value,0,0,,,容量已經到達所設置容量值,115125,,K1,,,,,2X4X中位机无此日志,,,,,,40
0x0000008F,LOG_IF_PROTECT_VOL,,,如果工步电压保护,,Condition step - voltage protection,2,0,,,如果工步電壓保護,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置电压值,实际值1：实际电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：設置電壓值,實際值1：實際電壓值",Set value: Set voltage value. Actual value 1: actual voltage value,20
0x00000090,LOG_IF_PROTECT_CUR,,,如果工步电流保护,,Condition step - current protection,2,0,,,如果工步電流保護,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置电流值,实际值1：实际电流值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：設置電流值,實際值1：實際電流值","Set value: Set current value, actual value 1: actual current value",10
0x00000091,LOG_IF_PROTECT_CAP,,,如果工步容量保护,,Condition step - capacity protection,2,0,,,如果工步容量保護,,,,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置容量值,实际值1：实际容量值","時間:{0},設置值:{1}Ah,實際值1:{2}Ah","Time :{0}, set value :{1}Ah, actual value :{2}Ah","設置值：設置容量值,實際值1：實際容量值",Set value: Set capacity value. Actual value 1: actual capacity value,40
0x00000092,LOG_IF_PROTECT_ENERGY,,,如果工步能量保护,,Condition step - energy protection,2,0,,,如果工步能量保護,,,,,,,"时间:{0},设置值:{1}Wh,实际值1:{2}Wh","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置能量值,实际值1：实际能量值","時間:{0},設置值:{1}Wh,實際值1:{2}Wh","Time :{0}, set value :{1}Wh, actual value :{2}Wh","設置值：設置能量值,實際值1：實際能量值",Set value: Set the energy value. Actual value 1: actual energy value,160
0x00000093,LOG_IF_PROTECT_POW,,,如果工步功率保护,,Condition step - power protection,2,0,,,如果工步功率保護,,,,,,,"时间:{0},设置值:{1}W,实际值1:{2}W","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置功率值,实际值1：实际功率值","時間:{0},設置值:{1}W,實際值1:{2}W","Time :{0}, set value :{1}W, actual value :{2}W","設置值：設置功率值,實際值1：實際功率值",Set value: Set power value. Actual value 1: actual power value,160
0x00000094,LOG_IF_PROTECT_STEP_TIME,,,如果工步工步时间保护,,Condition step - step time protection,2,0,,,如果工步工步時間保護,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置时间值,实际值1：实际时间值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：設置時間值,實際值1：實際時間值",Set value: Set time value. Actual value 1: actual time value,160
0x00000095,LOG_IF_PROTECT_CTRL_TEMP,,,如果工步环境温度保护,,Condition step - environment temperature protection,2,0,,,如果工步環境溫度保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,60
0x00000096,LOG_IF_STOP,,,如果工步停止,,Condition step to stop,0,0,,,如果工步停止,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：如果设置值,实际值1：实际值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：如果設置值,實際值1：實際值","Set value: If set, actual value 1: actual value",
0x00000097,LOG_NEILSBED_ABNORMAL_COMM_ERR,,,压床通讯异常(中位机与控制板),,Neilsbed communication abnormal,2,0,,,壓床通訊異常(中位機與控制板),115126,,,R3,,,,,,,,,,
0x00000098,LOG_NEILSBED_ABNORMAL_TRAY_UNPLACE,,,托盘未到位,,Trays are not in place,2,0,,,托盤未到位,115127,,,R4,,,,,,,,,,
0x00000099,LOG_NEILSBED_ABNORMAL_CYLINDER_PRESS_UNPLACE,,,压床未压到位,,Neilsbed is not in place,2,0,,,壓床未壓到位,115128,,,R5,,,,,,,,,,
0x0000009A,LOG_NEILSBED_ABNORMAL_NO_TRAY,,,无托盘,,No tray,2,0,,,無托盤,115129,,,R6,,,,,,,,,,
,,,,,,,,0,,,,,,,,,,,,,,,,,
,,,,,,,,0,,,,,,,,,,,,,,,,,
,,,,,,,,0,,,,,,,,,,,,,,,,,
,,,,,,,,0,,,,,,,,,,,,,,,,,
0x000000B0,LOG_RESTORE_STATE_FILE_CRC_ERR,,,"恢复通道状态时,状态文件校验错误",,"State-file-check error,on restoring the channel state",2,0,,,"恢復通道狀態時,狀態文件校驗錯誤",,,,R7,,,,,,,,,,
0x000000B1,LOG_RESTORE_STATE_FILE_FLAG_ERR,,,"恢复通道状态时,状态文件标志错误",,"State-file-flag error,on restoring the channel state",2,0,,,"恢復通道狀態時,狀態文件標誌錯誤",,,,R8,,,,,,,,,,
0x000000B2,LOG_RESTORE_STATE_FILE_CYCLE_ERR,,,"恢复通道状态时,恢复循环栈出错",,"Cycle-stack restoring error, on restoring the channel state",2,0,,,"恢復通道狀態時,恢復循環棧出錯",,,,R9,,,,,,,,,,
0x000000B3,LOG_RECV_INVALID_CMD,,,通道收到无效的命令消息,,Received invalid command-message in channel,2,0,,,通道收到無效的命令消息,,,,S0,,,,,,,,,,
0x000000B4,LOG_START_ALLOCK_ACH_ERR,,,分配不到辅助通道,,Unable to allocate any auxilary channel,2,0,,,分配不到輔助通道,,,,,,,,,,,,,,
0x000000B5,LOG_START_GET_TESTID_ERR,,,获取测试ID失败,,Failed to get test ID,2,0,,,獲取測試ID失敗,,,,S1,,,,,,,,,,
0x000000B6,LOG_START_STET_NO_ERR,,,分配不到辅助通道,,Step ID error,2,0,,,分配不到輔助通道,,,,,,,,,,,,,,
0x000000B7,LOG_LOAD_STEP_PARA_ERR,,,加载工步参数出错,,Step parameters loading error,2,0,,,加載工步參數出錯,115130,,,S2,,,,,,,,,,
0x000000B8,LOG_SEND_GET_STEP_EXT_PARA_ERR,,,发送获取工步附加资源消息失败,,Failed to send the message of getting extra resource of step,2,0,,,發送獲取工步附加資源消息失敗,,,,S3,,,,,,,,,,30
0x000000B9,LOG_EXE_STEP_FAIL,,,执行工步失败,,Failed to execute step,2,0,,,執行工步失敗,115131,,,S4,,,,,,,,,,30
0x000000BA,LOG_STOP_PARA_ERR,,,"通道停止时,参数错误",,"Parameter error,on stopping",2,0,,,"通道停止時,參數錯誤",,,,S5,,,,,,,,,,
0x000000BB,LOG_JUMP_PARA_ERR,,,"通道跳转时,参数错误",,"Parameter error,on jumping",2,0,,,"通道跳轉時,參數錯誤",,,,S6,,,,,,,,,,
0x000000BC,LOG_JUMP_CHL_STATE_ERR,,,在通道保护，停止，完成，暂停状态时收到跳转命令,,"Received jump-command,when channel is in the state of protecting/suspending/finished/pausing",2,0,,,在通道保護，停止，完成，暫停狀態時收到跳轉命令,,,,S8,,,,,,,,,,
0x000000BD,LOG_JUMP_CYCLE_STACK_GET_TOP_ERR,,,"跳转时,循环栈不为空,但取栈顶操作失败",,"Failed to pop nonempty cycle-stack,on jumping",2,0,,,"跳轉時,循環棧不為空,但取棧頂操作失敗",,,,S9,,,,,,,,,,
0x000000BE,LOG_JUMP_CYCLE_STACK_PUSH_ERR,,,"跳转时,循环栈出栈失败",,"Failed to pop cycle-stack,on jumping",2,0,,,"跳轉時,循環棧出棧失敗",,,,T0,,,,,,,,,,
0x000000BF,LOG_RESUME_ALLOCK_ACH_ERR,,,"接续时,分配不到辅助通道",,"Unable to allocate any auxilary channel, on continuing",2,0,,,"接續時,分配不到輔助通道",,,,T1,,,,,,,,,,
0x000000C0,LOG_RESUME_ALLOCK_PARALLEL_MCH_ERR,,,"接续时,分配不到并联的主通道",,"Unable to allocate paralleling main-channel, on continuing",2,0,,,"接續時,分配不到並聯的主通道",,,,T2,,,,,,,,,,
0x000000C1,LOG_RESUME_PARA_ERR,,,"接续时,参数错误",,"Parameter error, on continuing",2,0,,,"接續時,參數錯誤",,,,T3,,,,,,,,,,
0x000000C2,LOG_RESUME_STATE_ERR,,,在通道运行和完成状态时收到接续命令,,"Received continue-command,when channel in the state of running/finished",2,0,,,在通道運行和完成狀態時收到接續命令,,,,T4,,,,,,,,,,
0x000000C3,LOG_SEND_GET_STEP_MESSAGE_ERR,,,发送读取工步消息失败,,Failed to send the message of reading step,2,0,,,發送讀取工步消息失敗,,,,T5,,,,,,,,,,30
0x000000C4,LOG_SEND_GET_SIM_DATA_MESSAGE_ERR,,,发送读取模拟工况数据消息失败,,Failed to send the message of reading simulate-step data,2,0,,,發送讀取模擬工況數據消息失敗,,,,,,,,,,,,,,
0x000000C5,LOG_GET_SIM_DATA_ERR,,,"启动或工步跳转时,获取模拟工况数据发生错误",,"Error of getting simulate-step data,on starting or jumping",2,0,,,"啟動或工步跳轉時,獲取模擬工況數據發生錯誤",,,,,,,,,,,,,,
0x000000C6,LOG_GET_STEP_FILE_ERR,,,"启动或工步跳转时,获取工步文件错误",,"Error of getting step-file,on starting or jumping",2,0,,,"啟動或工步跳轉時,獲取工步文件錯誤",,,,T6,,,,,,,,,,
0x000000C7,LOG_SAMPLE_ERR,,,长时间连续采样数据错误,,Error of long time sampling,2,0,,,長時間連續採樣數據錯誤,115132,,,T7,,,,,,,,,,
0x000000C8,LOG_INVALID_ACH_TYPE_PROTECT_ERR,,,"保护判断时,出现无效的辅助通道类型",,"Invalid type of auxilary channel,on making protecting-judgement",2,0,,,"保護判斷時,出現無效的輔助通道類型",,,,,,,,,,,,,,
0x000000C9,LOG_PCCCVC_CHANGE_ERR,,,"电池组恒流恒压时,某一辅助通道达到设定值,转入恒压时,执行工步失败",,"Failed to execute step,while an auxilary channel has reached the set value and gets ready to change its step from CCCV to CV",2,0,,,"電池組恆流恆壓時,某一輔助通道達到設定值,轉入恆壓時,執行工步失敗",115133,,,,,,,,,,,,,
0x000000CA,LOG_GET_STEP_EXT_PARA_ERR,,,获取工步附加资源发生错误,,0x000000CA-LOG_GET_STEP_EXT_PARA_ERR---Error on receiving extra resource of step,2,0,,,獲取工步附加資源發生錯誤,,,,,,,,,,,,,,
0x000000CB,LOG_POWER_ON_ALLOCK_CHL_ERR,,,"上电自动接续时,分配通道失败",,"Failed to allocate channel,on power up auto-continuing",2,0,,,"上電自動接續時,分配通道失敗",115134,,,,,,,,,,,,,
0x000000CC,LOG_POWER_ON_ALLOCK_PARALLEL_MCH_ERR,,,"上电自动接续时,分配并联的主通道失败",,"Unable to allocate paralleling main-channel,on power up auto-continuing",2,0,,,"上電自動接續時,分配並聯的主通道失敗",115135,,,,,,,,,,,,,
0x000000CD,LOG_POST_MESSAGE_INSIDE_ERR,,,传递内部消息时失败,,Failed to post internal message,2,0,,,傳遞內部消息時失敗,,,,T8,,,,,,,,,,
0x000000CE,LOG_PUMP_AIR_ERR,,,抽真空异常保护,,Abnormal vacuum pumping protection,2,1,,,抽真空異常保護,115136,,,,,,,,,,,,,160
0x000000CF,LOG_LEAKAGE_RATE_ERR,,,真空漏率异常保护,,Abnormal leakage rate of vacuum relief protection,2,1,CGB35|CGB50|CGB63|CGB74,,真空漏率異常保護,,FJB60-4|FJB81-1|FJB99-2|FJBB5-3,,,,,,"充电能量、放电能量字段复用，起始点气压复用充电能量高4字节，单位KPa,结束点气压复用充电能量低4字节,单位KPa;气压漏率复用放电能量字段，单位KPa/min",,,,,,160
0x000000D0,LOG_LEAKAGE_RATE_INFO,,,真空漏率信息,,Information of leakage rate,0,0,,,真空漏率信息,115137,,,,,,,"充电能量、放电能量字段复用，起始点气压复用充电能量高4字节，单位KPa,结束点气压复用充电能量低4字节,单位KPa;气压漏率复用放电能量字段，单位KPa/min",,,,,,
0x000000D1,LOG_IF_PROTECT_PRESS_AIR_ERR,,,如果工步气压保护,,Condition step - pressure protection,2,1,,,如果工步氣壓保護,115138,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：气压设置值,实际值1：气压实际值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：氣壓設置值,實際值1：氣壓實際值","Set value: air pressure set value, actual value 1: actual air pressure value",160
0x000000D2,LOG_ALLOW_LEAK_AIR,,,允许泄真空,,Allow vacuum pumping,0,0,,,允許洩真空,,,,,,,,,,,,,,
0x000000D3,LOG_LEAKED_AIR,,,已执行泄真空控制,,Vacuum relief had been executed,0,0,,,已執行洩真空控制,,,,,,,,,,,,,,
0x000000D4,LOG_OPEN_HAIR,,,自动打开高真空,,Auto open high vacuum,0,0,,,自動打開高真空,,,,,,,,,,,,,,
0x000000D5,LOG_CLOSE_HAIR,,,自动关闭高真空,,Auto close high vacuum,0,0,,,自動關閉高真空,,,,,,,,,,,,,,
0x000000D6,LOG_OPEN_WZY,,,自动打开微正压,,Auto open micro positive pressure,0,0,,,自動打開微正壓,,,,,,,,,,,,,,
0x000000D7,LOG_CLOSE_WZY,,,自动关闭微正压,,Auto close micro positive pressure,0,0,,,自動關閉微正壓,,,,,,,,,,,,,,
0x000000D8,LOG_OPEN_HAIR_MAN,,,客户端操作打开高真空,,Client open high vacuum,0,0,,,客戶端操作打開高真空,,,,,,,,,,,,,,
0x000000D9,LOG_CLOSE_HAIR_MAN,,,客户端操作关闭高真空,,Client close high vacuum,0,0,,,客戶端操作關閉高真空,,,,,,,,,,,,,,
0x000000DA,LOG_OPEN_WZY_MAN,,,客户端操作打开微正压,,Client open micro positive pressure,0,0,,,客戶端操作打開微正壓,,,,,,,,,,,,,,
0x000000DB,LOG_CLOSE_WZY_MAN,,,客户端操作关闭微正压,,Client close micro positive pressure,0,0,,,客戶端操作關閉微正壓,,,,,,,,,,,,,,
0x000000DC,LOG_OPEN_LAIR_MAN,,,客户端操作打开低真空,,Client open low vacuum,0,0,,,客戶端操作打開低真空,,,,,,,,,,,,,,
0x000000DD,LOG_CLOSE_LAIR_MAN,,,客户端操作关闭低真空,,Client close low vacuum,0,0,,,客戶端操作關閉低真空,,,,,,,,,,,,,,
0x000000DE,LOG_OPEN_YC,,,自动打开压床,,Auto open neilsbed,0,0,,,自動打開壓床,,,,U2,,,,,,,,,,
0x000000DF,LOG_CLOSE_YC,,,自动关闭压床,,Auto close neilsbed,0,0,,,自動關閉壓床,,,,U3,,,,,,,,,,
0x000000E0,LOG_OPEN_YC_MAN,,,客户端打开压床,,Client open neilsbed,0,0,,,客戶端打開壓床,,,,U4,,,,,,,,,,
0x000000E1,LOG_CLOSE_YC_MAN,,,客户端关闭压床,,Client close neilsbed,0,0,,,客戶端關閉壓床,,,,U5,,,,,,,,,,
0x000000E2,LOG_RECEIVED_FORMATION_RESULT,,,收到化成结果,,Received formation result,0,0,,,收到化成結果,,,,T9,,,,,,,,,,
0x000000E3,LOG_SENDOUT_TRANSPLANT2_ORDER_SUCCEED,,,发送移栽2指令成功,,Send out transplant 2 order succeed,0,0,,,發送移栽2指令成功,,,,U0,,,,,,,,,,
0x000000E4,LOG_USER_CLEAR_CHANNEL,,,客户操作清除标记,,User clear channel,0,0,,,客戶操作清除標記,,,,U1,,,,,,,,,,
0x000000E5,LOG_RESERVE_STOP_STEP,,,预约指定工步完成后暂停,,Reserve to pause the specified step,0,0,,,預約指定工步完成後暫停,,,,,,,,,,,,,,
0x000000E6,LOG_RESERVE_STOP_CURRENT_CYCLE,,,预约当前循环完成后暂停,,Reserve to pause the current cycle,0,0,,,預約當前循環完成後暫停,,,,,,,,,,,,,,
0x000000E7,LOG_RESERVE_STOP_CURRENT_STEP,,,预约当前工步完成后暂停,,Reserve to pause the current step,0,0,,,預約當前工步完成後暫停,,,,,,,,,,,,,,
0x000000E8,LOG_RESERVE_STOP_CURRENT_CYCLE_STEP,,,预约当前循环指定工步完成后暂停,,Reserve to pause the specified step of current cycle,0,0,,,預約當前循環指定工步完成後暫停,,,,,,,,,,,,,,
0x000000E9,LOG_RESERVE_STOP_CYCLE,,,预约指定循环完成后暂停,,Reserve to pause the specified cycle,0,0,,,預約指定循環完成後暫停,,,,,,,,,,,,,,
0x000000EA,LOG_RESERVE_STOP_CYCLE_CURRENT_STEP,,,预约指定循环的当前工步完成后暂停,,Reserve to pause the current step of cycle,0,0,,,預約指定循環的當前工步完成後暫停,,,,,,,,,,,,,,
0x000000EB,LOG_RESERVE_STOP_CYCLE_STEP,,,预约某个循环某个工步完成后暂停,,Reserve to pause the specified step of cycle,0,0,,,預約某個循環某個工步完成後暫停,,,,,,,,,,,,,,
0x000000EC,LOG_RESERVE_STOP_ARRIVED,,,预约条件达到执行暂停,,Reserve to pause succeed,0,0,,,預約條件達到執行暫停,,,,,,,,,,,,,,
0x000000ED,LOG_RESERVE_STOP_JUMP,,,执行跳转，清除预约暂停信息,,Cancel reservation for jump,0,0,,,執行跳轉，清除預約暫停信息,,,,,,,,,,,,,,
0x000000EE,LOG_RESERVE_STOP_MODIFY,,,执行重置，清除预约暂停信息,,Cancel reservation for step modify,0,0,,,執行重置，清除預約暫停信息,,,,,,,,,,,,,,
0x000000EF,LOG_RESERVE_STOP_MOVE,,,执行迁移，清除预约暂停信息,,Cancel reservation for channel move,0,0,,,執行遷移，清除預約暫停信息,,,,,,,,,,,,,,
0x000000F0,LOG_CANCEL_RESERVE_STOP_STEP,,,取消 预约指定工步完成后暂停,,Cancel reservation of pause the specified step,0,0,,,取消 預約指定工步完成後暫停,,,,,,,,,,,,,,
0x000000F1,LOG_CANCEL_RESERVE_STOP_CURRENT_CYCLE,,,取消 预约当前循环完成后暂停,,Cancel reservation of pause the current cycle,0,0,,,取消 預約當前循環完成後暫停,,,,,,,,,,,,,,
0x000000F2,LOG_CANCEL_RESERVE_STOP_CURRENT_STEP,,,取消 预约当前工步完成后暂停,,Cancel reservation of pause the current step,0,0,,,取消 預約當前工步完成後暫停,,,,,,,,,,,,,,
0x000000F3,LOG_CANCEL_RESERVE_STOP_CURRENT_CYCLE_STEP,,,取消 预约当前循环指定工步完成后暂停,,Cancel reservation of pause the specified step of current cycle,0,0,,,取消 預約當前循環指定工步完成後暫停,,,,,,,,,,,,,,
0x000000F4,LOG_CANCEL_RESERVE_STOP_CYCLE,,,取消 预约指定循环完成后暂停,,Cancel reservation of pause specified cycle,0,0,,,取消 預約指定循環完成後暫停,,,,,,,,,,,,,,
0x000000F5,LOG_CANCEL_RESERVE_STOP_CYCLE_CURRENT_STEP,,,取消 预约指定循环的当前工步完成后暂停,,Cancel reservation of pause current step of cycle,0,0,,,取消 預約指定循環的當前工步完成後暫停,,,,,,,,,,,,,,
0x000000F6,LOG_CANCEL_RESERVE_STOP_CYCLE_STEP,,,取消 预约某个循环某个工步完成后暂停,,Cancel reservation of pause specified step of cycle,0,0,,,取消 預約某個循環某個工步完成後暫停,,,,,,,,,,,,,,
0x000000F7,LOG_CLEAR_RESERVE_STOP_BY_USER_STOP,,,执行停止，清除预约暂停信息,,Clear reserve stop by user stop,0,0,,,執行停止，清除預約暫停信息,,,,,,,,,,,,,,
0x000000F8,LOG_CCC_SUD_VOLDOWN_2,,CC电压突降2保护(单步),电压突降2保护,,CCC sud voltage down 2,2,0,,,電壓突降2保護,,,K2,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x000000F9,LOG_CCD_SUD_VOLUP_2,,DC电压突升2保护(单步),电压突升2保护,,CCD sud voltage up 2,2,0,,,電壓突升2保護,,,K3,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x000000FA,LOG_BY_OTHER_CH_VOL_SUD,,,因其它通道电压突变2保护而保护,,By other channel voltage sud,2,0,,,因其它通道電壓突變2保護而保護,,,K4,,,,,放电能量字段复用，二次电压保护单元号复用放电能量字段Byte1，通道号复用放电能量字段Byte0,,,,,,20
0x000000FB,LOG_BY_OTHER_CH_TEMP,,,因其它通道温度保护而保护,,By other channel temp,2,0,,,因其它通道溫度保護而保護,,,K5,,,,,放电能量字段复用，二次温度保护单元号复用放电能量字段Byte1，通道号复用放电能量字段Byte0,,,,,,60
0x000000FC,LOG_PROTECT_SECOND,,,因其它通道（二次）保护而保护,,Protect second,2,0,,,因其它通道（二次）保護而保護,,,K6,,,,,放电能量字段复用，二次电压保护单元号复用放电能量字段Byte1，通道号复用放电能量字段Byte0;二次温度保护单元号复用放电能量字段Byte2，通道号复用放电能量字段Byte3,,,,,,160
0x000000FD,LOG_VOL_UP_SECOND,,安全保护,主通道电压超出二次保护设定的电压上限,,Voltage up second,2,0,,,主通道電壓超出二次保護設定的電壓上限,115139,,K7,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x000000FE,LOG_VOL_DOWN_SECOND,,安全保护,主通道电压超出二次保护设定的电压下限,,Voltage down second,2,0,,,主通道電壓超出二次保護設定的電壓下限,115140,,K8,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x000000FF,LOG_AUX_VOL_UP_SECOND,,安全保护,辅助通道电压超出二次保护设定的电压上限,,Aux voltage up 2,2,0,,,輔助通道電壓超出二次保護設定的電壓上限,115141,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x00000100,LOG_AUX_VOL_DOWN_SECOND,,安全保护,辅助通道电压超出二次保护设定的电压下限,,Aux voltage down 2,2,0,CGB82,,輔助通道電壓超出二次保護設定的電壓下限,115142,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",20
0x00000101,LOG_AUX_TEMP_UP_SECOND,,安全保护,辅助通道温度超出二次保护设定的温度上限,,Aux temp up 2,2,0,,,輔助通道溫度超出二次保護設定的溫度上限,115143,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：溫度實際值","Set value: Set temperature value, actual value 1: actual temperature value",60
0x00000102,LOG_AUX_TEMP_DOWN_SECOND,,安全保护,辅助通道温度超出二次保护设定的温度下限,,Aux temp down 2,2,0,,,輔助通道溫度超出二次保護設定的溫度下限,115144,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：溫度實際值","Set value: Set temperature value, actual value 1: actual temperature value",60
0x00000103,LOG_LEAD_TEMP_UP,,极耳温度保护,极耳温度上限保护,,Lead temp up,2,0,,,極耳溫度上限保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：溫度實際值","Set value: Set temperature value, actual value 1: actual temperature value",60
0x00000104,LOG_LEAD_TEMP_DOWN,,极耳温度保护,极耳温度下限保护,,Lead temp down,2,0,,,極耳溫度下限保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：溫度實際值","Set value: Set temperature value, actual value 1: actual temperature value",60
0x00000105,LOG_SMOKE_ALARM_PROTECT_ALL_CH,,,设备烟雾报警，保护所有通道并断开功率电源,,Smoke alarm protect all channel,2,0,,,設備煙霧報警，保護所有通道並斷開功率電源,,,,S7,,,,,,,,,,160
0x00000106,LOG_THE_SAME_OF_STEP_RC,,,两个工步限制条件表达相同,,The same of step rc,2,0,,,兩個工步限制條件表達相同,,,,,,,,,,,,,,
0x00000107,LOG_STEP_RC_AND_STEP_EXP_RC,,,限制条件和限制条件表达都设置了,,Step rc and step exp rc,2,0,,,限制條件和限制條件表達都設置了,,,,,,,,,,,,,,
0x00000108,LOG_LADDER_PUMP_AIR_CTRL,,,抽真空,,Ladder pump air ctrl,0,0,,,抽真空,,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：真空设置值,实际值1：真空设置值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：真空設置值,實際值1：真空設置值","Setting value: vacuum setting value, actual value 1: vacuum setting value",
0x00000109,LOG_THRID_PARTY_VOL_UPPER,,,结束电压上限保护,,End Voltage upper limit protection,2,0,,,結束電壓上限保護,,,K9,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：实际电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：實際電壓值","Set value: Voltage set value, actual value 1: actual voltage",20
0x0000010A,LOG_THRID_PARTY_VOL_LOWER,,,结束电压下限保护,,End Voltage lower limit protection,2,0,,,結束電壓下限保護,,,L0,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：实际电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：實際電壓值","Set value: Voltage set value, actual value 1: actual voltage",20
0x0000010B,LOG_ALLOW_LOAD_BATTERY,,,允许化成入料,,Allow load battery,0,0,,,允許化成入料,,,,U6,,,,,,,,,,
0x0000010C,LOG_FORMATION_LOAD_BATTERY,,,化成入料,,Formation load battery,0,0,,,化成入料,,,,U7,,,,,,,,,,
0x0000010D,LOG_RECORD_SEND_FORMATION_RESULT,,,记录通道工作时下发了化成结果,,Record send formation result,0,0,,,記錄通道工作時下發了化成結果,,,,U8,,,,,,,,,,
0x0000010E,LOG_SPRAY_SECOND_VOL_SMOKE_ALARM,,,触发喷淋(二次|保护_电压_烟雾报警),,Bat vol temp neartemp,0,0,,,觸發噴淋(二次|保護_電壓_煙霧報警),115145,,,U9,,,,,,,,,,160
0x0000010F,LOG_SPRAY_SECOND_VOL_NEAR_TEMP_ALARM,,,触发喷淋(二次|保护_电压_邻位温度),,Bat vol temp smoke,0,0,,,觸發噴淋(二次|保護_電壓_鄰位溫度),115146,,,V0,,,,,,,,,,160
0x00000110,LOG_SPRAY_SECOND_VOL_LEVEL_TEMP,,,触发喷淋(二次|保护_电压_库位火警温度),,Localtemp smoke,0,0,,,觸發噴淋(二次|保護_電壓_庫位火警溫度),115147,,,V1,,,,,,,,,,160
0x00000111,LOG_SECOND_VOL_TEMP_PROTECT,,,触发喷淋(同一通道_二次电压保护_二次温度保护),,Neartemp smoke,0,0,,,觸發噴淋(同一通道_二次電壓保護_二次溫度保護),115148,,,V2,,,,,,,,,,160
0x00000112,LOG_SPRAY_SECOND_AUX_TEMP_SMOKE_ALARM,,,触发喷淋(二次|保护_辅助通道温度_烟雾报警),,Spray second aux temp upper smoke alarm,0,0,,,觸發噴淋(二次|保護_輔助通道溫度_煙霧報警),115149,,,V3,,,,,,,,,,160
0x00000113,LOG_SPRAY_SECOND_AUX_TEMP_LEVEL_TEMP_ALARM,,,触发喷淋(二次|保护_辅助通道温度_库位火警温度),,Spray second aux temp lower smoke alarm,0,0,,,觸發噴淋(二次|保護_輔助通道溫度_庫位火警溫度),115150,,,V4,,,,,,,,,,160
0x00000114,LOG_SEND_SPRAY_SIGNAL,,,下发喷淋信号,,Send spray signal,3,0,,,下發噴淋信號,115151,,,V5,1,,,,,,,,,
0x00000115,LOG_NEILSBED_ABNORMAL_SHELVES_FIRE_ALARM,,,邻库位火警保护,,Neilsbed abnormal shelves fire alarm,3,0,,,鄰庫位火警保護,115152,,,V6,,,,,,,,,,160
0x00000116,LOG_SEND_SHELVES_PROTECT_SIGNAL,,,下发库位保护信号,,Send shelves protect signal,0,0,,,下發庫位保護信號,,,,V7,,,,,,,,,,160
0x00000117,LOG_CLOSE_LAIR,,,关闭低真空,,Close low vaccum,0,0,,,關閉低真空,,,,,,,,,,,,,,
0x00000118,LOG_PUMP_AIR_STEADY_PRESS_BEGIN,,,抽真空稳压开始,,Pump air steady press begin,0,0,,,抽真空穩壓開始,,,,,,,,,,,,,,
0x00000119,LOG_PUMP_AIR_STEADY_PRESS_END,,,抽真空稳压结束,,Pump air steady press end,0,0,,,抽真空穩壓結束,,,,,,,,,,,,,,
0x0000011A,LOG_HOLD_AIR_DELAY_BEGIN,,,保压延时开始,,Hold air delay begin,0,0,,,保壓延時開始,,,,,,,,,,,,,,
0x0000011B,LOG_HOLD_AIR_DELAY_END,,,保压延时结束,,Hold air delay end,0,0,,,保壓延時結束,,,,,,,,,,,,,,
0x0000011C,LOG_HOLD_AIR_BEGIN,,,保压开始,,Hold air begin,0,0,,,保壓開始,,,,,,,,,,,,,,
0x0000011D,LOG_HOLD_AIR_END,,,保压结束,,Hold air end,0,0,,,保壓結束,,,,,,,,,,,,,,
0x0000011E,LOG_IDLE_VOL_SUDDENLY_UP_SECOND,,,二次保护_空闲时电压突升,,"Second protection, voltage rise in idle time",2,0,,,二次保護_空閒時電壓突升,115153,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压差设置值,实际值1：实际电压差变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓差設置值,實際值1：實際電壓差變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: voltage difference Set value, actual value 1: actual voltage difference change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x0000011F,LOG_IDLE_VOL_SUDDENLY_DOWN_SECOND,,,二次保护_空闲时电压突降,,"Second protection, voltage drop in idle time",2,0,,,二次保護_空閒時電壓突降,115154,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压差设置值,实际值1：实际电压差变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓差設置值,實際值1：實際電壓差變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: voltage difference Set value, actual value 1: actual voltage difference change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x00000120,LOG_IDLE_TEMP_SUDDENLY_UP_SECOND,,,[T2]二次保护_辅助通道空闲时温度突升,,[T2] second protection temperature rise when auxiliary channel is idle.,2,0,,,[T2]二次保護_輔助通道空閒時溫度突升,115155,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化设置值,实际值1：实际温度变化值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化設置值,實際值1：實際溫度變化值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Temperature change Set value, actual value 1: actual temperature change value, actual value 2: current temperature value, actual value 3: last temperature value",60
0x00000121,LOG_IDLE_TEMP_SUDDENLY_DOWN_SECOND,,,二次保护_辅助通道空闲时温度突降,,Second temperature protection when auxiliary channel is idle.,2,0,,,二次保護_輔助通道空閒時溫度突降,115156,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化设置值,实际值1：实际温度变化值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化設置值,實際值1：實際溫度變化值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Temperature change Set value, actual value 1: actual temperature change value, actual value 2: current temperature value, actual value 3: last temperature value",60
0x00000122,LOG_SPRAY_CHANNEL_PROTECT,,,通道因喷淋而保护,,The passage is protected by spray.,3,0,,,通道因噴淋而保護,115157,,L1,,,,,,,,,,,160
0x00000123,LOG_SPRAY_VOL_SUDDENLY_DOWN_SMOKE_ALARM,,,触发喷淋(二次|保护_电压突降_烟雾报警)(未使用),,"Triggered spray (second protection, voltage drop, smoke alarm) (Unused)",2,0,,,觸發噴淋(二次|保護_電壓突降_煙霧報警)(未使用),115158,,,,,,,,,,,,,160
0x00000124,LOG_SPRAY_AUX_TEMP_SUDDENLY_UP_SMOKE_ALARM,,,触发喷淋(二次|保护_辅助通道温度突升_烟雾报警)(未使用),,"Triggered spray (second times protection, auxiliary channel temperature rise, smoke alarm) (Unused)",2,0,,,觸發噴淋(二次|保護_輔助通道溫度突升_煙霧報警)(未使用),115159,,,,,,,,,,,,,160
0x00000125,LOG_SPRAY_AUX_TEMP_SUDDENLY_DOWN_SMOKE_ALARM,,,触发喷淋(二次|保护_辅助通道温度突降_烟雾报警)(未使用),,"Triggered spray (second times protection, auxiliary channel temperature drop, smoke alarm) (Unused)",2,0,,,觸發噴淋(二次|保護_輔助通道溫度突降_煙霧報警)(未使用),115160,,,,,,,,,,,,,160
0x00000126,LOG_NEILSBED_ABNORMAL_FIRE_ALARM_UPPER,,,压床异常_火警上限保护,,Abnormal fire alarm limit protection for press bed,3,0,,,壓床異常_火警上限保護,115161,,L2,,,,,,,,,,,160
0x00000127,LOG_NEILSBED_ABNORMAL_FIRE_ALARM_LOWER,,,压床异常_火警下限保护,,Lower limit protection for abnormal fire alarm of press bed,2,0,,,壓床異常_火警下限保護,115162,,L3,,,,,,,,,,,160
0x00000128,LOG_SPRAY_FIRE_SMOKE_ALARM,,,触发喷淋(库位火警温度_烟雾报警),,"Triggered sprinkler (location fire temperature, smoke alarm)",2,0,,,觸發噴淋(庫位火警溫度_煙霧報警),115163,,,V8,,,,,,,,,,
0x00000129,LOG_WAIT_SYN_OVERTIME_PROTECT,,,同步等待时间超时保护,,Synchronous waiting time timeout protection,2,0,,,同步等待時間超時保護,,,,,,,,,,,,,,160
0x0000012A,LOG_WORK_FINISH_SIGNAL,,,发送整层工作完成信号,,Send the whole layer work completion signal.,0,0,,,發送整層工作完成信號,,,,V9,,,,,,,,,,
0x0000012B,LOG_PLC_POSE_TEMP_ERR,,,工位温度异常,,Station temperature anomaly,2,0,,,工位溫度異常,,,,W0,,,,,,,,,,60
0x0000012C,LOG_PLC_PRESS_ERR,,,压力异常,,Pressure abnormality,2,0,,,壓力異常,,,,,,,,,,,,,,
0x0000012D,LOG_PLC_SET_FORMAT_ERR,,,下发化成异常信号设置,,Set up abnormal signal settings,0,0,,,下發化成異常信號設置,,,,W1,,,,,,,,,,
0x0000012E,LOG_PLC_CLR_FORMAT_ERR,,,下发化成异常信号清除,,Elimination of abnormal signal,0,0,,,下發化成異常信號清除,,,,W2,,,,,,,,,,
0x0000012F,LOG_REST_VOL_SUDDENLY_UP,,,[V5]电压突升保护,,[V5] voltage surge protection,2,0,CGB29,,[V5]電壓突升保護,115164,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x00000130,LOG_REST_VOL_SUDDENLY_DOWN,,,[V5]电压突降保护,,[V5] voltage drop protection,2,0,CGB29,,[V5]電壓突降保護,115165,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：上次電壓值","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: last voltage value",20
0x00000131,LOG_NEILSBED_ABNORMAL_CTROL_WITH_PLC,,,压床通讯异常(控制板与PLC),,Abnormal communication in press press (control panel and PLC),2,0,,,壓床通訊異常(控制板與PLC),115166,,,W3,,,,,,,,,,
0x00000132,LOG_OPEN_LAIR,,,自动打开低真空,,Open low vaccum,0,0,,,自動打開低真空,,,,,,,,,,,,,,
0x00000133,LOG_RECV_CLEAR_CHL_STATUS_CMD_WHILE_TESTING,,,测试中收到清除通道标记命令,,Receive clear channel flag command in test.,0,0,,,測試中收到清除通道標記命令,,,,W4,,,,,,,,,,
0x00000134,LOG_LOOP_RESI_PROTECTED,,,回路电阻保护,,Loop resistance protection,2,0,CGB88,,迴路電阻保護,,,,,,,"时间:{0},设置值:{1}Ω,实际值1:{2}Ω","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：回路阻抗设置值,实际值1：实际回路阻抗值","時間:{0},設置值:{1}Ω,實際值1:{2}Ω","Time :{0}, set value :{1}Ω, actual value :{2}Ω","設置值：回路阻抗設置值,實際值1：實際回路阻抗值","Set value: Set value of loop impedance, actual value 1: actual loop impedance",160
0x00000135,LOG_SECOND_PROECT_PARA_ERR,,,二次组合参数错误,,Second combination parameter error,2,0,,,二次組合參數錯誤,,,,,,,,,,,,,,
0x00000136,LOG_SECOND_PROECT_GROUP_1,,,二次保护在设置时间内因电压突降值达到异常点个数和温度突升值达到异常点个数而保护(组合1),,Secondary protection is protected during setting time because of the number of abnormal voltage drops and the number of abnormal temperature drops (combination 1),2,0,,,二次保護在設置時間內因電壓突降值達到異常點個數和溫度突升值達到異常點個數而保護(組合1),115167,,,,,,,,,,,,,60
0x00000137,LOG_SECOND_PROECT_GROUP_2,,,二次保护在设置时间内因电压突降值达到异常点个数和温度突升值达到异常点个数而保护(组合2),,Secondary protection is protected during setting time because of the number of abnormal voltage drops and the number of abnormal temperature drops (combination 2),2,0,,,二次保護在設置時間內因電壓突降值達到異常點個數和溫度突升值達到異常點個數而保護(組合2),115168,,,,,,,,,,,,,60
0x00000138,LOG_SECOND_PROECT_GROUP_3,,,二次保护在设置时间内因电压突降值达到异常点个数和温度突升值达到异常点个数而保护(组合3),,Secondary protection is protected during setting time because of the number of abnormal voltage drops and the number of abnormal temperature drops (combination 3),2,0,,,二次保護在設置時間內因電壓突降值達到異常點個數和溫度突升值達到異常點個數而保護(組合3),115169,,,,,,,,,,,,,60
0x00000139,LOG_SECOND_PROECT_GROUP_4,,,二次保护在设置时间内因电压突降值达到异常点个数和温度突升值达到异常点个数而保护(组合4),,Secondary protection is protected during setting time because of the number of abnormal voltage drops and the number of abnormal temperature drops (combination 4),2,0,,,二次保護在設置時間內因電壓突降值達到異常點個數和溫度突升值達到異常點個數而保護(組合4),115170,,,,,,,,,,,,,60
0x0000013A,LOG_NOT_SPRAY_SUDDENLY_VOL_DOWN_AUX_TEMP_UP,,,触发喷淋(二次|保护_组合_电压突变_温度突变),,"Triggered spray (second protection, combination voltage mutation, temperature mutation)",2,0,,,觸發噴淋(二次|保護_組合_電壓突變_溫度突變),115171,,,W5,,,,,,,,,,160
0x0000013B,LOG_SPRAY_AUX_TEMP_UPPER_COUNT_SMOKE_ALARM,,,触发喷淋(二次|保护_超设定温度上限个数_烟雾报警),,"Triggered sprinkler (second protection, over set temperature limit, smoke alarm)",2,0,,,觸發噴淋(二次|保護_超設定溫度上限個數_煙霧報警),115172,,,W6,,,,,,,,,,160
0x0000013C,LOG_WX_T_UPPER_LIMIT,,,温箱温度上限,,Upper limit of temperature box temperature,2,0,,,溫箱溫度上限,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：實際溫度值","Set value: Temperature set value, actual value 1: actual temperature",60
0x0000013D,LOG_WX_T_LOWER_LIMIT,,,温箱温度下限,,Lower temperature limit of thermostat,2,0,,,溫箱溫度下限,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：實際溫度值","Set value: Temperature set value, actual value 1: actual temperature",60
0x0000013E,LOG_AUX_P_UPPER_LIMIT,,,辅助通道压力上限,,Upper limit of auxiliary channel pressure,2,0,,,輔助通道壓力上限,,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：压力设置值,实际值1：实际压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg","Time :{0}, set value :{1}Kg, actual value :{2}Kg","設置值：壓力設置值,實際值1：實際壓力值","Set value: Pressure set value, actual value 1: actual pressure value",
0x0000013F,LOG_AUX_P_LOWER_LIMIT,,,辅助通道压力下限,,Auxiliary channel pressure lower limit,2,0,,,輔助通道壓力下限,,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：压力设置值,实际值1：实际压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg","Time :{0}, set value :{1}Kg, actual value :{2}Kg","設置值：壓力設置值,實際值1：實際壓力值","Set value: Pressure set value, actual value 1: actual pressure value",
0x00000140,LOG_NOT_SPRAY_SUDDENLY_VOL_DOWN,,,二次组合电压突降值达到设置值,,Second combined voltage sag reaches the set value.,0,0,,,二次組合電壓突降值達到設置值,115173,,,,,,,,,,,,,20
0x00000141,LOG_NOT_SPRAY_SUDDENLY_AUX_TEMP_UP,,,二次组合温度突升值达到设置值,,Second combined temperature rise reaches the set value.,0,0,,,二次組合溫度突升值達到設置值,115174,,,,,,,,,,,,,60
0x00000142,LOG_SEND_SECOND_GROUP_NOT_SPRAY,,,二次组合非自动喷淋保护,,Second combination non automatic sprinkler protection,3,0,,,二次組合非自動噴淋保護,115175,,,,,,,,,,,,,160
0x00000143,LOG_SEND_SECOND_GROUP_SPRAY,,,二次组合自动喷淋保护,,Second combined automatic spray protection,4,0,,,二次組合自動噴淋保護,115176,,,,2,,,,,,,,,160
0x00000144,LOG_SPRAY_AUX_TEMP_UPPER_COUNT,,,触发喷淋(二次温度上限保护_并超过设定个数),,Triggering spray (second temperature limit protection) and exceeding the set number.,0,0,,,觸發噴淋(二次溫度上限保護_並超過設定個數),115177,,,,,,,,,,,,,160
0x00000145,LOG_TEMP_SUDDENLY_UP,,,[T4]温度突升保护,,[T4] temperature rise protection,2,0,,,[T4]溫度突升保護,115178,,L4,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度突升设置值,实际值1：温度实际突升值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度突升設置值,實際值1：溫度實際突升值,實際值2：當前溫度值,實際值3：上次溫度值","Setting value: Temperature rise Setting value, actual value 1: actual temperature rise, actual value 2: current temperature, actual value 3: last temperature",60
0x00000146,LOG_CAPACITY_LOWER_PROTECT,,,"工步切换时,容量下限保护",,Capacity lower protection when working steps switch,2,0,,,"工步切換時,容量下限保護",,,L5,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x00000147,LOG_SECOND_PROECT_TEMP_GROUP,,,"电压突变保护后,在设置时间内,因温度突升值达到异常点个数而保护",,"After the sudden change of voltage protection, the temperature rise value reaches the number of outliers in the setting time.",2,0,,,"電壓突變保護後,在設置時間內,因溫度突升值達到異常點個數而保護",115179,,L6,,,,,,,,,,,60
0x00000148,LOG_SECOND_PROECT_VOL_GROUP,,,"温度突变保护后,在设置时间内,因电压突降值达到异常点个数而保护",,"After the sudden change of temperature, the number of abnormal points will be protected during the setting time.",2,0,,,"溫度突變保護後,在設置時間內,因電壓突降值達到異常點個數而保護",115180,,L7,,,,,,,,,,,60
0x00000149,LOG_PRESS_RELEASE_T1_BEGIN,,,"保护时,泄压T1开始时间",,"During protection, pressure relief T1 start time.",0,0,,,"保護時,洩壓T1開始時間",,,,,,,,,,,,,,
0x0000014A,LOG_PRESS_RELEASE_T1_END,,,"保护时,泄压T1结束时间",,"During protection, pressure relief T1 end time.",0,0,,,"保護時,洩壓T1結束時間",,,,,,,,,,,,,,
0x0000014B,LOG_PRESS_RELEASE_FAILURE,,,"保护时,泄压失败",,Failure of pressure relief during protection,1,0,,,"保護時,洩壓失敗",,,,,,,,,,,,,,
0x0000014C,LOG_PRESS_DELAY_T2_BEGIN,,,"保护时,延时打开压床T2开始时间",,"When protection, delay opens the press T2 start time.",0,0,,,"保護時,延時打開壓床T2開始時間",,,,,,,,,,,,,,
0x0000014D,LOG_PRESS_DELAY_T2_END,,,"保护时,延时打开压床T2结束时间",,"When protection, delay opens the press T2 end time.",0,0,,,"保護時,延時打開壓床T2結束時間",,,,,,,,,,,,,,
0x0000014E,LOG_WAIT_SYN_TIME_BEGIN,,,同步等待开始,,Synchronous wait starts,0,0,,,同步等待開始,,,,,,,,,,,,,,
0x0000014F,LOG_WAIT_SYN_TIME_END,,,同步等待结束,,End of synchronization wait,0,0,,,同步等待結束,,,,,,,,,,,,,,
0x00000150,LOG_IF_PROTECT_PRESS_ERR_TOPLIMIT,,,条件工步压力保护(上限),,Condition step pressure protection,2,0,,,條件工步壓力保護,,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：压力设置值,实际值1：实际压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg","Time :{0}, set value :{1}Kg, actual value :{2}Kg","設置值：壓力設置值,實際值1：實際壓力值","Set value: Pressure set value, actual value 1: actual pressure value",
0x00000151,LOG_CHECK_BLOCK_OPEN_AIRH_BEGIN,,,测堵打开高真空开始,,Measure the blockage and start the high vacuum.,0,0,,,測堵打開高真空開始,,,,,,,,,,,,,,
0x00000152,LOG_CHECK_BLOCK_OPEN_AIRH_END,,,测堵打开高真空结束,,Measurement of blockage opens high vacuum ends,0,0,,,測堵打開高真空結束,,,,,,,,,,,,,,
0x00000153,LOG_CHECK_BLOCK_STEADY_PRESS_BEGIN,,,测堵稳压开始,,Pressure measurement starts.,0,0,,,測堵穩壓開始,,,,,,,,,,,,,,
0x00000154,LOG_CHECK_BLOCK_STEADY_PRESS_END,,,测堵稳压结束,,End of metering and stabilizing,0,0,,,測堵穩壓結束,,,,,,,,,,,,,,
0x00000155,LOG_NEG_PRESS_SUDDENLY_UP,,,[P2]气压突升保护,,[P2] air pressure surge protection,2,0,,,[P2]氣壓突升保護,,FJB63-4|FJBC2-3,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa,实际值2:{3}Kpa,实际值3:{4}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：气压保护设置值,实际值1：实际气压值,实际值2：当前气压值,实际值3：上次气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa,實際值2:{3}Kpa,實際值3:{4}Kpa","Time: {0}, setting: {1} Kpa, the actual value 1: {2} Kpa, the actual value 2: {3} Kpa, the actual value 3: {4} Kpa","設置值：氣壓保護設置值,實際值1：實際氣壓值,實際值2：當前氣壓值,實際值3：上次氣壓值","Setting value: Pressure protection setting value, actual value 1: actual pressure value, actual value 2: current pressure value, actual value 3: last pressure value",160
0x00000156,LOG_CTRL_PRESS_CHANGE,,,发送压力变更控制信息,,Sending pressure change control information,0,0,,,發送壓力變更控制信息,,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置压力值,实际值1：设置压力值","時間:{0},設置值:{1}Kg,實際值1:{2}℃","Time :{0}, set value :{1}Kg, actual value :{2}℃","設置值：設置壓力值,實際值1：設置壓力值",Set value: Set the pressure value. Actual value 1: Set the pressure value,
0x00000157,LOG_SECOND_PROECT_VOL_TIME,,,"[V1]二次保护,在设置时间内,因电压突降值达到异常点个数而保护",,The two protection of [V1] is to protect the number of outliers due to the number of outliers during the set time.,2,0,,,"[V1]二次保護,在設置時間內,因電壓突降值達到異常點個數而保護",,,L8,,,,,,,,,,,20
0x00000158,LOG_SECOND_PROECT_TEMP_TIME,,,"[T1]二次保护,在设置时间内,因温度突升值达到异常点个数而保护",,"[T1] two protection, in the setting time, because the temperature rise value reaches the number of outliers.",2,0,,,"[T1]二次保護,在設置時間內,因溫度突升值達到異常點個數而保護",,,L9,,,,,,,,,,,60
0x00000159,LOG_VOL_LARGE_JUMP_PROTECT,,,[V2]忙时大幅突降保护,,[V2] sharp drop in busy hour protection,2,0,,,[V2]忙時大幅突降保護,,,M0,,,,,,,,,,,160
0x0000015A,LOG_ALLOW_EXECUTE_OPEN_CYLINDER,,,允许执行压床脱开,,Allow the execution of the press to disengage.,2,0,,,允許執行壓床脫開,,,,W7,,,,,,,,,,
0x0000015B,LOG_EXECUTE_PROTECT_ACTION,,,因执行N2自定义组合动作而保护,,Protect by performing N2 custom combination actions,2,0,,,因執行N2自定義組合動作而保護,,,,,,,,,,,,,,160
0x0000015C,LOG_V1_LIVE_TIME_START,,,V1时间生存周期开始,,V1 time life cycle begins,0,0,,,V1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000015D,LOG_V1_LIVE_TIME_END,,,V1时间生存周期结束,,V1 time life cycle ends,0,0,,,V1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x0000015E,LOG_V2_LIVE_TIME_START,,,V2时间生存周期开始,,V2 time life cycle begins,0,0,,,V2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000015F,LOG_V2_LIVE_TIME_END,,,V2时间生存周期结束,,V2 time life cycle ends,0,0,,,V2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000160,LOG_V3_LIVE_TIME_START,,,V3时间生存周期开始,,V3 time life cycle begins,0,0,,,V3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000161,LOG_V3_LIVE_TIME_END,,,V3时间生存周期结束,,V3 time life cycle ends,0,0,,,V3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000162,LOG_V4_LIVE_TIME_START,,,V4时间生存周期开始,,V4 time life cycle begins,0,0,,,V4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000163,LOG_V4_LIVE_TIME_END,,,V4时间生存周期结束,,V4 time life cycle ends,0,0,,,V4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000164,LOG_V5_LIVE_TIME_START,,,V5时间生存周期开始,,V5 time life cycle begins,0,0,,,V5時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000165,LOG_V5_LIVE_TIME_END,,,V5时间生存周期结束,,V5 time life cycle ends,0,0,,,V5時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000166,LOG_V6_LIVE_TIME_START,,,V6时间生存周期开始,,V6 time life cycle begins,0,0,,,V6時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000167,LOG_V6_LIVE_TIME_END,,,V6时间生存周期结束,,V6 time life cycle ends,0,0,,,V6時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000168,LOG_V7_LIVE_TIME_START,,,V7时间生存周期开始,,V7 time life cycle begins,0,0,,,V7時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000169,LOG_V7_LIVE_TIME_END,,,V7时间生存周期结束,,V7 time life cycle ends,0,0,,,V7時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x0000016A,LOG_T1_LIVE_TIME_START,,,T1时间生存周期开始,,T1 time life cycle begins,0,0,,,T1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000016B,LOG_T1_LIVE_TIME_END,,,T1时间生存周期结束,,T1 time life cycle ends,0,0,,,T1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x0000016C,LOG_T2_LIVE_TIME_START,,,T2时间生存周期开始,,T2 time life cycle begins,0,0,,,T2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000016D,LOG_T2_LIVE_TIME_END,,,T2时间生存周期结束,,T2 time life cycle ends,0,0,,,T2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x0000016E,LOG_T3_LIVE_TIME_START,,,T3时间生存周期开始,,T3 time life cycle begins,0,0,,,T3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000016F,LOG_T3_LIVE_TIME_END,,,T3时间生存周期结束,,T3 time life cycle ends,0,0,,,T3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000170,LOG_T4_LIVE_TIME_START,,,T4时间生存周期开始,,T4 time life cycle begins,0,0,,,T4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000171,LOG_T4_LIVE_TIME_END,,,T4时间生存周期结束,,T4 time life cycle ends,0,0,,,T4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000172,LOG_P1_LIVE_TIME_START,,,P1时间生存周期开始,,P1 time life cycle begins,0,0,,,P1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000173,LOG_P1_LIVE_TIME_END,,,P1时间生存周期结束,,P1 time life cycle ends,0,0,,,P1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000174,LOG_P2_LIVE_TIME_START,,,P2时间生存周期开始,,P2 time life cycle begins,0,0,,,P2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000175,LOG_P2_LIVE_TIME_END,,,P2时间生存周期结束,,P2 time life cycle ends,0,0,,,P2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000176,LOG_S1_LIVE_TIME_START,,,S1时间生存周期开始,,S1 time life cycle begins,0,0,,,S1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x00000177,LOG_S1_LIVE_TIME_END,,,S1时间生存周期结束,,S1 time life cycle ends,0,0,,,S1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x00000178,LOG_ALLOW_EXECUTE_MANUAL_FIRE,,,允许执行手动消防,,Manual fire control is permitted.,2,0,,,允許執行手動消防,,,,W8,,,,,,,,,,
0x00000179,LOG_ALLOW_EXECUTE_AUTO_FIRE,,,允许执行自动消防,,Automatic fire control is allowed.,4,0,,,允許執行自動消防,,,,W9,2,,,,,,,,,
0x0000017A,LOG_ALLOW_EXECUTE_CLOSE_FIRE_DOOR,,,允许执行关闭消防门,,Fire door is allowed to be closed.,2,0,,,允許執行關閉消防門,,,,X0,,,,,,,,,,
0x0000017B,LOG_ALLOW_EXECUTE_FORCE_MOVE_SMOKE,,,允许执行强制排烟,,Permission to enforce mandatory smoke extraction,2,0,,,允許執行強制排煙,,,,X1,,,,,,,,,,
0x0000017C,LOG_AUTO_RESUM_BY_UPTEMP,,,低于设定温度自动接续,,Automatic connection below set temperature,0,0,,,低於設定溫度自動接續,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：接续温度设置值,实际值1：当前温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：接續溫度設置值,實際值1：當前溫度值",Set value: Follows the temperature setting value. Actual value 1: indicates the current temperature,
0x0000017D,LOG_AUTO_RESUM_BY_LOTEMP,,,高于设定温度自动接续,,Automatic connection above set temperature,0,0,,,高於設定溫度自動接續,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：接续温度设置值,实际值1：当前温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：接續溫度設置值,實際值1：當前溫度值",Set value: Follows the temperature setting value. Actual value 1: indicates the current temperature,
0x0000017E,LOG_AUTO_RESUM_BY_TEMP_RANGE,,,在设定的温度范围内自动接续,,Automatic connection in set temperature range,0,0,,,在設定的溫度範圍內自動接續,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度上限设置值,实际值1：温度下限设置值,实际值2：最大温度值,实际值3：最小温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度上限設置值,實際值1：溫度下限設置值,實際值2：最大溫度值,實際值3：最小溫度值","Set value: upper limit temperature set value, actual value 1: lower limit temperature set value, actual value 2: maximum temperature, actual value 3: minimum temperature",
0x0000017F,LOG_LEAK_RATE_LOWER_LIMIT,,,测漏检测负压低于下限,,Leak detection negative pressure lower than the lower limit,1,1,,,測漏檢測負壓低於下限,,,,,,,,,,,,,,
0x00000180,LOG_LEAK_RATE_UPPER_LIMIT,,,测漏检测负压高于上限,,Leak detection is higher than the upper limit of negative pressure.,1,1,,,測漏檢測負壓高於上限,,,,,,,,,,,,,,
0x00000181,LOG_SLAVE_ERR,,,从中位机有异常发生,,There is an abnormal occurrence from the median machine.,2,0,,,從中位機有異常發生,,,,X2,,,,,,,,,,90
0x00000182,LOG_FIX_TEMP_ERR,,,夹具温度异常,,Fixture temperature anomaly,2,0,,,夾具溫度異常,,,,X3,,,,,,,,,,60
0x00000183,LOG_CLAMP_PRESS_ERR,,,夹具电机或者压力异常,,Fixture motor or pressure abnormal,2,0,,,夾具電機或者壓力異常,,,,X4,,,,,,,,,,
0x00000184,LOG_CLAMP_REQ_STOP,,,夹具异常，请求停止,,"Clamp abnormal, request stop",2,0,,,夾具異常，請求停止,,,,X5,,,,,,,,,,
0x00000185,LOG_REST_VOL_PICUP_DEV,,,"搁置时,电压回检偏差超出设定偏差范围",,"When used, the voltage return deviation exceeds the set deviation range.",2,0,CGB37,,"擱置時,電壓回檢偏差超出設定偏差範圍",,,M1,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：偏差电压设置值,实际值1：偏差电压实际值,实际值2：主通道电压值,实际值3：辅助通道电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：偏差電壓設置值,實際值1：偏差電壓實際值,實際值2：主通道電壓值,實際值3：輔助通道電壓值","Set value: Set value of deviation voltage, actual value 1: actual value of deviation voltage, actual value 2: voltage of the main channel, actual value 3: voltage of the auxiliary channel",20
0x00000186,LOG_CCC_VOL_UP,,,"恒流充电,超出电压上限",,"CC, exceeding voltage limit.",2,0,CGB46,,"恆流充電,超出電壓上限",,FJB76-1,M2,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,20
0x00000187,LOG_CCC_VOL_DOWN,,,"恒流充电,低于电压下限",,"CC, lower than the lower voltage limit.",2,0,CGB58,,"恆流充電,低於電壓下限",,FJB77-1,M3,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压下限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓下限值,實際值1：當前電壓值",Set value: lower voltage limit. Actual value 1: current voltage value,20
0x00000188,LOG_CCD_VOL_UP,,,"恒流放电,超出电压上限",,"DC, exceeding voltage limit",2,0,CGB76,,"恆流放電,超出電壓上限",,FJB92-2,M4,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,20
0x00000189,LOG_CCD_VOL_DOWN,,,"恒流放电,低于电压下限",,"DC, below voltage limit.",2,0,CGB47,,"恆流放電,低於電壓下限",,FJB93-2,M5,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压下限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓下限值,實際值1：當前電壓值",Set value: lower voltage limit. Actual value 1: current voltage value,20
0x0000018A,LOG_CVC_VOL_UP,,,"恒压充电,超出电压上限",,"CV, exceeding voltage limit.",2,0,CGB59,,"恆壓充電,超出電壓上限",,FJBB8-3,M6,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,20
0x0000018B,LOG_CVC_VOL_DOWN,,,"恒压充电,低于电压下限",,"CV, lower than voltage limit.",2,0,CGB77,,"恆壓充電,低於電壓下限",,FJBB9-3,M7,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压下限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓下限值,實際值1：當前電壓值",Set value: lower voltage limit. Actual value 1: current voltage value,20
0x0000018C,LOG_REST_IF_PROTECT_PRESS_AIR_ERR,,,"搁置时,条件工步气压保护",,"Rest, pneumatic pressure protection.",2,1,,,"擱置時,條件工步氣壓保護",,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x0000018D,LOG_CCC_IF_PROTECT_PRESS_AIR_ERR,,,"恒流充电时,条件工步气压保护",,"CC, condition step gas pressure protection.",2,1,,,"恆流充電時,條件工步氣壓保護",,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x0000018E,LOG_CCD_IF_PROTECT_PRESS_AIR_ERR,,,"恒流放电时,条件工步气压保护",,"DC, condition step gas pressure protection.",2,1,,,"恆流放電時,條件工步氣壓保護",,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x0000018F,LOG_CVC_IF_PROTECT_PRESS_AIR_ERR,,,"恒压充电时,条件工步气压保护",,"CV, condition step gas pressure protection.",2,1,,,"恆壓充電時,條件工步氣壓保護",,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000190,LOG_REST_PRESSURE_UPPER,,,"搁置时,气压上限保护",,"Rest, the upper limit of pressure is protected",2,1,CGB34,,"擱置時,氣壓上限保護",,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000191,LOG_REST_PRESSURE_LOWER,,,"搁置时,气压下限保护",,"Rest, the lower limit of pressure is protected.",2,1,CGB34,,"擱置時,氣壓下限保護",,,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000192,LOG_CCC_PRESSURE_UPPER,,,"恒流充电时,气压上限保护",,"CC, upper limit protection",2,1,CGB49,,"恆流充電時,氣壓上限保護",,FJB80-1,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000193,LOG_CCC_PRESSURE_LOWER,,,"恒流充电时,气压下限保护",,"CC, lower limit protection",2,1,CGB49,,"恆流充電時,氣壓下限保護",,FJB80-1,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000194,LOG_CCD_PRESSURE_UPPER,,,"恒流放电时,气压上限保护",,"DC, Upper limit protection",2,1,CGB62,,"恆流放電時,氣壓上限保護",,FJB98-2,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000195,LOG_CCD_PRESSURE_LOWER,,,"恒流放电时,气压下限保护",,"DC, lower limit protection",2,1,CGB62,,"恆流放電時,氣壓下限保護",,FJB98-2,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000196,LOG_CVC_PRESSURE_UPPER,,,"恒压充电时,气压上限保护",,"CV, upper limit protection",2,1,CGB73,,"恆壓充電時,氣壓上限保護",,FJBB4-3,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000197,LOG_CVC_PRESSURE_LOWER,,,"恒压充电时,气压下限保护",,"CV, lower limit protection",2,1,CGB73,,"恆壓充電時,氣壓下限保護",,FJBB4-3,,,,,"时间:{0},设置值:{1}Kpa,实际值1:{2}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置气压值,实际值1：实际气压值","時間:{0},設置值:{1}Kpa,實際值1:{2}Kpa","Time :{0}, set value :{1}Kpa, actual value :{2}Kpa","設置值：設置氣壓值,實際值1：實際氣壓值",Set value: Set the air pressure. Actual value 1: actual air pressure,160
0x00000198,LOG_CCC_CAPACITY,,,"恒流充电时,容量超出上限值",,"CC, capacity exceeds 1.2 times of the process value.",2,0,CGB48,,"恆流充電時,容量超出上限值",,FJB79-1,M8,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x00000199,LOG_CCD_CAPACITY,,,"恒流放电时,容量超出上限值",,"DC, capacity exceeds 1.2 times of the process value.",2,0,CGB60,,"恆流放電時,容量超出上限值",,FJB95-2,M9,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x0000019A,LOG_CVC_CAPACITY,,,"恒压充电时,容量超出上限值",,"CV, capacity exceeds 1.2 times of the process value.",2,0,CGB72,,"恆壓充電時,容量超出上限值",,FJBB3-3,N0,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x0000019B,LOG_CCC_CAPACITY_LOWER_PROTECT,,,"恒流充电结束时,工步切换时,容量下限保护",,"At the end of CC, the lower limit of capacity is protected when the step is switched.",2,0,CGB52,,"恆流充電結束時,工步切換時,容量下限保護",,FJB84-1,N1,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x0000019C,LOG_CCD_CAPACITY_LOWER_PROTECT,,,"恒流放电结束时,工步切换时,容量下限保护",,"At the end of DC, the lower limit of capacity is protected when the step is switched.",2,0,CGB65,,"恆流放電結束時,工步切換時,容量下限保護",,FJBA2-2,N2,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x0000019D,LOG_CVC_CAPACITY_LOWER_PROTECT,,,"恒压充电结束时,工步切换时,容量下限保护",,"At the end of CV, the lower limit of capacity is protected when the step is switched.",2,0,CGB78,,"恆壓充電結束時,工步切換時,容量下限保護",,FJBC1-3,N3,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x0000019E,LOG_REST_TEMP_SUDDENLY_UP,,,"搁置时,(T4)温度突升保护",,"Rest, (T4) temperature rise protection.",2,0,CGB36,,"擱置時,(T4)溫度突升保護",,FJB61-4,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度突升设置值,实际值1：温度实际突升值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度突升設置值,實際值1：溫度實際突升值,實際值2：當前溫度值,實際值3：上次溫度值","Setting value: Temperature rise Setting value, actual value 1: actual temperature rise, actual value 2: current temperature, actual value 3: last temperature",60
0x0000019F,LOG_CCC_TEMP_SUDDENLY_UP,,,"恒流充电时,(T4)温度突升保护",,"CC, (T4) temperature rise protection.",2,0,CGB51,,"恆流充電時,(T4)溫度突升保護",,FJB82-1,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度突升设置值,实际值1：温度实际突升值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度突升設置值,實際值1：溫度實際突升值,實際值2：當前溫度值,實際值3：上次溫度值","Setting value: Temperature rise Setting value, actual value 1: actual temperature rise, actual value 2: current temperature, actual value 3: last temperature",60
0x000001A0,LOG_CCD_TEMP_SUDDENLY_UP,,,"恒流放电时,(T4)温度突升保护",,"DC, (T4) temperature rise protection.",2,0,CGB64,,"恆流放電時,(T4)溫度突升保護",,FJBA0-2,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度突升设置值,实际值1：温度实际突升值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度突升設置值,實際值1：溫度實際突升值,實際值2：當前溫度值,實際值3：上次溫度值","Setting value: Temperature rise Setting value, actual value 1: actual temperature rise, actual value 2: current temperature, actual value 3: last temperature",60
0x000001A1,LOG_CVC_TEMP_SUDDENLY_UP,,,"恒压充电时,(T4)温度突升保护",,"CV, (T4) temperature rise protection",2,0,CGB75,,"恆壓充電時,(T4)溫度突升保護",,FJBB6-3,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度突升设置值,实际值1：温度实际突升值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度突升設置值,實際值1：溫度實際突升值,實際值2：當前溫度值,實際值3：上次溫度值","Setting value: Temperature rise Setting value, actual value 1: actual temperature rise, actual value 2: current temperature, actual value 3: last temperature",60
0x000001A2,LOG_CHECK_BLOCK_OPEN_NEILSBED_FAILED,,,测堵打开压床失败,,Failure of measuring press to open press bed,2,0,,,測堵打開壓床失敗,,,,,,,,,,,,,,
0x000001A3,LOG_CHECK_BLOCK_NOZZLE_BLOCK,,,测堵吸嘴堵塞,,Blockage of suction plug,2,0,,,測堵吸嘴堵塞,,,,,,,,,,,,,,
0x000001A4,LOG_CHECK_BLOCK_TIME_SET_ABNORMAL,,,测堵时间设置异常,,Abnormal setting time of measurement,2,0,,,測堵時間設置異常,,,,,,,,,,,,,,
0x000001A5,LOG_PUMP_AIR_CLOSE_NEILSBED_FAILED,,,抽真空时关闭压床失败,,Failure of closing press when vacuuming,2,0,,,抽真空時關閉壓床失敗,,,,,,,,,,,,,,
0x000001A6,LOG_PUMP_AIR_NEILSBED_UNCLOSED,,,抽真空压床未闭合,,Vacuum press is not closed.,2,0,,,抽真空壓床未閉合,,,,,,,,,,,,,,
0x000001A7,LOG_PUMP_AIR_VALUE_NOT_TO_STANDARD,,,真空未达标,,Vacuum is not up to standard.,2,1,,,真空未達標,,,,,,,,,,,,,,
0x000001A8,LOG_CHECK_LEAK_RATE_NEILSBED_UNCLOSED,,,测漏压床未闭合,,Leak detection press not closed,2,0,,,測漏壓床未閉合,,,,,,,,,,,,,,
0x000001A9,LOG_CHECK_LEAK_RATE_NOT_TO_STANDARD,,,漏率未达标,,Leakage rate is not up to standard.,2,1,,,漏率未達標,,,,,,,,,,,,,,
0x000001AA,LOG_CHECK_LEAK_RATE_PRESS_LOWER,,,测漏检测气压低于下限,,Leak detection pressure lower than the lower limit,2,1,,,測漏檢測氣壓低於下限,,,,,,,,,,,,,,
0x000001AB,LOG_CHECK_LEAK_RATE_PRESS_UPPER,,,测漏检测气压高于上限,,Leak detection pressure is higher than the upper limit.,2,1,,,測漏檢測氣壓高於上限,,,,,,,,,,,,,,
0x000001AC,LOG_CHECK_BLOCK_ABNORMAL,,,测堵异常,,Abnormality of plugging,2,0,,,測堵異常,,,,,,,,,,,,,,
0x000001AD,LOG_CTRL_TEMP_CHANGE,,,温度变更控制,,Temperature change control,0,0,,,溫度變更控制,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：设置温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：設置溫度值",Set value: Set the temperature. Actual value 1: Set the temperature,
0x000001AE,LOG_LONG_TIME_SAMPLE_ERR_PROTECT_ALL_CH,,,因长时连续采样数据错误而通道保护,,Channel protection for long time continuous sampling data errors,2,0,,,因長時連續採樣數據錯誤而通道保護,,,N8,,,,,,,,,,,
0x000001AF,LOG_SEND_FORMATION_DATA_ABNORMAL,,,下发化成柜通信异常信号,,Communication abnormal signal of incoming generation cabinet,0,0,,,下發化成櫃通信異常信號,,,N9,,,,,,,,,,,
0x000001B0,LOG_VOL_LIMIT_CONDITION_JUMP,,,电压限制条件跳转,,Voltage limit jump,-1,0,,,電壓限制條件跳轉,,,NA,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置电压值,实际值1：实际电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：設置電壓值,實際值1：實際電壓值",Set value: Set voltage value. Actual value 1: actual voltage value,
0x000001B1,LOG_CUR_LIMIT_CONDITION_JUMP,,,电流限制条件跳转,,Current limit jump,-1,0,,,電流限制條件跳轉,,,NB,,,,,,,,,,,
0x000001B2,LOG_TIME_LIMIT_CONDITION_JUMP,,,时间限制条件跳转,,Time limit jump,-1,0,,,時間限制條件跳轉,,,NC,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置时间值,实际值1：实际时间值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：設置時間值,實際值1：實際時間值",Set value: Set time value. Actual value 1: actual time value,
0x000001B3,LOG_CAP_LIMIT_CONDITION_JUMP,,,容量限制条件跳转,,Capacity limit jump,-1,0,,,容量限制條件跳轉,,,ND,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：容量设置值,实际值1：容量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：容量設置值,實際值1：容量實際值,實際值2：設置電流放大系數","Set value: Capacity Set value, actual value 1: capacity set value, actual value 2: current amplification factor",40
0x000001B4,LOG_ENERGY_LIMIT_CONDITION_JUMP,,,能量限制条件跳转,,Energy limit jump,-1,0,,,能量限制條件跳轉,,,NE,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah,实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：能量设置值,实际值1：能量实际值,实际值2：设置电流放大系数","時間:{0},設置值:{1}Ah,實際值1:{2}Ah,實際值2:{3}","Time :{0}, set value :{1}Ah, actual value 1:{2}Ah, actual value 2:{3}","設置值：能量設置值,實際值1：能量實際值,實際值2：設置電流放大系數","Setting value: energy setting value, actual value 1: actual energy value, actual value 2: current amplification factor",
0x000001B5,LOG_SOTP_CUR_CONDITION_JUMP,,,截止电流条件跳转,,Cut-off current condition jump,-1,0,,,截止電流條件跳轉,,,NF,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3},实际值3:{4}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：电流实际值,实际值2：采样电流放大系数,实际值3：设置电流放大系数","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3},實際值3:{4}","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}, actual value 3:{4}","設置值：電流設置值,實際值1：電流實際值,實際值2：采樣電流放大系數,實際值3：設置電流放大系數","Set value: current set value, actual value 1: actual current value, actual value 2: sampled current amplification factor, actual value 3: Set current amplification factor",
0x000001B6,LOG_SOTP_VOL_CONDITION_JUMP,,,截止电压条件跳转,,Cut-off voltage condition jump,-1,0,,,截止電壓條件跳轉,,,NG,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置电压值,实际值1：实际电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：設置電壓值,實際值1：實際電壓值",Set value: Set voltage value. Actual value 1: actual voltage value,
0x000001B7,LOG_LOWER_LIMIT_VOL_CONDITION_JUMP,,,下限截止电压(脉冲工步使用)条件跳转,,Lower limit cut-off voltage (pulse step use) conditional jump,-1,0,,,下限截止電壓(脈衝工步使用)條件跳轉,,,NH,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置电压值,实际值1：实际电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：設置電壓值,實際值1：實際電壓值",Set value: Set voltage value. Actual value 1: actual voltage value,
0x000001B8,LOG_VOL_CHANGE_LIMIT_CONDITION_JUMP,,,ΔV限制条件跳转,,ΔV conditional jump,-1,0,,,ΔV限制條件跳轉,,,NI,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置电压变化值,实际值1：实际电压变化值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：設置電壓變化值,實際值1：實際電壓變化值",Set value: Set voltage change value. Actual value 1: actual voltage change value,
0x000001B9,LOG_AUX_TEMP_UPPER_CONDITION_JUMP,,,辅助通道温度限制上限跳转,,Auxiliary channel temperature limit upper limit jump,-1,0,,,輔助通道溫度限制上限跳轉,,,NJ,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：辅助温度设置值,实际值1：辅助温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：輔助溫度設置值,實際值1：輔助溫度實際值","Set value: auxiliary temperature set value, actual value 1: auxiliary temperature actual value",
0x000001BA,LOG_AUX_TEMP_UPPER_CONDITION_JUMP,,,辅助通道温度限制下限跳转,,Auxiliary channel temperature limit lower limit jump,-1,0,,,輔助通道溫度限制下限跳轉,,,NK,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：辅助电压温度值,实际值1：辅助温度实际值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：輔助電壓溫度值,實際值1：輔助溫度實際值","Set value: auxiliary voltage temperature value, actual value 1: actual auxiliary temperature value",
0x000001BB,LOG_AUX_VOL_UPPER_CONDITION_JUMP,,,辅助通道电压限制上限跳转,,Auxiliary channel voltage limit upper limit jump,-1,0,,,輔助通道電壓限制上限跳轉,,,P0,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：辅助电压设置值,实际值1：辅助电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：輔助電壓設置值,實際值1：輔助電壓實際值","Set value: Set value of the auxiliary voltage, actual value 1: actual value of the auxiliary voltage",
0x000001BC,LOG_AUX_VOL_LOWER_CONDITION_JUMP,,,辅助通道电压限制下限跳转,,Auxiliary channel voltage limit lower limit jump,-1,0,,,輔助通道電壓限制下限跳轉,,,P1,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：辅助电压设置值,实际值1：辅助电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：輔助電壓設置值,實際值1：輔助電壓實際值","Set value: Set value of the auxiliary voltage, actual value 1: actual value of the auxiliary voltage",
0x000001BD,LOG_IF_CONDITION_AUTO_JUMP_NEXT_STEP,,,如果条件跳到下一工步,,If conditions jump to the next step,-1,0,,,如果條件跳到下一工步,,,P2,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：如果设置值,实际值1：实际值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：如果設置值,實際值1：實際值","Set value: If set, actual value 1: actual value",
0x000001BE,LOG_ALLOW_EXECUTE_FIX_STOP_HEAT,,,允许执行单夹具停止加热,,Allow single clamp to stop heating.,0,0,,,允許執行單夾具停止加熱,,,,,,,,,,,,,,
0x000001BF,LOG_ALLOW_EXECUTE_NEAR_FIX_STOP_HAET,,,允许执行相邻夹具停止加热,,Allow adjacent clamp to stop heating.,0,0,,,允許執行相鄰夾具停止加熱,,,,,,,,,,,,,,
0x000001C0,LOG_ALLOW_EXECUTE_TROLLRY_KEEP_CLEAR,,,允许执行天车避位,,Permission to avoid overhead travelling crane,0,0,,,允許執行天車避位,,,,X6,,,,,,,,,,
0x000001C1,LOG_ALLOW_EXECUTE_FIX_HOLD_ON,,,允许执行夹具夹紧,,Clamp clamping is allowed.,0,0,,,允許執行夾具夾緊,,,,X7,,,,,,,,,,
0x000001C2,LOG_ALLOW_EXECUTE_DEVICE_FIRE_ALARM,,,允许执行设备火警报警,,Fire alarm is allowed to execute equipment.,0,0,,,允許執行設備火警報警,,,,X8,,,,,,,,,,
0x000001C3,LOG_FTL1_LIVE_TIME_START,,,FTL1时间生存周期开始,,FTL1 time life cycle begins,0,0,,,FTL1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001C4,LOG_FTL1_LIVE_TIME_END,,,FTL1时间生存周期结束,,FTL1 time life cycle ends,0,0,,,FTL1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001C5,LOG_FTL2_LIVE_TIME_START,,,FTL2时间生存周期开始,,FTL2 time life cycle begins,0,0,,,FTL2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001C6,LOG_FTL2_LIVE_TIME_END,,,FTL2时间生存周期结束,,FTL2 time life cycle ends,0,0,,,FTL2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001C7,LOG_FTL3_LIVE_TIME_START,,,FTL3时间生存周期开始,,FTL3 time life cycle begins,0,0,,,FTL3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001C8,LOG_FTL3_LIVE_TIME_END,,,FTL3时间生存周期结束,,FTL3 time life cycle ends,0,0,,,FTL3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001C9,LOG_FTL4_LIVE_TIME_START,,,FTL4时间生存周期开始,,FTL4 time life cycle begins,0,0,,,FTL4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001CA,LOG_FTL4_LIVE_TIME_END,,,FTL4时间生存周期结束,,FTL4 time life cycle ends,0,0,,,FTL4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001CB,LOG_FTL5_LIVE_TIME_START,,,FTL5时间生存周期开始,,FTL5 time life cycle begins,0,0,,,FTL5時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001CC,LOG_FTL5_LIVE_TIME_END,,,FTL5时间生存周期结束,,FTL5 time life cycle ends,0,0,,,FTL5時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001CD,LOG_FTR1_LIVE_TIME_START,,,FTR1时间生存周期开始,,FTR1 time life cycle begins,0,0,,,FTR1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001CE,LOG_FTR1_LIVE_TIME_END,,,FTR1时间生存周期结束,,FTR1 time life cycle ends,0,0,,,FTR1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001CF,LOG_FTR2_LIVE_TIME_START,,,FTR2时间生存周期开始,,FTR2 time life cycle begins,0,0,,,FTR2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001D0,LOG_FTR2_LIVE_TIME_END,,,FTR2时间生存周期结束,,FTR2 time life cycle ends,0,0,,,FTR2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001D1,LOG_FTR3_LIVE_TIME_START,,,FTR3时间生存周期开始,,FTR3 time life cycle begins,0,0,,,FTR3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001D2,LOG_FTR3_LIVE_TIME_END,,,FTR3时间生存周期结束,,FTR3 time life cycle ends,0,0,,,FTR3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001D3,LOG_FTR4_LIVE_TIME_START,,,FTR4时间生存周期开始,,FTR4 time life cycle begins,0,0,,,FTR4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001D4,LOG_FTR4_LIVE_TIME_END,,,FTR4时间生存周期结束,,FTR4 time life cycle ends,0,0,,,FTR4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001D5,LOG_FTR5_LIVE_TIME_START,,,FTR5时间生存周期开始,,FTR5 time life cycle begins,0,0,,,FTR5時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001D6,LOG_FTR5_LIVE_TIME_END,,,FTR5时间生存周期结束,,FTR5 time life cycle ends,0,0,,,FTR5時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001D7,LOG_FTU1_LIVE_TIME_START,,,FTU1时间生存周期开始,,FTU1 time life cycle begins,0,0,,,FTU1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001D8,LOG_FTU1_LIVE_TIME_END,,,FTU1时间生存周期结束,,FTU1 time life cycle ends,0,0,,,FTU1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001D9,LOG_FTU2_LIVE_TIME_START,,,FTU2时间生存周期开始,,FTU2 time life cycle begins,0,0,,,FTU2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001DA,LOG_FTU2_LIVE_TIME_END,,,FTU2时间生存周期结束,,FTU2 time life cycle ends,0,0,,,FTU2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001DB,LOG_FTU3_LIVE_TIME_START,,,FTU3时间生存周期开始,,FTU3 time life cycle begins,0,0,,,FTU3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001DC,LOG_FTU3_LIVE_TIME_END,,,FTU3时间生存周期结束,,FTU3 time life cycle ends,0,0,,,FTU3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001DD,LOG_FTU4_LIVE_TIME_START,,,FTU4时间生存周期开始,,FTU4 time life cycle begins,0,0,,,FTU4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001DE,LOG_FTU4_LIVE_TIME_END,,,FTU4时间生存周期结束,,FTU4 time life cycle ends,0,0,,,FTU4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001DF,LOG_FTU5_LIVE_TIME_START,,,FTU5时间生存周期开始,,FTU5 time life cycle begins,0,0,,,FTU5時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001E0,LOG_FTU5_LIVE_TIME_END,,,FTU5时间生存周期结束,,FTU5 time life cycle ends,0,0,,,FTU5時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001E1,LOG_FTD1_LIVE_TIME_START,,,FTD1时间生存周期开始,,FTD1 time life cycle begins,0,0,,,FTD1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001E2,LOG_FTD1_LIVE_TIME_END,,,FTD1时间生存周期结束,,FTD1 time life cycle ends,0,0,,,FTD1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001E3,LOG_FTD2_LIVE_TIME_START,,,FTD2时间生存周期开始,,FTD2 time life cycle begins,0,0,,,FTD2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001E4,LOG_FTD2_LIVE_TIME_END,,,FTD2时间生存周期结束,,FTD2 time life cycle ends,0,0,,,FTD2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001E5,LOG_FTD3_LIVE_TIME_START,,,FTD3时间生存周期开始,,FTD3 time life cycle begins,0,0,,,FTD3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001E6,LOG_FTD3_LIVE_TIME_END,,,FTD3时间生存周期结束,,FTD3 time life cycle ends,0,0,,,FTD3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001E7,LOG_FTD4_LIVE_TIME_START,,,FTD4时间生存周期开始,,FTD4 time life cycle begins,0,0,,,FTD4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001E8,LOG_FTD4_LIVE_TIME_END,,,FTD4时间生存周期结束,,FTD4 time life cycle ends,0,0,,,FTD4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001E9,LOG_FTD5_LIVE_TIME_START,,,FTD5时间生存周期开始,,FTD5 time life cycle begins,0,0,,,FTD5時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001EA,LOG_FTD5_LIVE_TIME_END,,,FTD5时间生存周期结束,,FTD5 time life cycle ends,0,0,,,FTD5時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001EB,LOG_FP1_LIVE_TIME_START,,,FP1时间生存周期开始,,FP1 time life cycle begins,0,0,,,FP1時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001EC,LOG_FP1_LIVE_TIME_END,,,FP1时间生存周期结束,,FP1 time life cycle ends,0,0,,,FP1時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001ED,LOG_FP2_LIVE_TIME_START,,,FP2时间生存周期开始,,FP2 time life cycle begins,0,0,,,FP2時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001EE,LOG_FP2_LIVE_TIME_END,,,FP2时间生存周期结束,,FP2 time life cycle ends,0,0,,,FP2時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001EF,LOG_FP3_LIVE_TIME_START,,,FP3时间生存周期开始,,FP3 time life cycle begins,0,0,,,FP3時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001F0,LOG_FP3_LIVE_TIME_END,,,FP3时间生存周期结束,,FP3 time life cycle ends,0,0,,,FP3時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001F1,LOG_FP4_LIVE_TIME_START,,,FP4时间生存周期开始,,FP4 time life cycle begins,0,0,,,FP4時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001F2,LOG_FP4_LIVE_TIME_END,,,FP4时间生存周期结束,,FP4 time life cycle ends,0,0,,,FP4時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001F3,LOG_FP5_LIVE_TIME_START,,,FP5时间生存周期开始,,FP5 time life cycle begins,0,0,,,FP5時間生存週期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000001F4,LOG_FP5_LIVE_TIME_END,,,FP5时间生存周期结束,,FP5 time life cycle ends,0,0,,,FP5時間生存週期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000001F5,LOG_FTL1_TEMP_UPPER,,,层板左热电阻全时上限(FTL1),,Full time upper limit of left thermal resistance of laminate (FTL1),2,0,,,層板左熱電阻全時上限(FTL1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001F6,LOG_FTL1_TEMP_LOWER,,,层板左热电阻全时下限(FTL1),,The full time lower limit of the left thermal resistance of the laminate (FTL1),2,0,,,層板左熱電阻全時下限(FTL1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001F7,LOG_FTR1_TEMP_UPPER,,,层板右热电阻全时上限(FTR1),,Full time upper limit of right thermal resistance of laminate (FTR1),2,0,,,層板右熱電阻全時上限(FTR1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001F8,LOG_FTR1_TEMP_LOWER,,,层板右热电阻全时下限(FTR1),,Full time lower limit of thermal resistance of laminate (FTR1),2,0,,,層板右熱電阻全時下限(FTR1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001F9,LOG_FTU1_TEMP_UPPER,,,层板上极耳电偶全时上限(FTU1),,Full time upper limit of polar ear couple on layer (FTU1),2,0,,,層板上極耳電偶全時上限(FTU1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001FA,LOG_FTU1_TEMP_LOWER,,,层板上极耳电偶全时下限(FTU1),,The full time lower limit of polar ear couple on the laminate (FTU1),2,0,,,層板上極耳電偶全時下限(FTU1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001FB,LOG_FTD1_TEMP_UPPER,,,层板下极耳电偶全时上限(FTD1),,Full time upper limit of lower pole ear couple (FTD1),2,0,,,層板下極耳電偶全時上限(FTD1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001FC,LOG_FTD1_TEMP_LOWER,,,层板下极耳电偶全时下限(FTD1),,Full time lower limit of the lower pole ear couple (FTD1),2,0,,,層板下極耳電偶全時下限(FTD1),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,
0x000001FD,LOG_FP1_PRESS_UPPER,,,层板压力全时上限(FP1),,Full time pressure limit of laminates (FP1),2,0,,,層板壓力全時上限(FP1),,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置压力值,实际值1：实际压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg","Time :{0}, set value :{1}Kg, actual value :{2}Kg","設置值：設置壓力值,實際值1：實際壓力值",Set value: Set pressure value. Actual value 1: actual pressure value,
0x000001FE,LOG_FP1_PRESS_LOWER,,,层板压力全时下限(FP1),,Full time and lower limit of laminate pressure (FP1),2,0,,,層板壓力全時下限(FP1),,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置压力值,实际值1：实际压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg","Time :{0}, set value :{1}Kg, actual value :{2}Kg","設置值：設置壓力值,實際值1：實際壓力值",Set value: Set pressure value. Actual value 1: actual pressure value,
0x000001FF,LOG_FTL4_TEMP_SUDDENLY,,,层板左热电阻全时突升(FTL4),,The left thermal resistance of the laminate rises all the time (FTL4).,2,0,,,層板左熱電阻全時突升(FTL4),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化设置值,实际值1：实际温度变化值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化設置值,實際值1：實際溫度變化值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Temperature change Set value, actual value 1: actual temperature change value, actual value 2: current temperature value, actual value 3: last temperature value",
0x00000200,LOG_FTR4_TEMP_SUDDENLY,,,层板右热电阻全时突升(FTR4),,All right time thermal resistance of laminated plates rises suddenly (FTR4).,2,0,,,層板右熱電阻全時突升(FTR4),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化设置值,实际值1：实际温度变化值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化設置值,實際值1：實際溫度變化值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Temperature change Set value, actual value 1: actual temperature change value, actual value 2: current temperature value, actual value 3: last temperature value",
0x00000201,LOG_FTU4_TEMP_SUDDENLY,,,层板上极耳电偶全时突升(FTU4),,All ears on the laminate are suddenly raised (FTU4).,2,0,,,層板上極耳電偶全時突升(FTU4),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化设置值,实际值1：实际温度变化值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化設置值,實際值1：實際溫度變化值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Temperature change Set value, actual value 1: actual temperature change value, actual value 2: current temperature value, actual value 3: last temperature value",
0x00000202,LOG_FTD4_TEMP_SUDDENLY,,,层板下极耳电偶全时突升(FTD4),,Full time spike of the lower pole ear couple (FTD4),2,0,,,層板下極耳電偶全時突升(FTD4),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化设置值,实际值1：实际温度变化值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化設置值,實際值1：實際溫度變化值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Temperature change Set value, actual value 1: actual temperature change value, actual value 2: current temperature value, actual value 3: last temperature value",
0x00000203,LOG_FP4_PRESS_SUDDENLY,,,层板压力全时突升\突降(FP4),,Laminates full time pressure rise / fall (FP4),2,0,,,層板壓力全時突升\突降(FP4),,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg,实际值2:{3}Kg,实际值3:{4}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：压力变化设置值,实际值1：实际压力变化值,实际值2：当前压力值,实际值3：上次压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg,實際值2:{3}Kg,實際值3:{4}Kg","Time :{0}, set value :{1}Kg, actual value 1:{2}Kg, actual value 2:{3}Kg, actual value 3:{4}Kg","設置值：壓力變化設置值,實際值1：實際壓力變化值,實際值2：當前壓力值,實際值3：上次壓力值","Setting value: pressure change Setting value, actual value 1: actual pressure change value, actual value 2: current pressure value, actual value 3: last pressure value",
0x00000204,LOG_EXECUTE_PPM_PROTECT_ACTION,,,因执行PPM自定义组合动作而保护,,Protect by performing PPM custom combination actions,2,0,,,因執行PPM自定義組合動作而保護,,,,,,,,,,,,,,160
0x00000205,LOG_SEND_PPM_FIRE_ACTION,,,下发PPM消防动作信号,,Send PPM fire signal,2,0,,,下發PPM消防動作信號,,,,,,,,,,,,,,
0x00000206,LOG_SECOND_GROUP_AUX_PRESS_SUDDENLY,,,二次组合压力突变值达到设置值,,The two combined pressure catastrophe reaches the set value.,0,0,,,二次組合壓力突變值達到設置值,,,,,,,,,,,,,,
0x00000207,LOG_SET_FORMAT_TESTING,,,设置化成状态测试中,,Set into a state test,0,0,,,設置化成狀態測試中,,,,,,,,,,,,,,
0x00000208,LOG_DT_dt,,,DT/dt保护,,DT/dt protection,2,0,,,DT/dt保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度设置值,实际值1：温度值实际值,实际值2：当前温度值,实际值3：上次温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度設置值,實際值1：溫度值實際值,實際值2：當前溫度值,實際值3：上次溫度值","Set value: Set temperature value, actual value 1: actual temperature value, actual value 2: current temperature value, actual value 3: last temperature value",160
0x00000209,LOG_FIX_ABNORMAL_SIGNAL,,,整列异常信号保护,,Whole column anomaly signal protection,2,0,,,整列異常信號保護,,,,Y5,,,,,,,,,,160
0x0000020A,LOG_JIG_CHL_PROTECT_SIGNAL,,,发送化成异常报警信号,,Sending into an abnormal alarm signal,1,0,,,發送化成異常報警信號,,,,,,,,,,,,,,
0x0000020B,LOG_SEND_FIX_FIRE_SIGNAL,,,发送火警报警信号,,Send fire alarm signal,1,0,,,發送火警報警信號,,,,,,,,,,,,,,
0x0000020C,LOG_SEND_FAN_ABNORMAL_SIGNAL,,,发送风扇异常告警信号,,Send a fan exception alarm,1,0,,,發送風扇異常告警信號,,,,,,,,,,,,,,
0x0000020D,LOG_SEND_FAN_RECOVERY_NORMAL,,,发送风扇恢复正常解除告警信号,,Send fan to resume normal release alarm signal,1,0,,,發送風扇恢復正常解除告警信號,,,,,,,,,,,,,,
0x0000020E,LOG_AUTO_SPRAY_TEMP_UPPER_ABNORMAL_POINTS,,,因达到自动喷淋二次温度上限异常点个数而保护,,Protected by the number of abnormal points in the upper limit of the automatic spray secondary temperature,4,0,,,因達到自動噴淋二次溫度上限異常點個數而保護,,,,,,,,,,,,,,60
0x0000020F,LOG_SEND_ALARM_LIGHT_RED_BUZZER,,,三色报警灯_红灯亮_蜂鸣器响,,Tri-color warning light _ red light _ buzzer sound,1,0,,,三色報警燈_紅燈亮_蜂鳴器響,,,,,,,,,,,,,,
0x00000210,LOG_SEND_ALARM_LIGHT_YELLOW,,,三色报警灯_黄灯亮,,Tricolor alarm light _ yellow light,0,0,,,三色報警燈_黃燈亮,,,,,,,,,,,,,,
0x00000211,LOG_SEND_ALARM_LIGHT_YELLOW_TOGGLE,,,三色报警灯_黄灯闪烁,,Tri-color warning light _ yellow light flashes,0,0,,,三色報警燈_黃燈閃爍,,,,,,,,,,,,,,
0x00000212,LOG_SEND_ALARM_LIGHT_GREEN,,,三色报警灯_绿灯亮,,Tri-color warning light _ green light,0,0,,,三色報警燈_綠燈亮,,,,,,,,,,,,,,
0x00000213,LOG_SOC_LIMIT_CONDITION_JUMP,,,SOC截止限制条件跳转,,SOC cutoff limit condition jump,0,0,,,SOC截止限制條件跳轉,,,,,,,"时间:{0},设置值:{1}%,实际值1:{2}%,实际值2:{3}Ah,实际值3:{4}Ah","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：SOC截止条件,实际值1：实际SOC值,实际值2：当前容量值,实际值3：上次容量值","時間:{0},設置值:{1}%,實際值1:{2}%,實際值2:{3}Ah,實際值3:{4}Ah","Time :{0}, set value :{1}%, actual value 1:{2}%, actual value 2:{3}Ah, actual value 3:{4}Ah","設置值：SOC截止條件,實際值1：實際SOC值,實際值2：當前容量值,實際值3：上次容量值","Set value: SOC cut-off condition, actual value 1: actual SOC value, actual value 2: current capacity, actual value 3: last capacity",
0x00000214,LOG_VOLT_BIAS_PROT_BY_TIME,,,间隔时间电压偏差保护,,Interval voltage deviation protection,2,,,,間隔時間電壓偏差保護,,FJB85-1|FJBA3-2,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：电压变化值,实际值2：当前电压值,实际值3：前面电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：電壓變化值,實際值2：當前電壓值,實際值3：前面電壓值","Set value: Voltage set value, actual value 1: voltage change value, actual value 2: current voltage value, actual value 3: previous voltage value",160
0x00000215,LOG_PROT_BY_OTHER_VOLT_BIAS_TIME,,,因其它通道间隔时间电压偏差保护而保护,,Protected by other channel interval time voltage deviation protection,2,,,,因其它通道間隔時間電壓偏差保護而保護,,,,,,,,放电能量字段复用，放电能量字段复用，单元号复用放电能量字段Byte1，通道号复用放电能量字段Byte0,,,,,,20
0x00000216,LOG_REST_V6_VOL_SUDDENLY_DOWN,,,[V6] 搁置时电压突降2,,[V6] Voltage dips on hold 2,2,,,,[V6] 擱置時電壓突降2,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压差值,实际值2：当前电压值,实际值3：上一次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓差值,實際值2：當前電壓值,實際值3：上一次電壓值","Set value: Set value of voltage, actual value 1: actual voltage difference, actual value 2: current voltage, actual value 3: last voltage",20
0x00000217,LOG_IF_DBC,,,DBC条件表达式判断,,DBC Condition,2,0,,,DBC設定值和採集的實際值做條件比較,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置BMS值,实际值1：实际BMS值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：設置BMS值,實際值1：實際BMS值",Set value: Set the BMS value. Actual value 1: actual BMS value,
0x00000218,LOG_IF_AUXTEMP,,,如果参数的辅助通道温度保护,,If parameters in the auxiliary channel temperature protection,2,,,,如果參數的輔助通道溫度保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：辅助温度设置值,实际值1：实际辅助温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：輔助溫度設置值,實際值1：實際輔助溫度值","Set value: auxiliary temperature set value, actual value 1: actual auxiliary temperature",60
0x00000219,LOG_IF_AUXVOL,,,如果参数的辅助通道电压保护,,If parameters in the auxiliary channel voltage protection,2,,,,如果參數的輔助通道電壓保護,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：辅助电压设置值,实际值1：实际辅助电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：輔助電壓設置值,實際值1：實際輔助電壓值","Set value: auxiliary voltage set value, actual value 1: actual auxiliary voltage",20
0x0000021A,LOG_DELTA_V_UP,,,Delta V上限保护（正电压间隔变化量）,,Delta Voltage Upper Limit Protection (Positive Voltage Interval Variation),2,,,,Delta V上限保護（正電壓間隔變化量）,,,P6,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压差值,实际值2：当前电压值,实际值3：上一次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓差值,實際值2：當前電壓值,實際值3：上一次電壓值","Set value: Set value of voltage, actual value 1: actual voltage difference, actual value 2: current voltage, actual value 3: last voltage",20
0x0000021B,LOG_DELTA_V_DOWN,,,Delta V下限保护（负电压间隔变化量）,,Delta Voltage Lower Limit Protection (Negative Voltage Interval Variation),2,,,,Delta V下限保護（負電壓間隔變化量）,,,P7,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压差值,实际值2：当前电压值,实际值3：上一次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓差值,實際值2：當前電壓值,實際值3：上一次電壓值","Set value: Set value of voltage, actual value 1: actual voltage difference, actual value 2: current voltage, actual value 3: last voltage",20
0x0000021C,LOG_DELTA_I_UP,,,Delta I上限保护（正电流间隔变化量）,,Delta Current Upper Limit Protection (Positive Current Interval Variation),2,,,,Delta I上限保護（正電流間隔變化量）,,,P8,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：实际电流差值,实际值2：当前电流值,实际值3：上一次电流值","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流設置值,實際值1：實際電流差值,實際值2：當前電流值,實際值3：上一次電流值","Setting value: current setting value, actual value 1: actual current difference value, actual value 2: current current value, actual value 3: last current value",10
0x0000021D,LOG_DELTA_I_DOWN,,,Delta I下限保护（负电流间隔变化量）,,Delta Current Lower Limit Protection (Negative Current Interval Variation),2,,,,Delta I下限保護（負電流間隔變化量）,,,P9,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流设置值,实际值1：实际电流差值,实际值2：当前电流值,实际值3：上一次电流值","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流設置值,實際值1：實際電流差值,實際值2：當前電流值,實際值3：上一次電流值","Setting value: current setting value, actual value 1: actual current difference value, actual value 2: current current value, actual value 3: last current value",10
0x0000021E,LOG_CAN_OFFLINE,,,CAN通讯超时或异常掉线保护,,CAN Communication Overtime or Abnormal Offline Protection,2,,,,CAN通訊超時或異常掉線保護,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：第几路CAN掉线（1-8）,实际值1：第几路CAN掉线（1-8）","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：第幾路CAN掉線（1-8）,實際值1：第幾路CAN掉線（1-8）","Set value: Which route CAN be disconnected (1-8), Actual value 1: Which route CAN be disconnected (1-8)",160
0x0000021F,LOG_COMM_ABNOR_TO_START,,,通讯异常时保护通道，并且开始做超时标记,,Protect the channel when communication is abnormal and start timeout marking,2,,,,通訊異常時保護通道，並且開始做超時標記,,,,,,,,,,,,,,160
0x00000220,LOG_COMM_ABNOR_TO_END,,,通讯异常超时时间到，保护所有通道并标记,,"Communication abnormal timeout arrives, protect all channels and mark",2,,,,通訊異常超時時間到，保護所有通道並標記,,,,,,,,,,,,,,160
0x00000221,LOG_NEILSBED_ABNORMAL_CO1,,,压床异常---一氧化碳(CO)报警,,Abnormal Pressure Bed - Carbon Monoxide Alarm,3,0,,,壓床異常---一氧化碳(CO)報警,,,,,1,,,,,,,,,
0x00000222,LOG_POWERON_ALLOCK_CHL_ERR,,,"上电时,分配通道失败",,Distribution channel fails when power is on,0,,,,"上電時,分配通道失敗",,,,,,,,,,,,,,
0x00000223,LOG_POWERON_ALLOCK_PARALLEL_CHL_ERR,,,"上电时,分配并联通道失败",,Failure to allocate parallel channels when power is on,0,,,,"上電時,分配並聯通道失敗",,,,,,,,,,,,,,
0x00000224,LOG_RTC_VL,,,"RTC电池低压保护,请更换电池并同步RTC",,"Low Voltage Protection for RTC Batteries, Replace Batteries and Synchronize RTC",2,,,,"RTC電池低壓保護,請更換電池並同步RTC",,,,,,,,,,,,,,160
0x00000225,LOG_RTC_DAYDIFF,,,"RTC时间天数间隔异常保护,请同步RTC",,"RTC time interval abnormal protection, please synchronize RTC",2,,,,"RTC時間天數間隔異常保護,請同步RTC",,,,,,,,,,,,,,160
0x00000226,LOG_BACKUP_SUCC,,,成功备份中位机文件到上位机,,Successful Backup of ZWJ Files to SWJ,0,,,,成功備份中位機文件到上位機,,,,,,,,,,,,,,
0x00000227,LOG_UPDATE_SUCC,,,中位机升级成功,,The control unit Upgrade Successful,0,,,,中位機升級成功,,,,,,,,,,,,,,
0x00000228,LOG_CVC_RAISESPEED,,,恒压电流上升速率,,Constant Voltage Current Rising Rate,2,,,,恆壓電流上升速率,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流变化设置值,实际值1：实际电流变化值,实际值2：当前电流值,实际值3：前面电流值","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流變化設置值,實際值1：實際電流變化值,實際值2：當前電流值,實際值3：前面電流值","Setting value: current change Setting value, actual value 1: actual current change value, actual value 2: current value, actual value 3: previous current value",10
0x00000229,LOG_FILESMOKE_AND_TP_ALARM,,,触发喷淋(烟雾与TP组合触发报警),,Triggered Spray (Combination of Smoke and TP Triggered Alarm),3,,,,觸發噴淋(煙霧與TP組合觸發報警),,,,,,,,,,,,,,
0x0000022A,LOG_SET_PARALLEL,,,设置下位机硬件并联失败,,Set the lower computer hardware parallel failure,2,,,,設置下位機硬件並聯失敗,,,,,,,,,,,,,,100
0x0000022B,LOG_FIX_TEMP_RANGE_ABNORMAL,,,夹具温度范围异常,,Abnormal fixture temperature range,2,,,,夾具溫度範圍異常,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：夹具温度范围设置值,实际值1：当前夹具温度范围实际值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：夾具溫度範圍設置值,實際值1：當前夾具溫度範圍實際值","Set value: Set value, actual value 1: actual value of the current temperature range of the fixture",60
0x0000022C,LOG_CURRENT_ABNORMAL_CHARGE,,,充电电流异常,,Abnormal charging current,2,,,,充電電流異常,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：前面实际电流值,实际值1：当前实际电流值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：前面實際電流值,實際值1：當前實際電流值","Setting value: previous actual current value, actual value 1: current actual current value",10
0x0000022D,LOG_CURRENT_ABNORMAL_DISCHARGE,,,放电电流异常,,Abnormal discharge current,2,,,,放電電流異常,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：前面实际电流值,实际值1：当前实际电流值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：前面實際電流值,實際值1：當前實際電流值","Setting value: previous actual current value, actual value 1: current actual current value",10
0x0000022E,LOG_CURRENT_ABNORMAL_OTHER_CHL,,,其它通道充放电电流异常,,Other channels charge and discharge current abnormal,2,,,,其它通道充放電電流異常,,,,,,,,放电能量字段复用，放电能量字段复用，单元号复用放电能量字段Byte1，通道号复用放电能量字段Byte0,,,,,,10
0x0000022F,LOG_OCV_VOL_UP,,,OCV工步，超出电压上限,,"OCV step, exceeding the upper voltage limit",2,,,,OCV工步，超出電壓上限,,,G8,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,20
0x00000230,LOG_OCV_VOL_DOWN,,,OCV工步，低于电压下限,,"OCV step, below the lower voltage limit",2,,,,OCV工步，低於電壓下限,,,G9,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压下限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓下限值,實際值1：當前電壓值",Set value: lower voltage limit. Actual value 1: current voltage value,20
0x00000231,LOG_WX_SMOKE1_AND_SMOKE2,,,触发消防火警报警(烟感1与烟感2),,Fire alarm fire (smoke 1 and smoke 2),2,,,,觸發消防火警報警(煙感1與煙感2),,,,Y6,,,,,,,,,,
0x00000232,LOG_WX_SMOKE1_AND_TP2,,,触发消防火警报警(烟感1与TP2),,Fire alarm fire (smoke sensor 1 and TP2),2,,,,觸發消防火警報警(煙感1與TP2),,,,Y7,,,,,,,,,,
0x00000233,LOG_WX_SMOKE2_AND_TP2,,,触发消防火警报警(烟感2与TP2),,Fire alarm fire (smoke 2 and TP2),2,,,,觸發消防火警報警(煙感2與TP2),,,,Y8,,,,,,,,,,
0x00000234,LOG_WX_TP1_AND_TP2,,,触发消防火警报警(TP1与TP2),,Fire alarm fire (TP1 and TP2),2,,,,觸發消防火警報警(TP1與TP2),,,,Y9,,,,,,,,,,
0x00000235,LOG_WX_TP2,,,超出火警温度上限(TP2),,Fire alarm temperature limit exceeded (TP2),2,,,,超出火警溫度上限(TP2),,,,Y4,,,,,,,,,,60
0x00000236,LOG_WX_TP1,,,超出环境温度上限(TP1),,Exceeding ambient temperature limit (TP1),2,,,,超出環境溫度上限(TP1),,,,Y3,,,,,,,,,,60
0x00000237,LOG_MAIN_FAN_ERR,,,设备风机异常,,The equipment fan is abnormal,2,,,,設備風機異常,,,,,,,,,,,,,,
0x00000238,LOG_INTO_CV,,,开始恒压阶段的电流截止判断,,Current cut-off judgment at the beginning of the constant voltage phase,-1,,,,開始恆壓階段的電流截止判斷,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：恒压时的电压设置值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：恒壓時的電壓設置值,實際值1：當前電壓值",Set value: Voltage set value at constant voltage. Actual value 1: current voltage,10
0x00000239,LOG_WYB_COMM_ERR,,,万用表通信失败,,Multimeter communication failed,2,,,,萬用表通信失敗,,,,,,,,,,,,,,
0x0000023A,LOG_NEILSBED_ABNORMAL_SMOKE_SOLID,,,烟雾颗粒报警,,Smoke particle alarm,2,,,,煙霧顆粒報警,,,,,,,,,,,,,,
0x0000023B,LOG_NEILSBED_ABNORMAL_SMOKE_GAS,,,烟雾气敏浓度报警,,Smoke gas concentration alarm,2,,,,煙霧氣敏濃度報警,,,,,,,,,,,,,,
0x0000023C,LOG_DC_MODULE_STATUE_ERR,,,模块开关状态错误,,Module switch status error,2,,,,模塊開關狀態錯誤,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：状态设置值,实际值1：当前状态值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：狀態設置值,實際值1：當前狀態值","Set value: Status set value, actual value 1: current status value",
0x0000023D,LOG_BATTERY_CONTACT_RESULT,,,发送电池接触检测结果,,Send battery contact test results,0,,,,發送電池接觸檢測結果,,,,,,,,,,,,,,
0x0000023E,LOG_BATT_DIR_ERR,,,通道极性状态错误,,Channel polarity status error,2,,,,通道極性狀態錯誤,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：通道极性设置值,实际值1：当前通道极性状态值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：通道極性設置值,實際值1：當前通道極性狀態值",Set value: Set channel polarity value. Actual value 1: Current channel polarity status value,
0x0000023F,LOG_IF_PROTECT_CTRL_H,,,湿度条件保护,,Humidity protection,2,,,,濕度條件保護,,,,,,,"时间:{0},设置值:{1}%,实际值1:{2}%","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：湿度设置值,实际值1：当前湿度值","時間:{0},設置值:{1}%,實際值1:{2}%","Time :{0}, set value :{1}%, actual value :{2}%","設置值：濕度設置值,實際值1：當前濕度值",Set value: Humidity Set value. Actual value 1: Current humidity,160
0x00000240,LOG_CTRL_H_CHANGE,,,湿度变更控制,,Humidity change control,0,,,,濕度變更控制,,,,,,,"时间:{0},设置值:{1}%,实际值1:{2}%","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：湿度设置值,实际值1：湿度设置值","時間:{0},設置值:{1}%,實際值1:{2}%","Time :{0}, set value :{1}%, actual value :{2}%","設置值：濕度設置值,實際值1：濕度設置值",Set value: Humidity set value. Actual value 1: humidity set value,
0x00000241,LOG_CTRL_THICK,,,测厚控制,,Thickness control,0,,,,測厚控制,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：厚度设置值,实际值1：厚度设置值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：厚度設置值,實際值1：厚度設置值","Set value: thickness set value, actual value 1: thickness set value",
0x00000242,LOG_IF_PROTECT_BATT_THICK,,,电池厚度保护,,Battery thickness protection,2,,,,電池厚度保護,,,,,,,"时间:{0},设置值:{1}mm,实际值1:{2}mm","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：厚度设置值,实际值1：当前厚度值","時間:{0},設置值:{1}mm,實際值1:{2}mm","Time :{0}, set value :{1}mm, actual value :{2}mm","設置值：厚度設置值,實際值1：當前厚度值","Set value: Thickness set value, actual value 1: current thickness value",160
0x00000243,LOG_ENABLE_PRESET_APPOINTMENT_PAUSE_TIME_SET,,,启用预设时间暂停,,Enable preset time pause,0,,,,啟用預設時間暫停,,,,,,,,,,,,,,
0x00000244,LOG_ENABLE_PRESET_APPOINTMENT_PAUSE_TIME_CANCLE,,,取消预设时间暂停,,Cancel preset time pause,0,,,,取消預設時間暫停,,,,,,,,,,,,,,
0x00000245,LOG_JSWL_SMOKE_AND_SECONDPROTECT,,,触发火警（烟感与电压或温度的二次保护）,,Fire alarm (secondary protection of smoke and voltage or temperature),2,,,,觸發火警（煙感與電壓或溫度的二次保護）,,,,,,,,,,,,,,60
0x00000246,LOG_JSWL_TWO_SECONDPROTECT,,,触发火警（电压与温度都产生二次保护）,,Fire alarm triggered (secondary protection for both voltage and temperature),2,,,,觸發火警（電壓與溫度都產生二次保護）,,,,,,,,,,,,,,60
0x00000247,LOG_THICKNESS_COMM_ERR,,,厚度仪通信异常,,Thickness communication err,2,,,,厚度儀通信異常,,,,,,,,,,,,,,
0x00000248,LOG_EF_COMM_ERR,,,膨胀力通信异常,,EF communication err,2,,,,膨脹力通信異常,,,,,,,,,,,,,,
0x00000249,LOG_IF_SMBUS,,,SMBUS条件表达式,,SMBUS condition expression,2,,,,SMBUS條件表達式,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：SMBUS设置值,实际值1：SMBUS实时值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：SMBUS設置值,實際值1：SMBUS實時值","Set value: SMBUS Set value, actual value 1: SMBUS real-time value",
0x0000024A,LOG_SMBXWJ_OFFLINE,,,SMBUS下位机掉线,,SMBUS device offline,2,,,,SMBUS下位機掉線,,,,,,,,,,,,,,100
0x0000024B,LOG_SMBXWJ_CHL_NO_SYNCCFG,,,SMBUS下位机通道未更新配置或更新配置失败,,SMBUS did not update channel config,2,,,,SMBUS下位機通道未更新配置或更新配置失敗,,,,,,,,,,,,,,100
0x0000024C,LOG_SMBXWJ_CHL_COMM_TO,,,SMBUS下位机通道通讯超时,,SMBUS channel communication timeout,2,,,,SMBUS下位機通道通訊超時,,,,,,,,,,,,,,100
0x0000024D,LOG_LOAD_CANCFG_PARA_ERR,,,同步CAN下位机配置时加载资源发生错误,,SMBUS download resources err,2,,,,同步CAN下位機配置時加載資源發生錯誤,,,,,,,,,,,,,,100
0x0000024E,LOG_WX_COMM_ERR,,,温箱通讯异常,,Abnormal communication of WX,2,,,,溫箱通訊異常,,,,,,,,,,,,,,
0x0000024F,LOG_COOLWATER_OpenON,,,开启冷水机,,Turn on the cooling-water machine,0,,,,開啟冷水機,,,,,,,,,,,,,,
0x00000250,LOG_COOLWATER_OpenOFF,,,关闭冷水机,,Turn on the cooling-water machine,0,,,,關閉冷水機,,,,,,,,,,,,,,
0x00000251,LOG_COOLWATER_COMM_ERR,,,冷水机通讯异常,,Abnormal communication of cooling-water machine,2,,,,冷水機通訊異常,,,,,,,,,,,,,,
0x00000252,LOG_COOLWATER_NOSET_PARM,,,无设置冷水机的参数,,There is no parameter for setting water chiller,2,,,,無設置冷水機的參數,,,,,,,,,,,,,,
0x00000253,LOG_COOLWATER_SET_TEMP_FLOW,,,设置温度和流量,,Setting temperature and flow,0,,,,設置溫度和流量,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}L/min","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：设置流量值","時間:{0},設置值:{1}℃,實際值1:{2}L/min","Time :{0}, set value :{1}℃, actual value :{2}L/min","設置值：設置溫度值,實際值1：設置流量值",Set value: Set temperature value. Actual value 1: Set traffic value,
0x00000254,LOG_CH_VOLT_FALL,,,通道停止保护时电压达到下降条件,,The voltage reaches the falling condition when the channel stops protection,2,,,,通道停止保護時電壓達到下降條件,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压下降设置值,实际值1：实际电压变化值,实际值2：当前电压值,实际值3：前面电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓下降設置值,實際值1：實際電壓變化值,實際值2：當前電壓值,實際值3：前面電壓值","Set value: voltage drop Set value, actual value 1: actual voltage change value, actual value 2: current voltage value, actual value 3: previous voltage value",160
0x00000255,LOG_OTHER_CH_VOLT_FALL,,,有通道停止保护时电压达到了下降条件,,"When the channel stops protection, the voltage reaches the drop condition",2,,,,有通道停止保護時電壓達到了下降條件,,,,,,,,放电能量字段复用，放电能量字段复用，单元号复用放电能量字段Byte1，通道号复用放电能量字段Byte0,,,,,,160
0x00000256,LOG_CTRL_CANDATA,,,控制工步发送CAN数据帧,,Control step sends can data frame,0,,,,控制工步發送CAN數據幀,,,,,,,,,,,,,,
0x00000257,LOG_CUTOFF_BY_VOLT_DIFF_OF_TIMES,,,指定时间间隔满足电压差截止,,The specified time interval meets the voltage difference cut-off,-1,,,,指定時間間隔滿足電壓差截止,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压偏差设置值,实际值1：实际偏差值,实际值2：最新电压值,实际值3：最早电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓偏差設置值,實際值1：實際偏差值,實際值2：最新電壓值,實際值3：最早電壓值","Set value: Voltage deviation Set value, actual value 1: actual deviation value, actual value 2: latest voltage value, actual value 3: earliest voltage value",
0x00000258,LOG_485BMS_OFFLINE,,,485 BMS通讯掉线,,485 BMS communication dropped,2,,,,485 BMS通訊掉線,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：第几路485掉线（1-8）,实际值1：第几路485掉线（1-8）","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：第幾路485掉線（1-8）,實際值1：第幾路485掉線（1-8）","Set value: Route 485 disconnected (1-8), Actual value 1: Route 485 disconnected (1-8)",
0x00000259,LOG_DBC_ARRIVE_JUMP_CONDICTION,,,DBC发送给BMS数据后符合跳转下一个工步条件,,The DBC triggers the next work step condition,0,,,,DBC發送給BMS數據後符合跳轉下壹個工步條件,,,,,,,,,,,,,,
0x0000025A,LOG_CYCLE_DCHG_SOH_LOW_LIMIT,,,循环放电结束容量下限,,Lower limit of capacity at the end of cycle discharge,2,,,,循環放電結束容量下限,,,,,,,,,,,,,,40
0x0000025B,LOG_CYCLE_CHG_CAP_BIAS,,,循环充电容量偏差,,Cycle charge capacity deviation,2,,,,循環充電容量偏差,,,,,,,,,,,,,,40
0x0000025C,LOG_CYCLE_DCHG_CAP_BIAS,,,循环放电容量偏差,,Cycle charge capacity deviation,2,,,,循環充電容量偏差,,,,,,,,,,,,,,40
0x0000025D,LOG_VOLT_BIAS_BEGNG,,,指定时间内启始电压偏差,,Starting voltage deviation within specified time,2,,,,指定時間內啟始電壓偏差,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：电压偏差设置值,实际值1：当前电压值,实际值2：上次电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V","設置值：電壓偏差設置值,實際值1：當前電壓值,實際值2：上次電壓值","Set value: Voltage deviation Set value, actual value 1: current voltage value, actual value 2: last voltage value",
0x0000025E,LOG_STEP_TIME_OVERTIME,,,工步运行时间上限,,Upper limit of step running time,2,,,,工步運行時間上限,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置时间上限,实际值1：当前工步时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：設置時間上限,實際值1：當前工步時間",Set value: Set the upper limit of time. Actual value 1: current working time,
0x0000025F,LOG_FOLLOW_STEP_CFGERR,,,跟随工步参数配置错误,,Configuration error of following step parameters,2,,,,跟隨工步參數配置錯誤,,,,,,,,,,,,,,
0x00000260,LOG_FOLLOW_STEP_DOWNLOAD_ERR,,,跟随工步值下发失败,,Follow step value distribution failed,2,,,,跟隨工步值下發失敗,,,,,,,,,,,,,,
0x00000261,LOG_FOLLOW_STEP_CUTOFF,,,跟随工步截止,,Follow the work step to the end,0,,,,跟隨工步截止,,,,,,,"时间:{0},设置值:{1},实际值1:{2},实际值2:{3}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：截止电流或功率设置值,实际值1：当前BMS实际值,实际值2：截止上下偏移值","時間:{0},設置值:{1},實際值1:{2},實際值2:{3}","Time :{0}, set value :{1}, actual value 1:{2}, actual value 2:{3}","設置值：截止電流或功率設置值,實際值1：當前BMS實際值,實際值2：截止上下偏移值",Set value: Cut-off current or power set value. Actual value 1: actual value of the current BMS. Actual value 2: offset value of the upper and lower cut-off,
0x00000262,LOG_FILESYSTERM_RO,,,中位机文件系统只读，请重启(冷启动)系统尝试修复；,,The file system is read-only,2,,,,中位機文件系統只讀,,,,,,,,,,,,,,90
0x00000263,LOG_TAIL_DELAY_PROTECT,,,拖尾电芯强制判死保护,,Tail-cell mandatory death protection,2,,,,拖尾電芯強制判死保護,,,,,,,,,,,,,,160
0x00000264,LOG_SYSTEM_FAN_ABNORMAL_PRE_STEP,,,工步切换时发现设备风扇异常,,The fan of the device is found to be abnormal when the work step is switched,2,,,,工步切換時發現設備風扇異常,,,,,,,,,,,,,,
0x00000265,LOG_JIG_OPEN_WHEN_WORKING,,,测试中夹具张开保护,,Fixture opening protection during test,2,,,,測試中夾具張開保護,,,,,,,,,,,,,,160
0x00000266,LOG_START_OCV_ACR,,,启动OCV&ACR测试,,Start OCV&ACR test,-1,,,,啟動OCV&ACR測試,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置OCV值,实际值1：设置OCV值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：設置OCV值,實際值1：設置OCV值",Set value: Set the OCV value. Actual value 1: set the OCV value,
0x00000267,LOG_IF_PROTECT_PRESS_ERR_LOWLIMIT,,,条件工步压力保护(下限),,Conditional step pressure protection (lower limit),2,,,,條件工步壓力保護(下限),,,,,,,"时间:{0},设置值:{1}Kg,实际值1:{2}Kg","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置压力值,实际值1：实际压力值","時間:{0},設置值:{1}Kg,實際值1:{2}Kg","Time :{0}, set value :{1}Kg, actual value :{2}Kg","設置值：設置壓力值,實際值1：實際壓力值",Set value: Set pressure value. Actual value 1: actual pressure value,160
0x00000268,LOG_FIX_TEMP_RANGE_ABNORMAL_TOPLIMIT,,,夹具温度范围异常(上限),,Fixture temperature range abnormal (upper limit),2,,,,夾具溫度範圍異常(上限),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,60
0x00000269,LOG_FIX_TEMP_RANGE_ABNORMAL_LOWLIMIT,,,夹具温度范围异常(下限),,Fixture temperature range abnormal (lower limit),2,,,,夾具溫度範圍異常(下限),,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置温度值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：設置溫度值,實際值1：實際溫度值",Set value: Set the temperature. Actual value 1: actual temperature,60
0x0000026A,LOG_FIRE_ALARM_TO_POWEROFF,,,火警保护并断功率电源,,Fire protection and power cut off,2,,,,火警保護並斷功率電源,,,,,,,,,,,,,,
0x0000026B,LOG_CCC_ADJACENT_VOLT_DROP,,,恒流充电相邻电压下降异常保护,,Constant current charging adjacent voltage drop abnormal protection,2,,,,恆流充電相鄰電壓下降異常保護,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压变化量,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓變化量,實際值1：當前電壓值",Set value: Voltage change. Actual value 1: current voltage,20
0x0000026C,LOG_OCV_ACR_IF_PROTEDTED,,,OCV ACR 测试如果条件保护,,OCV ACR test if conditions protect,2,,,,OCV ACR 測試如果條件保護,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置OCV值,实际值1：实际OCV值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：設置OCV值,實際值1：實際OCV值",Set value: Set the OCV value. Actual value 1: actual OCV value,160
0x0000026D,LOG_THIRD_PARTY_COMM_ERR,,,与第三方的通信中断,,Interruption of communication with third parties,2,,,,與第三方的通信中斷,,,,,,,,,,,,,,
0x0000026E,LOG_CTRL_NOSUPPORT_CANDATA,,,不支持此类型CAN控制工步,,This type of CAN control step is not supported,2,,,,不支持此類型CAN控制工步,,,,,,,,,,,,,,
0x0000026F,LOG_CTRL_BMSDATA,,,控制工步发送BMS数据帧,,Control process step to send BMS data frame,-1,,,,控制工步發送BMS數據幀,,,,,,,,,,,,,,
0x00000270,LOG_CV_CURRENT_PERCENT,,,恒压充电时电流突升超出百分比,,The current surge exceeds the percentage during constant voltage charging,2,,,,恆壓充電時電流突升超出百分比,,,,,,,"时间:{0},设置值:{1}%,实际值1:{2}A,实际值2:{3}A,实际值3:{4}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电流百分比,实际值1：电流变化值,实际值2：当前电流,实际值3：上一个电流","時間:{0},設置值:{1}%,實際值1:{2}A,實際值2:{3}A,實際值3:{4}A","Time :{0}, set value :{1}%, actual value 1:{2}A, actual value 2:{3}A, actual value 3:{4}A","設置值：電流百分比,實際值1：電流變化值,實際值2：當前電流,實際值3：上一個電流","Setting value: current percentage, actual value 1: current change value, actual value 2: current current, actual value 3: previous current",10
0x00000271,LOG_CV_VOLT_GREATER_THAN_VOLT_SETTED,,,恒压充电时电池电压大于等于设置恒压值,,"During constant voltage charging, the battery voltage is greater than or equal to the set constant voltage value",1,,,,恆壓充電時電池電壓大於等於設置恆壓值,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：设置电压值,实际值1：当前电压,实际值2：设置值加上电压偏差值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V","設置值：設置電壓值,實際值1：當前電壓,實際值2：設置值加上電壓偏差值",Set value: Set voltage value. Actual value 1: current voltage. Actual value 2: set value plus voltage deviation value,
0x00000272,LOG_DV_VOLT_LESS_THAN_VOLT_SETTED,,,恒压放电时电池电压小于等于设置恒压值,,"During constant voltage discharge, the battery voltage is less than or equal to the set constant voltage value",1,,,,恆壓放電時電池電壓小於等於設置恆壓值,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：设置电压值,实际值1：当前电压,实际值2：设置值减去电压偏差值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V","設置值：設置電壓值,實際值1：當前電壓,實際值2：設置值減去電壓偏差值",Set value: Set voltage value. Actual value 1: current voltage. Actual value 2: set value minus voltage deviation,
0x00000273,LOG_CTRL_TEMPHEATING_SWITCH,,,温度加热开关,,Temperature heating switch ,-1,,,,溫度加熱開關,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度加热开关设置值,实际值1：温度加热开关设置值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：溫度加熱開關設置值,實際值1：溫度加熱開關設置值","Set value: Set value of the temperature heating switch, actual value 1: set value of the temperature heating switch",
0x00000274,LOG_LONG_TIME_NULL_DF,,,中位机DF数据记录异常，请尝试关电重启恢复,,"The DF data record of the Control unit is abnormal, please try to power off and restart",2,,,,中位機DF數據記錄異常，請嘗試關電重啟恢復,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：错误码,实际值1：错误码","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：錯誤碼,實際值1：錯誤碼",Set value: error code. Actual value 1: error code,90
0x00000275,LOG_FIRST_REST_STEP_VOL_DOWN,,,"首次搁置电压保护,低于电压下限",,"First use voltage protection, lower than the voltage limit",2,,,,"首次擱置電壓保護,低於電壓下限",,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,20
0x00000276,LOG_WXT_RETURN_ERRCODE,,,转换器返回错误码,,Converter returns error code,2,,,,轉換器返回錯誤碼,,,,,,,,,,,,,,
0x00000277,LOG_WXT_LISTNULLBUTSETPLC,,,转换器查询链表为空,,The converter query chain is empty,2,,,,轉換器查詢鍊錶為空,,,,,,,,,,,,,,
0x00000278,LOG_WXT_NOCHLINTHISLEVEL,,,"转换器设置该层参数时,查询到该层不存在任何通道",,"When the converter sets the parameters for the layer, no channel exists for the layer queried",2,,,,"轉換器設置該層參數時,查詢到該層不存在任何通道",,,,,,,,,,,,,,
0x00000279,LOG_WXT_PLCOFFLINE,,,转换器与温箱掉线,,Converter dropped connection to temperature box,2,,,,轉換器與溫箱掉線,,,,,,,,,,,,,,
0x0000027A,LOG_WXT_SET_TEMP_FAILED,,,转换器设置温度失败,,Converter failed to set temperature,2,,,,轉換器設置溫度失敗,,,,,,,,,,,,,,60
0x0000027B,LOG_WXT_SET_HUMIDITY_FAILED,,,转换器设置湿度失败,,Converter failed to set humidity,2,,,,轉換器設置濕度失敗,,,,,,,,,,,,,,
0x0000027C,LOG_WXT_SET_THICK_FAILED,,,转换器设置厚度失败,,Converter failed to set thickness,2,,,,轉換器設置厚度失敗,,,,,,,,,,,,,,
0x0000027D,LOG_WXT_SET_PRESS_FAILED,,,转换器设置压力失败,,Converter failed to set pressure,2,,,,轉換器設置壓力失敗,,,,,,,,,,,,,,
0x0000027E,LOG_WXT_SET_WATERFLOW_FAILED,,,转换器设置水流量失败,,Converter failed to set water flow rate,2,,,,轉換器設置水流量失敗,,,,,,,,,,,,,,
0x0000027F,LOG_WXT_SET_DATATYPE_DIFF,,,转换器存在不同通道设置数据类型不一致的情况,,Converter has inconsistency in setting data type for different channels,2,,,,轉換器存在不同通道設置數據類型不一致的情況,,,,,,,,,,,,,,
0x00000280,LOG_WXT_SET_DATAVALUE_DIFF,,,转换器存在不同通道设置数据值不一致的情况,,The converter has inconsistent setting data values for different channels,2,,,,轉換器存在不同通道設置數據值不一致的情況,,,,,,,,,,,,,,
0x00000281,LOG_WXT_NOCONNPLCNOSETDATA,,,转换器没有连接温箱，不能设置数据,,The converter is not connected to the temperature box and cannot set the data,2,,,,轉換器沒有連接溫箱，不能設置數據,,,,,,,,,,,,,,
0x00000282,LOG_IDLE_ABNORMAL_PROTECT_ALL,,,空闲异常整机保护,,Idle abnormal whole machine protection,2,,,,空閒異常整機保護,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压上限值,实际值1：当前电压值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓上限值,實際值1：當前電壓值",Set value: Upper voltage limit. Actual value 1: Current voltage value,160
0x00000283,LOG_CTRL_YLJ_FLOW,,,设置液冷机流量,,Set the liquid cooler flow rate,-1,,,,設置液冷機流量,,,,,,,"时间:{0},设置值:{1}L/min,实际值1:{2}L/min","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置流量值,实际值1：设置流量值","時間:{0},設置值:{1}L/min,實際值1:{2}L/min","Time :{0}, set value :{1}L/min, actual value :{2}L/min","設置值：設置流量值,實際值1：設置流量值",Set value: Set traffic value. Actual value 1: Set traffic value,
0x00000284,LOG_YLJ_FLOW_IF_PROTEDTED,,,液冷流量 测试如果条件保护,,"Liquid cooling flow, test if conditions are protected",2,,,,"液冷流量,測試如果條件保護",,,,,,,"时间:{0},设置值:{1}L/min,实际值1:{2}L/min","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置流量值,实际值1：实际流量值","時間:{0},設置值:{1}L/min,實際值1:{2}L/min","Time :{0}, set value :{1}L/min, actual value :{2}L/min","設置值：設置流量值,實際值1：實際流量值",Set value: Set traffic value. Actual value 1: actual traffic value,160
0x00000285,LOG_CHL_STATUS_CHANGED,,,通道状态改变,,Channel status changed,0,,,,通道狀態改變,,,,,,,,,,,,,,
0x00000286,LOG_CTRL_VOC                       ,,,设置VOC控制工步,,Set VOC control worksteps,-1,,,,設置VOC控製工步,,,,,,,"时间:{0},设置值:{1}ppm,实际值1:{2}ppm","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置VOC值,实际值1：设置VOC值","時間:{0},設置值:{1}ppm,實際值1:{2}ppm","Time :{0}, set value :{1}ppm, actual value :{2}ppm","設置值：設置VOC值,實際值1：設置VOC值",Set value: Set VOC value. Actual value 1: Set VOC value,
0x00000287,LOG_WXT_SET_AIRPRESS_FAILED      ,,,设置气压失败,,Air pressure setting failed,2,,,,設置氣壓失敗,,,,,,,,,,,,,,
0x00000288,LOG_WXT_SET_VOC_FAILED            ,,,设置VOC失败,,VOC setting failed,2,,,,設置VOC失敗,,,,,,,,,,,,,,
0x00000289,LOG_WXT_SET_LIGHTSTATE_FAILED     ,,,设置三色灯失败,,Failed to set tricolor light,2,,,,設置三色燈失敗,,,,,,,,,,,,,,
0x0000028A,LOG_WXTS_BOXOVERTEMPALARM  ,,,温箱状态-温箱箱内超温报警,,status-temperature box inside the overtemperature alarm,2,,,,溫箱狀態-溫箱箱內超溫報警,,,,,,,,,,,,,,
0x0000028B,LOG_WXTS_HEATTUBEOVERTEMPALARM     ,,,温箱状态-发热管超温报警,,status-heating pipe overtemperature alarm,2,,,,溫箱狀態-發熱管超溫報警,,,,,,,,,,,,,,
0x0000028C,LOG_WXTS_TEMP1TEMP2_OUTOFRANGE     ,,,"温箱状态-温度1和2偏差大,超出范围",,"status-temperature 1 and 2 deviations are large, out of range",2,,,,"溫箱狀態-溫度1和2偏差大,超出範圍",,,,,,,,,,,,,,60
0x0000028D,LOG_WXTS_TEMPCONTROL_PROTACTION    ,,,温箱状态-机械温控表保护动作,,status-mechanical temperature control meter protection action,2,,,,溫箱狀態-機械溫控表保護動作,,,,,,,,,,,,,,160
0x0000028E,LOG_WXTS_OCV_ALARM                 ,,,温箱状态-VOC报警,,status-OC alarm,2,,,,溫箱狀態-VOC報警,,,,,,,,,,,,,,
0x0000028F,LOG_WXTS_SMOKEFILE_ALARM           ,,,温箱状态-烟感消防报警,,status-smoke fire alarm,2,,,,溫箱狀態-煙感消防報警,,,,,,,,,,,,,,
0x00000290,LOG_WXTS_COATFAN_ALARM             ,,,温箱状态-散热风扇故障,,status-cooling fan failure,2,,,,溫箱狀態-散熱風扇故障,,,,,,,,,,,,,,
0x00000291,LOG_WXTS_AIRTRANSPORTFAN_ALARM     ,,,温箱状态-运风风机故障,,status-air transport fan failure,2,,,,溫箱狀態-運風風機故障,,,,,,,,,,,,,,
0x00000292,LOG_WXTS_SMOKEEXHAUSTFAN_ALARM     ,,,温箱状态-排烟风机故障,,status-smoke exhaust fan failure,2,,,,溫箱狀態-排煙風機故障,,,,,,,,,,,,,,
0x00000293,LOG_WXTS_PLCANDVOCGAS_COMM_FAILED  ,,,温箱状态-PLC与VOC气体检测仪通讯异常,,status-abnormal communication between PLC and VOC gas detector,2,,,,溫箱狀態-PLC與VOC氣體檢測儀通訊異常,,,,,,,,,,,,,,
0x00000294,LOG_WXTS_PLCANDTEMPCONTRL_COMM_FAILED,,,温箱状态-PLC与温控器通讯异常,,status-abnormal communication between PLC and temperature controller,2,,,,溫箱狀態-PLC與溫控器通訊異常,,,,,,,,,,,,,,
0x00000295,LOG_WXTS_EMERGENCYSTOP_BUTTON_TRIGGER,,,温箱状态-急停按钮已按下,,status-emergency stop button has been pressed,2,,,,溫箱狀態-急停按鈕已按下,,,,,,,,,,,,,,
0x00000296,LOG_FOLLOW_STEP_BMS_DATA_ERR,,,跟随工步时BMS信号值错误,,BMS signal value error when following work step,2,,,,跟隨工步時BMS信號值錯誤,,,,,,,,,,,,,,
0x00000297,LOG_HIST_RESUME,,,历史接续,,History continues,0,,,,歷史接續,,,,,,,,,,,,,,
0x00000298,LOG_LASTVOL_FIRSTVOL_DIFF,,,非搁置工步-上一工步最后一条电压与当前工步第一条电压的差值超过设定的值,,Non reset step - the difference between the last voltage of the previous work step and the real-time voltage of the current work step exceeds the set value,1,,,,非擱置工步-上一工步最後一條電壓與當前工步第一條電壓的差值超過設定的值,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压差设置值,实际值1：实际电压差值,实际值2：上个工步最后电压,实际值3：此工步第一个电压","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓差設置值,實際值1：實際電壓差值,實際值2：上個工步最後電壓,實際值3：此工步第一個電壓","Set value: voltage difference Set value, actual value 1: actual voltage difference, actual value 2: the last voltage of the last working step, actual value 3: the first voltage of this working step",20
0x00000299,LOG_IF_CONDICTION_TOO_MANY,,,设置的如果条件参数过多保护,,Protection if too many condition parameters are set,2,,,,設置的如果條件參數過多保護,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：内部支持如果条件个数,实际值1：实际设置如果条件个数","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：內部支持如果條件個數,實際值1：實際設置如果條件個數",Set value: Number of internal supported if conditions. Actual value 1: number of actual set if conditions,160
0x0000029A,LOG_CURRENT_TOLERANCE_ABNORMAL_CNT,,,电流超差保护点数超过异常个数,,The number of current out of tolerance protection points exceeds the number of exceptions,1,,,,電流超差保護點數超過異常個數,,,,,,,"时间:{0},设置值:{1},实际值1:{2},实际值2:{3}mS,实际值3:{4},实际值4:{5}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)/放电能量(实际值4，NULL)","设置值：设置个数,实际值1：当前个数,实际值2：设置时间长度,实际值3：开始时间,实际值4：结束时间","時間:{0},設置值:{1},實際值1:{2},實際值2:{3}mS,實際值3:{4},實際值4:{5}","Time: {0}, setting: {1}, the actual value 1: {2}, the actual value 2: {3} mS, actual value 3: {4}, the actual value 4: {5}","設置值：設置個數,實際值1：當前個數,實際值2：設置時間長度,實際值3：開始時間,實際值4：結束時間","Set value: Set number, actual value 1: current number, actual value 2: set time length, actual value 3: start time, actual value 4: end time",10
0x0000029B,LOG_V8_PROTECT,,,V8组合忙时保护,,V8 combination busy time protection,2,,,,V8組合忙時保護,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压,实际值3：前面电压","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓,實際值3：前面電壓","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage, actual value 3: previous voltage",160
0x0000029C,LOG_V8_LIVE_TIME_START,,,V8时间生存周期开始,,V8 time life cycle begins,0,,,,V8時間生存周期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000029D,LOG_V8_LIVE_TIME_END,,,V8时间生存周期结束,,V8 time life cycle ends,0,,,,V8時間生存周期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x0000029E,LOG_V9_LIVE_TIME_START,,,V9时间生存周期开始,,V9 time life cycle begins,0,,,,V9時間生存周期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x0000029F,LOG_V9_LIVE_TIME_END,,,V9时间生存周期结束,,V9 time life cycle ends,0,,,,V9時間生存周期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000002A0,LOG_I1_LIVE_TIME_START,,,I1时间生存周期开始,,I1 time life cycle begins,0,,,,I1時間生存周期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000002A1,LOG_I1_LIVE_TIME_END,,,I1时间生存周期结束,,I1 time life cycle ends,0,,,,I1時間生存周期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000002A2,LOG_V10_LIVE_TIME_START,,,V10时间生存周期开始,,V10 time life cycle begins,0,,,,V10時間生存周期開始,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間",Set value: survival time. Actual value 1: survival time,
0x000002A3,LOG_V10_LIVE_TIME_END,,,V10时间生存周期结束,,V10 time life cycle ends,0,,,,V10時間生存周期結束,,,,,,,"时间:{0},设置值:{1}mS,实际值1:{2}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：生存时间,实际值1：生存时间差值","時間:{0},設置值:{1}mS,實際值1:{2}mS","Time :{0}, set value :{1}mS, actual value 1:{2}mS","設置值：生存時間,實際值1：生存時間差值",Set value: survival time. Actual value 1: survival time difference,
0x000002A4,LOG_CTRL_PRESS_SENDOR_CHLSET,,,控制工步-压力传感器控制操作,,Control step - pressure sensor control operation,-1,,,,控製工步-壓力傳感器控製操作,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：压力设置值,实际值1：同设置值为压力值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：壓力設置值,實際值1：同設置值為壓力值","Set value: pressure set value, actual value 1: the same as the set value is the pressure value",
0x000002A5,LOG_CTRL_AIRNITROGEN_STATUS,,,控制工步-真空氮气触发设置,,Control step - vacuum nitrogen trigger setting,-1,,,,控製工步-真空氮氣觸發設置,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：氮气设置值,实际值1：同设置值为氮气触发值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：氮氣設置值,實際值1：同設置值為氮氣觸發值",Set value: Nitrogen set value. Actual value 1: The same value is the nitrogen trigger value,
0x000002A6,LOG_IF_AIR_NITROGEN_STATUS_CHECK,,,真空氮气状态反馈如果条件判断,,Vacuum nitrogen status feedback condition judgment,2,,,,真空氮氣狀態反饋如果條件判斷,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：状态设置值,实际值1：实际状态值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：狀態設置值,實際值1：實際狀態值","Set value: Status Set value, actual value 1: actual status value",
0x000002A7,LOG_PROCESS_PAULSE_TRIGGER,,,压床流程暂停触发,,Press process pause trigger,0,,,,壓床流程暫停觸發,,,,,,,,,,,,,,
0x000002A8,LOG_MAIN_FAN_RECOVER_NORMAL,,,设备风机恢复正常,,The equipment fan returns to normal,0,,,,設備風機恢復正常,,,,,,,,,,,,,,
0x000002A9,LOG_FOLLOW_ACH_TEMP,,,温度跟随下发温度值,,Temperature Follows the delivered temperature,0,,,,溫度跟隨下發溫度值,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)",实际值1：同设置值为温度值,"時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃",實際值1：同設置值為溫度值,Actual value 1: The temperature is the same as the set value,
0x000002AA,LOG_CXCVX_ABNORMAL,,,转恒压充放电工步运行异常;可能电压精度异常or电流输出有异常;,,The constant X constant voltage charging and discharging process runs abnormally; the voltage accuracy may be abnormal or the current output is abnormal;,2,,,,恆X恆壓充放電工步運行異常;可能電壓精度異常or電流輸出有異常;,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3},实际值3:{4}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","实际值1：当前电流,实际值2：采样电流放大系数,实际值3：设置电流放大系数","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3},實際值3:{4}","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}, actual value 3:{4}","實際值1：當前電流,實際值2：采樣電流放大系數,實際值3：設置電流放大系數",Actual value 1: current current; Actual value 2: sampling current amplification coefficient; Actual value 3: Set current amplification coefficient,30
0x000002AB,LOG_IF_WX_TEMP_STATE,,,温箱温度状态如果条件判断,,Judgment of temperature condition of incubator,2,,,,溫箱溫度狀態如果條件判斷,,,,,,,"时间:{0},设置值:{1},实际值1:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：状态设置值,实际值1：实际状态值","時間:{0},設置值:{1},實際值1:{2}","Time :{0}, set value :{1}, actual value 1:{2}","設置值：狀態設置值,實際值1：實際狀態值","Set value: Status Set value, actual value 1: actual status value",
0x000002AC,LOG_START_WAIT_EXTDEVICE_STATE,,,开始等待第三方设备状态,,Start waiting for third party device status,0,,,,開始等待第三方設備狀態,,,,,,,,,,,,,,
0x000002AD,LOG_END_WAIT_EXTDEVICE_STATE,,,结束等待第三方设备状态,,End waiting for third party device status,0,,,,結束等待第三方設備狀態,,,,,,,,,,,,,,
0x000002AE,LOG_IF_PROTECTED,,,如果工步保护,,If step protection,2,,,,如果工步保護,,,,,,,,,,,,,,160
0x000002AF,LOG_EXP_CODER_LIMIT           ,,,"表达式编译文件长度限制,可能需要更新客户端重发流程",,Expression compile file length limits and may need to update the client resend process,1,,,,"表达式编译文件长度限制,可能需要更新客户端重发流程",,,,,,,,,,,,,,
0x000002B0,LOG_CHARGE_TO_VALUE_IN_TIMES    ,,,规定时间内充到设定电压,,Charge to set voltage within specified time,2,,,,规定时间内充到设定电压,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",
0x000002B1,LOG_MORE_CHARGE_TO_VALUE_IN_TIMES,,,规定时间内超出设定电压,,Exceeds the set voltage within the specified time,2,,,,规定时间内超出设定电压,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",
0x000002B2,LOG_CHARGE_VOL_RISERATE_ERR_IN_CAP,,,容量范围内充电电压上升速率,,Rate of charging voltage rise within capacity range,2,,,,容量范围内充电电压上升速率,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：电压变化设置值,实际值1：实际电压变化值,实际值2：当前电压","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V","設置值：電壓變化設置值,實際值1：實際電壓變化值,實際值2：當前電壓","Set value: Voltage change Set value, actual value 1: actual voltage change value, actual value 2: current voltage",40
0x000002B3,LOG_VOLTAGE_SUDDEN_DROP_IN_CAP   ,,,容量区间内-电压突降,,Capacity range. - Voltage drop,2,,,,容量区间内-电压突降,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：电压变化量设置值,实际值1：实际电压变化量,实际值2：当前电压","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V","設置值：電壓變化量設置值,實際值1：實際電壓變化量,實際值2：當前電壓",Set value: Voltage change Set value: Actual value 1: actual voltage change value 2: current voltage,40
0x000002B4,LOG_CURRENT_SUDDEN_RISE_IN_CAP   ,,,容量区间内-电流突升,,Capacity range. - Current surge,2,,,,容量区间内-电流突升,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A,实际值2:{3}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：电流变化量设置值,实际值1：实际电流变化量,实际值2：当前电流","時間:{0},設置值:{1}A,實際值1:{2}A,實際值2:{3}A","Time :{0}, set value :{1}A, actual value 1:{2}A, actual value 2:{3}A","設置值：電流變化量設置值,實際值1：實際電流變化量,實際值2：當前電流","Set value: Current change Set value, actual value 1: actual current change value, actual value 2: current current",40
0x000002B5,LOG_GLOBAL_TEMP_UPPER            ,,,全局保护-温度上限,,Global protection - Temperature upper limit,2,,,,全局保护-温度上限,,,,,,,,充电能量、放电能量字段复用，充电能量高4字节表示时间戳，低4字节表示触发保护时的温度值;放电能量高4字节表示设置的温度上限值，低4字节表示通道号,,,,,,60
0x000002B6,LOG_GLOBAL_TEMP_LOWER            ,,,全局保护-温度下限,,Global protection - Temperature lower limit,2,,,,全局保护-温度下限,,,,,,,,充电能量、放电能量字段复用，充电能量高4字节表示时间戳，低4字节表示触发保护时的温度值;放电能量高4字节表示设置的温度下限值，低4字节表示通道号,,,,,,60
0x000002B7,LOG_GLOBAL_VOL_EMERGSTOP_UPPER   ,,,全局保护-急停电压上限,,Global protection - upper limit of emergency stop voltage,2,,,,全局保护-急停电压上限,,,,,,,,充电能量、放电能量字段复用，充电能量高4字节表示时间戳，低4字节表示触发保护时的电压值;放电能量高4字节表示设置的电压上限值，低4字节表示通道号,,,,,,20
0x000002B8,LOG_GLOBAL_VOL_EMERGSTOP_LOWER,,,全局保护-急停电压下限,,Global protection - Lower limit of emergency stop voltage,2,,,,全局保护-急停电压下限,,,,,,,,充电能量、放电能量字段复用，充电能量高4字节表示时间戳，低4字节表示触发保护时的电压值;放电能量高4字节表示设置的电压下限值，低4字节表示通道号,,,,,,20
0x000002B9,LOG_PLC_SYSTEM_AIR_STATUS_ERR     ,,,压床系统气压状态异常保护,,Abnormal protection of air pressure state of press system,2,,,,壓床系統氣壓狀態異常保護,,,,,,,,,,,,,,160
0x000002BA,LOG_PLC_EMERGENCY_POWEROFF_PROTECT,,,压床紧急断电保护,,Press emergency power-off protection,2,,,,壓床緊急斷電保護,,,,,,,,,,,,,,160
0x000002BB,LOG_PLC_FIRE_RESPORNSE_LEVEL1     ,,,压床消防响应级别1级保护,,Press fire response level 1 protection,2,,,,壓床消防響應級別1級保護,,,,,,,,,,,,,,160
0x000002BC,LOG_PLC_FIRE_RESPORNSE_LEVEL2     ,,,压床消防响应级别2级保护,,Press fire response level 2 protection,2,,,,壓床消防響應級別2級保護,,,,,,,,,,,,,,160
0x000002BD,LOG_PLC_FIRE_RESPORNSE_LEVEL3,,,压床消防响应级别3级保护,,Press fire response level 3 Protection,2,,,,壓床消防響應級別3級保護,,,,,,,,,,,,,,160
0x000002BE,LOG_V10_PROTECT,,,V10组合保护,,V10 combined protection,2,,,,V10組合保護,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电压设置值,实际值1：电压实际值","時間:{0},設置值:{1}V,實際值1:{2}V","Time :{0}, set value :{1}V, actual value :{2}V","設置值：電壓設置值,實際值1：電壓實際值","Set value: Set voltage value, actual value 1: actual voltage value",160
0x000002BF,LOG_WX_CHL_ABNORMAL,,,温箱中通道存在异常,,Abnormal channel protection in temperature box,2,,,,溫箱中通道存在異常,,,,,,,,,,,,,,
0x000002C0,LOG_ABSOLUTE_TIME_LIMIT_CONDITION_JUMP,,,绝对时间限制条件跳转,,Absolute time limit jump,-1,,,,絕對時間限製條件跳轉,,,,,,,"时间:{0},设置值:{1},实际值1:{2},实际值2:{3}mS","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：绝对时间设置值,实际值1：当前系统时间,实际值2：时间间隔","時間:{0},設置值:{1},實際值1:{2},實際值2:{3}mS","Time :{0}, set value :{1}, actual value :{2}, actual value :{3}mS","設置值：絕對時間設置值,實際值1：當前系統時間,實際值2：時間間隔",Set value: Absolute time Set value. Actual value 1: current system time. Actual value 2: interval,
0x000002C1,LOG_2X_UPGRADE_4X,,,2X升级4X成功,,2X upgrade 4X successful,0,,,,2X升級4X成功,,,,,,,,,,,,,,
0x000002C2,LOG_IF_BAR_PRESS,,,如果条件-压强保护,,If conditions - pressure protection,2,,,,如果條件-壓強保護,,,,,,,"时间:{0},设置值:{1}bar,实际值1:{2}bar","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：压强设置值,实际值1：压强温度值","時間:{0},設置值:{1}bar,實際值1:{2}bar","Time :{0}, set value :{1}bar, actual value :{2}bar","設置值：壓強設置值,實際值1：壓強溫度值","Set value: pressure set value, actual value 1: pressure temperature value",160
0x000002C3,LOG_IF_INLET_TEMP,,,如果条件-进水口温度保护,,If conditions - inlet temperature protection,2,,,,如果條件-進水口溫度保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：實際溫度值","Set value: Temperature set value, actual value 1: actual temperature",60
0x000002C4,LOG_IF_OUTLET_TEMP,,,如果条件-出水口温度保护,,If conditions - outlet temperature protection,2,,,,如果條件-出水口溫度保護,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：温度设置值,实际值1：实际温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃","Time :{0}, set value :{1}℃, actual value :{2}℃","設置值：溫度設置值,實際值1：實際溫度值","Set value: Temperature set value, actual value 1: actual temperature",60
0x000002C5,LOG_FORM_CAP_LOWER,,,化成阶段完成后，最后一个恒流充电工步的最后一条数据的容量不在设定的范围，下限,,"When the formation test ends, the last capcity in last CCCahrge step beyond the limit，the lower limit.",2,,,,化成階段完成後，最後壹個恒流充電工步的最後壹條數據的容量不在設定的範圍，下限,,,,,,过放保护容量,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：容量设置值,实际值1：当前容量","時間:{0},設置值:{1}Ah,實際值1:{2}Ah","Time :{0}, set value :{1}Ah, actual value :{2}Ah","設置值：容量設置值,實際值1：當前容量",Set value: Capacity Set value. Actual value 1: Current capacity,40
0x000002C6,LOG_DVDT_CONDITION_JUMP,,,dV/dT截止条件跳转,,dV/dT cut-off conditional jump,-1,,,,dV/dT截止條件跳轉,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压设置值,实际值1：实际电压变化值,实际值2：当前电压,实际值3：前面电压","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓設置值,實際值1：實際電壓變化值,實際值2：當前電壓,實際值3：前面電壓","Set value: Voltage set value, actual value 1: actual voltage change value, actual value 2: current voltage, actual value 3: previous voltage",
0x000002C7,LOG_IDLE_VOL_ABNORMAL_PROTECT_ALL,,,空闲电压异常整机保护,,The device is protected when the idle voltage is abnormal,2,,,,空閑電壓異常整機保護,,,,,,,,,,,,,,20
0x000002C8,LOG_IDLE_CHL_TRIGGER_WARN_TEMP_UP,,,空闲通道触发温度预警上限,,Idle channel trigger temperature warning upper limit,2,,,,空閑通道觸發溫度預警上限,,,,,,,,充电能量、放电能量字段复用，充电能量高4字节表示时间戳，低4字节表示触发保护时的温度值;放电能量高4字节表示设置的空闲通道触发温度预警上限值，低4字节表示通道号,,,,,,60
0x000002C9,LOG_IDLE_CHL_TRIGGER_ALARM_TEMP_UP,,,空闲通道触发温度报警上限,,Idle channel trigger temperature alarm upper limit,2,,,,空閑通道觸發溫度報警上限,,,,,,,,充电能量、放电能量字段复用，充电能量高4字节表示时间戳，低4字节表示触发保护时的温度值;放电能量高4字节表示设置的空闲通道触发温度预警上限值，低4字节表示通道号,,,,,,60
0x000002CA,LOG_HEIGHTGAUGE_DEVICE_COMM_ERR,,,高度规设备通讯异常,,Abnormal communication of altitude gauge equipment,2,,,,高度規設備通訊異常,,,,,,,,,,,,,,
0x000002CB,LOG_PRESS_DEVICE_COMM_ERR,,,压力设备通讯异常,,Abnormal communication of pressure equipment,2,,,,壓力設備通訊異常,,,,,,,,,,,,,,
0x000002CC,LOG_CTRL_ENVIRONMENT_PARAM,,,环境参数设置控制工步,,Environmental parameter setting control step,-1,,,,環境參數設置控製工步,,,,,,,"时间:{0},设置值:{1},实际值1:{2}℃,实际值2:{3}L/min,实际值2:{4}KG","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,NULL)","设置值：参数类型,实际值1：温度,实际值2：流量,实际值3：压力","時間:{0},設置值:{1},實際值1:{2}℃,實際值2:{3}L/min,實際值2:{4}KG","Time :{0}, set value :{1}, actual value :{2}℃, actual value 2:{3}L/min, actual value 2:{4}KG","設置值：參數類型,實際值1：溫度,實際值2：流量,實際值3：壓力","Setting value: Parameter type, actual value 1: temperature, actual value 2: flow, actual value 3: pressure",
0x000002CD,LOG_GLOBAL_IDLE_CUR_LIMIT_UPPER,,,全局保护-闲时电流保护上限,,Global protection - upper limit of idle time current protection,2,,,,全局保護-閑時電流保護上限,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电流设置值,实际值1：电流实时值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：電流設置值,實際值1：電流實時值","Set value: Current set value, actual value 1: current real-time value",10
0x000002CE,LOG_GLOBAL_IDLE_CUR_LIMIT_LOWER,,,全局保护-闲时电流保护下限,,Global protection - lower limit of idle time current protection,2,,,,全局保護-閑時電流保護下限,,,,,,,"时间:{0},设置值:{1}A,实际值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：电流设置值,实际值1：电流实时值","時間:{0},設置值:{1}A,實際值1:{2}A","Time :{0}, set value :{1}A, actual value :{2}A","設置值：電流設置值,實際值1：電流實時值","Set value: Current set value, actual value 1: current real-time value",10
0x000002CF,LOG_VOLTAGE_MAXMIN_CHANGE_RATE,,,电压变化率,,Voltage change rate,1,,,,電壓變化率,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压变化率设置值,实际值1：最高最低电压差值,实际值2：最高电压值,实际值3：最低电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓變化率設置值,實際值1：最高最低電壓差值,實際值2：最高電壓值,實際值3：最低電壓值","Set value: Voltage change rate Set value, actual value 1: difference between the highest voltage and the lowest voltage, actual value 2: highest voltage, actual value 3: lowest voltage",
0x000002D0,LOG_TEMP_MAXMIN_CHANGE_RATE,,,温度变化率,,Temperature change rate,1,,,,溫度變化率,,,,,,,"时间:{0},设置值:{1}℃,实际值1:{2}℃,实际值2:{3}℃,实际值3:{4}℃","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：温度变化率设置值,实际值1：最高最低温度差值,实际值2：最高温度值,实际值3：最低温度值","時間:{0},設置值:{1}℃,實際值1:{2}℃,實際值2:{3}℃,實際值3:{4}℃","Time :{0}, set value :{1}℃, actual value 1:{2}℃, actual value 2:{3}℃, actual value 3:{4}℃","設置值：溫度變化率設置值,實際值1：最高最低溫度差值,實際值2：最高溫度值,實際值3：最低溫度值","Set value: Temperature change rate Set value, actual value 1: difference between the highest and lowest temperature, actual value 2: highest temperature, actual value 3: lowest temperature",
0x000002D1,LOG_SET_ENVIRONMENT_PARAM,,,环境参数设置,,Environmental parameter setting,0,,,,環境參數設置,,,,,,,"时间:{0},设置值:{1},实际值1:{2}℃,实际值2:{3}L/min,实际值3:{4}KG,实际值4:{5}Kpa","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：参数类型,实际值1：温度,实际值2：流量,实际值3：压力,实际值4：气压","時間:{0},設置值:{1},實際值1:{2}℃,實際值2:{3}L/min,實際值3:{4}KG,實際值4:{5}Kpa","Time: {0}, setting: {1}, the actual value 1: {2} ℃, the actual value 2: {3} L/min, the actual value 3: {4} KG, actual value 4: {5} Kpa","設置值：參數類型,實際值1：溫度,實際值2：流量,實際值3：壓力,實際值4：氣壓","Setting value: Parameter type, actual value 1: temperature, actual value 2: flow, actual value 3: pressure, actual value 4: air pressure",
0x000002D2,LOG_DOOR_ABNORMAL,,,门禁异常,,Abnormal access control,2,,,,門禁異常,,,,,,,,,,,,,,
0x000002D3,LOG_POWER_CPD_LIMIT_UPPER,,,恒功率放电时功率上限保护,,Constant power discharge,2,,,,恒功率放電時功率上限保護,,,,,,,"时间:{0},设置值:{1}W,实际值1:{2}W","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：功率上限设置值,实际值1：当前功率值","時間:{0},設置值:{1}W,實際值1:{2}W","Time :{0}, set value :{1}W, actual value :{2}W","設置值：功率上限設置值,實際值1：當前功率值",Set value: Set power upper limit. Actual value 1: current power value,160
0x000002D4,LOG_POWER_CPD_LIMIT_LOWER,,,恒功率放电时功率下限保护,,Constant power discharge,2,,,,恒功率放電時功率下限保護,,,,,,,"时间:{0},设置值:{1}W,实际值1:{2}W","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：功率下限设置值,实际值1：当前功率值","時間:{0},設置值:{1}W,實際值1:{2}W","Time :{0}, set value :{1}W, actual value :{2}W","設置值：功率下限設置值,實際值1：當前功率值",Set value: power lower limit Set value. Actual value 1: current power value,160
0x000002D5,LOG_POWER_CPC_LIMIT_UPPER,,,恒功率充电时功率上限保护,,Constant power flush,2,,,,恒功率充電時功率上限保護,,,,,,,"时间:{0},设置值:{1}W,实际值1:{2}W","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：功率上限设置值,实际值1：当前功率值","時間:{0},設置值:{1}W,實際值1:{2}W","Time :{0}, set value :{1}W, actual value :{2}W","設置值：功率上限設置值,實際值1：當前功率值",Set value: Set power upper limit. Actual value 1: current power value,160
0x000002D6,LOG_POWER_CPC_LIMIT_LOWER,,,恒功率充电时功率下限保护,,Constant power flush,2,,,,恒功率充電時功率下限保護,,,,,,,"时间:{0},设置值:{1}W,实际值1:{2}W","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：功率下限设置值,实际值1：当前功率值","時間:{0},設置值:{1}W,實際值1:{2}W","Time :{0}, set value :{1}W, actual value :{2}W","設置值：功率下限設置值,實際值1：當前功率值",Set value: power lower limit Set value. Actual value 1: current power value,160
0x000002D7,LOG_ABNORMAL_POWEROFF_CANNOT_RESUME,,,"存在异常断电或系统只读,不可直接接续,请检查硬件供电电路并检查数据完整性，如有需要通过历史数据接续",,"If an abnormal power failure occurs,or system read-only, the connection cannot be directly connected. Check the hardware power supply circuit and data integrity. If necessary, use historical data connection",2,,,,"存在異常斷電或系統只讀,不可直接接續,請檢查硬件供電電路並檢查數據完整性，如有需要通過歷史數據接續",,,,,,,,,,,,,,100
0x000002D8,LOG_ABNORMAL_RESTART_CANNOT_RESUME,,,"存在异常重启或系统只读,不可直接接续,请检查数据完整性，如有需要通过历史数据接续",,"If an abnormal restart occurs,or system read-only, the connection cannot be directly connected. Please check data integrity and connect historical data if necessary",2,,,,"存在異常重啟或系統只讀,不可直接接續,請檢查數據完整性，如有需要通過歷史數據接續",,,,,,,,,,,,,,
0x000002D9,LOG_IF_PROTECT_PRESS_N_ERR_TOPLIMIT,,,条件工步力(牛顿)保护(上限),,Conditional working step force (Newton) protection (upper limit),1,,,,條件工步力(牛頓)保護(上限),,,,,,,"时间:{0},设置值:{1}N,实际值:{2}N","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：如果条件力(牛顿)设置值,实际值：当前力(牛顿)值","時間:{0},設置值:{1}N,實際值:{2}N","Time :{0}, set value :{1}N, actual value :{2}N","設置值：如果條件力(牛頓)設置值,實際值：當前力(牛頓)值","Set value: If the conditional force (Newton) is set, the actual value: the current force (Newton) value",160
0x000002DA,LOG_IF_PROTECT_PRESS_N_ERR_LOWLIMIT,,,条件工步力(牛顿)保护(下限),,Conditional working step force (Newton) protection (lower limit),1,,,,條件工步力(牛頓)保護(下限),,,,,,,"时间:{0},设置值:{1}N,实际值:{2}N","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：如果条件力(牛顿)设置值,实际值：当前力(牛顿)值","時間:{0},設置值:{1}N,實際值:{2}N","Time :{0}, set value :{1}N, actual value :{2}N","設置值：如果條件力(牛頓)設置值,實際值：當前力(牛頓)值","Set value: If the conditional force (Newton) is set, the actual value: the current force (Newton) value",160
0x000002DB,LOG_FOLLOW_CCCV_SET_CHARGE,,,"跟随工步,恒流恒压类型,下发充电参数",,"Follow the work step, constant current and constant voltage type, and issue charging parameters",-1,,,,"跟隨工步,恒流恒壓類型,下發充電參數",,,,,,,"时间:{0},电压值:{1}V,电流值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：恒流恒压的电压值,实际值1：恒流恒压的电流值","時間:{0},電壓值:{1}V,電流值1:{2}A","Time :{0}, voltage value :{1}V, current value :{2}V","設置值：恒流恒壓的電壓值,實際值1：恒流恒壓的電流值",Voltage value: Voltage value of constant current and voltage. Current value 1: Constant current and voltage current value,
0x000002DC,LOG_FOLLOW_CCCV_SET_DISCHAREGE,,,"跟随工步,恒流恒压类型,下发放电参数",,"Follow the work step, constant current and constant voltage type, and issue discharge parameters",-1,,,,"跟隨工步,恒流恒壓類型,下發放電參數",,,,,,,"时间:{0},电压值:{1}V,电流值1:{2}A","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：恒流恒压的电压值,实际值1：恒流恒压的电流值","時間:{0},電壓值:{1}V,電流值1:{2}A","Time :{0}, voltage value :{1}V, current value :{2}V","設置值：恒流恒壓的電壓值,實際值1：恒流恒壓的電流值",Voltage value: Voltage value of constant current and voltage. Current value 1: Constant current and voltage current value,
0x000002DD,LOG_IF_PROTECT_SUM_CAP,,,如果工步总容量保护,,If the total capacity of the work step is protected,1,,,,如果工步總容量保護,,,,,,,"时间:{0},设置值:{1}Ah,实际值1:{2}Ah","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置总容量值,实际值1：实际总容量值","時間:{0},設置值:{1}Ah,實際值1:{2}Ah","Time :{0}, set value :{1}Ah, actual value :{2}Ah","設置值：設置總容量值,實際值1：實際總容量值",Set value: Set total capacity value. Actual value 1: actual total capacity value,40
0x000002DE,LOG_IF_PROTECT_SUM_ENERGY,,,如果工步总能量保护,,If the total energy of the work step is protected,1,,,,如果工步總能量保護,,,,,,,"时间:{0},设置值:{1}Wh,实际值1:{2}Wh","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：设置总能量值,实际值1：实际总能量值","時間:{0},設置值:{1}Wh,實際值1:{2}Wh","Time :{0}, set value :{1}Wh, actual value :{2}Wh","設置值：設置總能量值,實際值1：實際總能量值",Set value: Set total energy value. Actual value 1: actual total energy value,160
0x000002DF,LOG_4X_UPGRADE_5X,,,4X升级5X成功,,4X upgrade 5X successful,0,,,,4X升級5X成功,,,,,,,,,,,,,,
0x000002E0,LOG_BMS485_BUS_COMM_TYPE_CLASH,,,"BMS485总线通信类型存在冲突,请清空当前通道重新编辑正确工步后再启动",,There is a conflict in the communication type of the BMS485 bus. Please clear the current channel and edit the correct steps before starting again,1,,,,"BMS485總線通信類型存在沖突,請清空當前通道重新編輯正確工步後再啟動",,,,,,,,,,,,,,
0x000002E1,LOG_EXE_PULSE_FAILED,,,执行脉冲工步失败,,Failed to execute pulse step,1,,,,執行脈沖工步失敗,,,,,,,,,,,,,,
0x000002E2,LOG_DFDT_PRESSN,,,DF/dt保护,,DF/dt protection,1,0,,,DF/dt保護,,,,,,,"时间:{0},设置值:{1}N,实际值1:{2}N,实际值2:{3}N,实际值3:{4}N","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：膨胀力设置值,实际值1：膨胀力差值,实际值2：当前膨胀力值,实际值3：上次膨胀力值","時間:{0},設置值:{1}N,實際值1:{2}N,實際值2:{3}N,實際值3:{4}N","Time :{0}, set value :{1}N, actual value 1:{2}N, actual value 2:{3}N, actual value 3:{4}N","設置值：膨脹力設置值,實際值1：膨脹力差值,實際值2：當前膨脹力值,實際值3：上次膨脹力值","Set value: Set pressure value, actual value 1: pressure value difference, actual value 2: current pressure value, actual value 3: last pressure value",
0x000002E3,LOG_EXTDEV_CTRL,,,外部设备控制工步,,External device control steps,-1,,,,外部設備控製工步,,,,,,,,,,,,,,
0x000002E4,LOG_IF_EXTDEV,,,如果条件-外部设备信号判断,,If conditions - Judgment of external device signal parameters,2,,,,如果條件-外部設備信號判斷,,,,,,,"时间:{0},设置值:{1},实际值:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)","设置值：外部设备信号判断比较值,实际值：外部设备信号判断实际值","時間:{0},設置值:{1},實際值:{2}","Time :{0}, set value :{1}, actual value :{2}","設置值：外部設備信號判斷比較值,實際值：外部設備信號判斷實際值","Set value: external device signal Set value, actual value: external device signal Real-time value",
0x000002E5,LOG_EXTDEV_SET_CTRLSTEP_CUSTOM_FAIL,,,控制工步设置转换器-自定义协议失败,,Control step setting converter - custom protocol failed,2,,,,控製工步設置轉換器-自定義協議失敗,,,,,,,,,,,,,,
0x000002E6,LOG_EXTDEV_SET_CTRLSTEP_SPECIFY_FAIL,,,控制工步设置转换器-特定协议失败,,Control step setting converter - specific protocol failure,2,,,,控製工步設置轉換器-特定協議失敗,,,,,,,,,,,,,,
0x000002E7,LOG_EXTDEV_CTRL_LEVEL_NO_IN_MAP,,,控制工步设置转换器的层号不在映射表中,,The layer number of the control step setting converter is not in the mapping table,2,,,,控製工步設置轉換器的層號不在映射表中,,,,,,,,,,,,,,
0x000002E8,LOG_IF_DFDT_PRESSN_TOPLIMIT,,,如果工步膨胀力变化率保护上限,,If conditions - DF/dt protection ,1,0,,,如果工步膨脹力變化率保護上限,,,,,,,"时间:{0},设置值:{1}N,实际值1:{2}N,实际值2:{3}N,实际值3:{4}N","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：膨胀力设置值,实际值1：膨胀力差值,实际值2：当前膨胀力值,实际值3：上次膨胀力值","時間:{0},設置值:{1}N,實際值1:{2}N,實際值2:{3}N,實際值3:{4}N","Time :{0}, set value :{1}N, actual value 1:{2}N, actual value 2:{3}N, actual value 3:{4}N","設置值：膨脹力設置值,實際值1：膨脹力差值,實際值2：當前膨脹力值,實際值3：上次膨脹力值","Set value: Set pressure value, actual value 1: pressure value difference, actual value 2: current pressure value, actual value 3: last pressure value",
0x000002E9,LOG_IF_EXTDEV_SET,,,外部设备参数如果条件设置,,"External device parameters, if conditions are set",0,,,,外部設備參數如果條件設置,,,,,,,,,,,,,,
0x000002EA,LOG_EXTDEV_PARSE_CTRL_ERR,,,解析外部设备参数控制信息错误,,Error in parsing external device parameter control information,2,,,,解析外部設備參數控製信息錯誤,,,,,,,,"充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)",,,,,,
0x000002EB,LOG_IF_EXTDEV_BLOCK,,,如果条件-外部参数判断,,If conditions - Judgment of external device parameters,2,,,,如果條件-外部參數判斷,,,,,,,"时间:{0},设置值:{1},实际值:{2}","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(功能块编号,null)","设置值：外部参数判断比较值,实际值：外部参数实际值","時間:{0},設置值:{1},實際值:{2}","Time :{0}, set value :{1}, actual value :{2}","設置值：外部參數判斷比較值,實際值：外部參數實際值","Set value: external device value comparison, actual value: external device Real-time value",
0x000002EC,LOG_VOL_LINE_MISTAKE_PROOFING,,,电压线接线防呆-主辅电压差超出设定范围,,Voltage line is mistake proofing,1,0,,,電壓線接線防呆-主輔電壓差超出設定範圍,,,,,,,"时间:{0},设置值:{1}V,实际值1:{2}V,实际值2:{3}V,实际值3:{4}V","充电容量(时间,设置值)/放电容量(通道-条件类型-BMS,实际值1)/充电能量(实际值2,实际值3)","设置值：电压差设置值,实际值1：主辅电压差实际值,实际值2：主通道电压值,实际值3：辅助通道电压值","時間:{0},設置值:{1}V,實際值1:{2}V,實際值2:{3}V,實際值3:{4}V","Time :{0}, set value :{1}V, actual value 1:{2}V, actual value 2:{3}V, actual value 3:{4}V","設置值：電壓差設置值,實際值1：主輔電壓差實際值,實際值2：主通道電壓值,實際值3：輔助通道電壓值","Set value: Set value of difference voltage, actual value 1: actual value of difference voltage, actual value 2: voltage of the main channel, actual value 3: voltage of the auxiliary channel",
0x000002ED,LOG_XWJ_CHL_MODULE_ERR,,,通道模块故障,,Channel Module error,2,0,,,通道模塊故障,,,,,,,"时间:{0},故障码:{1}","充电容量(时间,null)/放电容量(故障码,设备代码)",故障码：通道模块故障码,"時間:{0},故障碼:{1}","Time :{0}, fault code :{1}",故障碼：道模塊故障碼,Fault Code: Channel Module Fault Code,
